Imaging Metal-on-Metal Hip Replacements: the Norwich Experience
- First Online:
- Cite this article as:
- Thomas, M.S., Wimhurst, J.A., Nolan, J.F. et al. HSS Jrnl (2013) 9: 247. doi:10.1007/s11420-013-9357-5
- 1k Downloads
Adverse reaction to metal debris is a relatively recently described and often a silent complication of metal-on-metal (MOM) total hip replacements (THR). The Norfolk & Norwich University Hospital has been performing metal artefact reduction (MARS) MRI for 8 years in a variety of different types of MOM THR.
The aims of this review are to describe the experience of using MARS MRI in Norwich and to compare our experience with that published by other groups.
A MEDLINE keyword search was performed for studies including MRI in MOM THR. Relevant publications were reviewed and compared with published data from the Norfolk & Norwich University Hospital. The similarities and differences between these data were compared and possible explanations for these discussed.
MARS MRI appears to be the most useful tool for diagnosing, staging and monitoring adverse reactions to metal debris (ARMD). There appears to be no clinically useful association between clinical and serological markers of disease and the severity of MR findings. Although severe early ARMD is associated with significant morbidity, mild disease is often stable for years. If patients with normal initial MR examinations develop ARMD, this usually occurs 7 years. A 1-year interval between MRI examinations is reasonable in asymptomatic patients.
There is a general international consensus that ARMD is prevalent in symptomatic and asymptomatic patients with MOM THR and that while appearances vary with the type of prosthesis, there are characteristic features that make MARS MRI essential for diagnosis, staging and surveillance of the disease.
Keywordship arthroplasty metal-on-metal ALVAL MRI
Orthopaedic surgery in Norwich first gained recognition with Tommy Brittain who pioneered extra-articular arthrodesis for arthritis , but it was his one-time registrar Ken McKee who set the scene for the later experience in imaging of metal-on-metal (MOM) arthroplasties. Ken McKee had also worked for Philip Wiles who in 1938 performed what is widely recognized as the first documented total hip replacement (THR) at the Middlesex Hospital, in London, using stainless steel acetabular and femoral prostheses . McKee was hooked. While his medical career was interrupted by active service with the Royal Army Medical Corp during World War II, he continued to think about the possibilities of hip arthroplasty writing in 1940 that “If one could replace the bearing of a motor car then it must be possible in human joints” (personal communication: Mr Hugh Phillips, Consultant Surgeon, Norwich, 2002).
The McKee–Farrar became the first widely implanted total hip replacement, but it was not without its problems. Early- to medium-term failure was caused by mechanical loosening with two thirds of these hip replacements demonstrating radiographic loosening followed by a revision rate of 10% to 17% by 10 years [3, 11]. The cause of this relatively high rate of early aseptic loosening has been attributed to both surgical technique and biomechanical problems of the bearing itself, which were felt to reflect the limits of the engineering techniques of the time prior to 1971. The McKee–Farrar was an equatorial loaded bearing as opposed to later polar bearing designs, and this was considered to be a major cause of loosening in the earlier implants due to high friction. Elevated serum metal ion levels were found in patients with MOM prostheses, and the concern that this may be associated with carcinogenicity contributed to the decline in popularity of the McKee–Farrar and the coming of age of Charnley’s metal on polyethylene (MOP) “low friction arthroplasty” in the early 1970s  despite the fact that the intermediate- and long-term results of the McKee–Farrar and Charnley were comparable [3, 24].
After the predominance of MOP systems in the late 1970s and 1980s, aseptic loosening caused by osteolysis from polyethylene wear debris became an increasingly common problem. In an attempt to find an alternative, manufacturers revisited the MOM concept. Long-term data revealed that when MOM implants were successful, outcomes were excellent with significantly lower wear (two orders of magnitude lower) of the metal bearing components than MOP arthroplasty [2, 24, 39] with reports of MOM bearings lasting for over two decades . The expectation was that with improved tolerances resulting from new machining techniques, the short- to medium-term failures of the McKee–Farrar could be addressed, and so by 1988, MOM hip replacement systems had been reintroduced and by 1997 MOM arthroplasty had returned to Norwich.
This review describes the experience of imaging these modern MOM THR in Norwich and compares our experience with that of other groups. In particular, this review will focus on four specific questions. These are: (1) What are the MR appearances of the normal post-operative hip? (2) What are the characteristic MR features of ARMD? (3) How do these features correlate with other biomarkers of disease? (4) What is the prognosis for patients with mild disease or normal MRI early in their post-operative course?
Method and Materials
On the 14th July 2013, a word search of MEDLINE was performed using the terms “metal-on-metal” AND “MRI”. This search produced 51 articles. Case reports, reviews, letters and lecture notes (12) were excluded along with publications that did not specifically report analysis of radiological findings in their results (11), any publications from Norwich (6) and a single paper not in English. This left 21 original scientific articles from institutions other than our own. The comparison of these papers with the results from Norwich forms the basis of this review.
Types of MOM THR
Experience of imaging failing MOM arthroplasties in Norwich includes several different types including total hybrid, total uncemented, large and small bearing and resurfacing arthroplasties.
Hybrid Total Hip Arthroplasty: Small Bearing
Uncemented Large Bearing: Total and Resurfacing
Between 2005 and 2008, 79 ASR-XL hip arthroplasties were performed in 68 patients (17 resurfacing and 62 total). Cobalt-chrome (CoCr) acetabular components and large bearing CoCr femoral heads were used in all cases, with the addition of a Corail titanium femoral prosthesis in the total hip arthroplasties . Following the experience with the Ultima TPS hybrid THR and following an index case that was revised early for pain and soft tissue abnormalities on MRI, the whole cohort of these patients underwent annual clinical assessment and self-assessment questionnaires including annual radiographs and MRI. There was a high early revision rate of 11% at 40 months for the ASR cohort  which was comparable with published results from other groups [15, 16, 17].
From 2001 to 2007, 463 Birmingham hip resurfacing arthroplasties (BHR) were performed in Norwich and a local district general hospital. The revision rate at 5 years was 3.1% with risk factors identified as female gender, high acetabular inclination, obesity and small femoral components .
The BHR is the most successful MOM prosthesis in the National Joint Registry of England & Wales , with a revision rate of 5.09% at 7 years, but even this compares poorly with a cemented Exeter MOP THR at 1.32%. The BHR is now reserved, by most British hip surgeons, for younger men with large femoral heads, who appear to have the lowest risk of revision.
One of the key radiological findings in ARMD is that conventional radiographs are usually normal. The most commonly recognized radiographic finding is resorption of the medial calcar, which is often subtle and does not reflect the sometimes extensive soft tissue changes [26, 47] (Fig. 1b, c). Interestingly some of the features that are now widely recognized on MRI were first described using conventional arthrography in 1975 although at the time ALVAL and ARMD were not recognized as a unique disease process .
Our experience of using CT for the detection of ARMD is limited to those patients with absolute contraindications to MR although some institutions appear to use it routinely with success . Others have used it for 3D measurements of the position of the prosthesis in which there appears to be no difference between symptomatic and asymptomatic patients supporting the hypothesis that patient-specific factors are more influential [16, 17, 29].
The Norwich MAR MR protocol at 1.5 T (Siemens Symphony, Erlangen, Germany)
Field of view (cm)
34 × 27
34 × 34
34 × 34
34 × 34
34 × 34
Slice thickness (mm)
Interslice gap (mm)
Receiver bandwidth (Hz/pixel)
Phase encoding direction
Not all radiologists use T2W sequences as part of a MARS MRI protocol. The longer echo times are more susceptible to dephasing, but we find this useful to identify microscopic metal debris which is not always visible on very short echo T1W or PD sequences.
This protocol comprises basic modifications to standard spin echo and STIR sequences. There are a number of more advanced techniques for metal artefact reduction which include view angle tilting and multi-spectral imaging techniques such as MAVRIC and SEMAC, but we have not used these as part of our routine practice [15, 21, 40]
MRI of the Normal THR
Early work optimizing MR for imaging concentrated on imaging the predominantly soft tissue complications of THR [35, 36, 45]. Soon after we started to MR image large numbers of patients with MOM THR, we realized that the normal post-operative MR appearances for THR were not clearly defined. This meant that some abnormalities such as muscle wasting might be attributable to the pre-existing disease or the operation. The presence of gluteus minimus atrophy and fluid collections around the greater trochanter had been previously demonstrated on MR imaging of asymptomatic patients following lateral transgluteal arthroplasty .
In order to define the normal spectrum of post-operative MRI findings, a cross-sectional observational study of 22 asymptomatic hips after MOP (n = 10) and MOM (n = 12) arthroplasty was performed. This revealed that short external rotator muscle atrophy was a standard finding in THRs performed using a posterior approach, even when the tendon was reattached, which it was in all these cases. It was also apparent that small simple thin-walled periprosthetic fluid collections were commonly found in the surgical bed and presumably represented post-operative seromas . Similar results have been described by other authors .
One of the original features described as part of the spectrum of findings in ARMD on MRI was bone marrow oedema on the STIR sequences, but it became apparent that the inter-observer reliability for this feature was poor and it was dropped from routine clinical reporting practices.
MRI of ARMD
The Norwich ASR and Ultima TPS cohorts are not directly comparable. The process of case selection was different: a cross-sectional study compared to a symptomatic case series. The ASR patients were also imaged considerably earlier in their postoperative course than the Ultima TPS group, and it is now clear that the mechanism of production of debris is different: crevice corrosion in the TPS and wear in the ASR.
It seems clear from our work and the work of others that the spectrum of ARMD on MRI does vary with the type of prosthesis. It seems likely that the bearing size, composition, method of fixation and replacement of the femoral neck may all contribute to the severity, distribution and MRI appearances of ARMD.
Grading ARMD with MRI
The key features of the Anderson staging system for severity of ARMD on MRI
This grading system is by no means universally accepted. The units at Oxford and Imperial College, London have both described their own scoring methods which are based on MR signal characteristics which the Oxford group have reported correlates with symptoms in that those patients with solid lesions are more likely to experience pain . This is a fundamentally different approach to the Norwich grading system which stages the severity of the disease in much the same way as a local staging of a cancer would be performed.
Correlation of MRI with Clinical Symptoms
In the Norwich cohort of ASR patients, 27 (34%) of the 75 imaged hips showed MRI features of ARMD. The prevalence of disease on MARS MRI in asymptomatic patients has been reported by other groups in a range of 4% to 68% with the highest prevalence in THR and the lowest in RSA [6, 17, 18, 29, 32, 44] . When the grade of disease on MR, or simply the presence or absence of ARMD, was compared with clinical symptom scores, there was no statistically significant correlation in the Norwich ASR cohort. The mean Oxford hip score (OHS) for patients with a normal MRI was 23 (original OHS, with “best” 12 and “worst” 48) and 19 in those with MR findings of ARMD. Severe ARMD was also observed in asymptomatic patients with OHS of 12 . These findings, which suggest that clinical scores cannot be used as a reliable tool for diagnosing or monitoring ARMD, have since been replicated by other groups . Other groups have taken a different approach to staging AMRD by using continuous measures of disease, rather than ordinal data, and have demonstrated that there is an association between synovial volume, and maximal synovial thickness, with the severity of ALVAL graded histologically  as well as a correlation between the severity of synovitis and symptoms in the presence of adverse local tissue reaction .
Correlation of MRI with Metal Ion Levels
In response to the concern over the increasing numbers of revisions for ARMD, the UK Medicines and Healthcare Products Regulatory Agency (MHRA) released a Medical Device Alert in April 2010 . These recommendations included regular clinical follow-up of patients and investigation of those patients with painful MOM arthroplasties with regular measurement of serum Co and Cr ion levels. Radiological follow-up was particularly recommended for those patients with serum metal ion levels above 7 ppb (in micrograms per litre).
In an attempt to assess whether or not serum metal ion levels are useful in monitoring ARMD, a retrospective cohort study of all patients who had received ASR implants and who had not undergone revisions were reviewed. Sixty-two hip replacements in 57 patients were included in the study . There were significant statistical associations between elevated serum metal ions and acetabular inclination greater than 50° (no association with acetabular version), larger heads, female gender (although head size may be a confounding influence), THR rather than resurfacings and bilateral implants.
Several questions arose from the MHRA Medical Device Alert and the recommendation for routine imaging of all patients with MOM THR. The minimal frequency of MR imaging required to adequately screen populations of relatively asymptomatic patients with MOM THR was not known because the natural history of the disease was not understood.
This study did not include a substantial number of patients in whom their MOM THR failed and were revised early, and so the conclusion is that for this cohort, ARMD appears to develop in the early post-operative phase. Those patients with severe disease present early on. Those with mild disease are often stable for many years, and only a small proportion of those with normal MRIs will develop ARMD but when they do it is usually 7 to 10 years after the operation. In light of this, annual assessment of asymptomatic patients with MOM THR with MAR MRI, as recommended by the MHRA, would seem to be an adequate frequency for screening, but our evidence does not necessarily apply to all types of prostheses.
ARMD is often a silent complication of MOM THR. Metal artefact reduction MRI allows adequate visualization of the periprosthetic soft tissues on most standard clinical MR systems. MR characteristics are now well described and appear to vary from prosthesis to prosthesis. There are different approaches to the grading of the severity of ARMD on MRI; however, there does not appear to be any good correlation between the severity of changes at MRI and either clinical symptoms scores or metal ion levels. Therefore, MRI appears to be the most useful tool for diagnosing and monitoring ARMD at this moment in time. Although severe early ARMD is associated with significant morbidity, mild disease is often stable for years. Depending on the type of prosthesis, a patient with a normal post-operative MRI may have a 10% chance of developing ARMD and if they do it seems to occur 7 to 11 years after surgery. A 1-year interval between MRI examinations is reasonable in asymptomatic patients.
DePuy International Ltd. (Leeds, UK) funded some, but not all, of the MRI scans used in three of the studies performed at the Norfolk & Norwich University Hospital, and cited in this review, as part of the Medicines and Healthcare products Regulatory Agency investigation in to the Ultima TPS hybrid THR.
Conflict of Interest:
Marianna S. Thomas, FRCR, John F. Nolan, FRCS(Orth) have declared that they have no conflict of interest. Andoni P. Toms FRCR, PhD received research support from DePuy International Ltd. (Leeds, UK) for the study. James A. Wimhurst, FRCS(Orth) received grant funding from Gwen Fish Trust for the study and assisted place attending AAOS 2011 for DePuy, outside the work.
This article does not contain any studies with human or animal subjects performed by the any of the authors.
Open Access This article is distributed under the terms of the Creative Commons Attribution License which permits any use, distribution, and reproduction in any medium, provided the original author(s) and the source are credited.