The journey of creating the first dedicated platform for robot-assisted (super)microsurgery in reconstructive surgery
- 218 Downloads
Incorporation of new technologies advances surgery in general. One example is robot assistance which can potentially overcome limitations in accessibility, dexterity, and ergonomics, thereby improving outcome of surgical procedures. To date, robot assistance has been used successfully in various fields of surgery. Also, in microsurgery robot assistance could offer great advantages as it requires great precision and meticulous technique. Currently available surgical robotic platforms are primarily designed for endoscopic procedures and lack the specific requirements for true microsurgery. Therefore, plastic surgeons of the Maastricht University Medical Center (Maastricht, the Netherlands) teamed up with engineers from the Eindhoven University of Technology (Eindhoven, the Netherlands) and developed the first dedicated robotic platform for (super)microsurgery, the MicroSure Robot (MSR).
Level of evidence: Level V, therapeutic study.
KeywordsRobot assistance Robotics Reconstructive surgery Microsurgery Supermicrosurgery
Surgical interventions are demanding processes, which require agile decision-making and meticulous execution. Especially in microsurgery qualities such as dexterity, maneuverability, steadiness, and technical quality are of utmost importance. The expertise needed to perform these procedures is acquired through extensive training and clinical experience. Over time, the instruments become extensions of the surgeons’ hands. Notwithstanding, the surgical qualities required to perform microsurgery are limited to the individual surgeon and are susceptible to human fatigue, aging, and mindset. Undesired involuntary and inadvertent movement results in damage of tissue and substandard operation outcome . Robot assistance can potentially overcome the aforementioned issues. Robot assistance provides filtering of tremor, enhances dexterity, and may enable more surgeons to perform (super)microsurgery. Incorporation of robotics in microsurgery would increase the quality of current microsurgical and supermicrosurgical procedures and create new possibilities in the field of (super)microsurgery. This master class article aims to describe the challenging journey of developing a dedicated robotic platform for (super)microsurgery—from inception to first introduction in a clinical setting.
Inception of the idea
From the 1970s onward, ideas have led to the development of the first (tele)robots in surgery . Whether it was to operate on astronauts in space or to assist and function as supporting manipulators, the ideas have evolved into a new era of robot-assisted surgery. The first Food and Drug Administration–approved robot was the da Vinci Surgical System (Intuitive Surgical, Inc.) which has been used successfully in various surgical specialties, such as cardiac surgery, neurosurgery, and urology [3, 4, 5, 6]. This robotic system was designed for endoscopic surgery and has been successfully used for endoscopic harvesting and in setting of flaps [7, 8]. However, the system showed significant limitations when used for microsurgical anastomoses in free flap reconstruction [9, 10, 11, 12, 13]. The actual idea of creating a dedicated robotic platform for microsurgery arose when the da Vinci system was used during the first robot-assisted microsurgical anastomosis in reconstructive microsurgery in 2007 . In a mini-muscle TRAM-flap breast reconstruction, the arterial anastomosis on the internal mammary artery was performed using robot assistance. Although the anastomosis was successfully completed, the authors concluded that the system had many limitations for application in microsurgery. The limited optic magnification and the bulky and powerful instruments of the system were significant drawbacks for use in true microsurgery.
Design concept phase
Based on this first experience with robot-assisted microsurgery, a Dutch collaboration between microsurgeons of the Maastricht University Medical Center (MUMC+, Maastricht, the Netherlands) and engineers of the Eindhoven University of Technology (TU/e, Eindhoven, the Netherlands) started the development of a dedicated robotic platform for (super)microsurgery: the Microsure Robot (MSR) . Analyses of conventional methodologies and available alternative solutions were taken into account as a guideline for the appropriate requirements of the MSR. The development of the MSR was primarily focused on improving the surgeons’ performance by filtering tremor and scaling movements, while at the same time being easily integrated into the existing workflow and infrastructure in the operating theatre. Complementary to this, but not least important, were safety, ease-of-use, and cost-effectiveness.
Suspension ring, that can be mounted onto the surgical table or the microscope
User interface, forceps-like joysticks which copy the surgeons’ movement to a micro-instrument held by the device (robotic arms)
Robotic arms, equipped with genuine microsurgical instruments which effectuate the surgeons’ movement
Food pedals, controlling the motion scaling
The device software, scaling down motion and filtering out tremor
The setup of the MSR allows the surgeon to freely use a conventional surgical microscope in combination with the device. The unchanged proximity of the surgeon to the patient allows the surgeon to get a direct view and access to the surgical field at all times if necessary and the robotic arms can easily be moved aside to continue the microsurgical procedure manually. This flexibility in switching between robot-assisted and conventional microsurgery during the procedure allows hybrid operations to be performed. This reduces total surgical time by enabling high-precision robot assistance only for phases of the operation where this is necessary.
The system is based on a modular design, permitting it to be used in different configurations depending on the level of required robot assistance. Up to four robotic arms can be used simultaneously, allowing one or two operating surgeons to control them by means of the user interface.
During the surgical procedure, the robot is entirely covered in sterile draping, except for the sterilized adapters that can be loaded with genuine microsurgical instruments.
This feature of the robotic platform allows the surgeon to use his or her preferred original instruments and skill set yet having the additional benefits of tremor filtration and motion scaling. The small footprint of the MSR creates a virtually unchanged setup and workflow in the operating theatre with respect to the conventional approach.
This study showed promising results of the feasibility of performing anastomoses using the MSR on silicone vessels in a pre-clinical setting. Although time to perform the anastomosis was longer using robot assistance, and quality assessed by the modified SAMS was lower compared with the conventional anastomoses, steeper learning curves were observed for the robot-assisted approach in both time and surgical skills.
Clinical evaluation and future perspectives
Evolution in technology has changed the possibilities in the medical field. Robot assistance in microsurgery can overcome human limitations in precision and ergonomics. Furthermore, additional benefits of robot assistance can be explored in the future as well, namely;
Conventional (super)microsurgery is performed without tactile feedback as forces are too low to rely on . Incorporation of enhanced tactile feedback in robot assistance would enable the surgeon to feel extremely delicate forces and therefore enhance tissue handling and precision which could reduce adverse events and improve patient outcome in (super)microsurgery.
The training of microsurgical skills is currently based on subjective observations of other trained microsurgeons. Parameters such as completion time, path length, depth perception, speed, smoothness, efficiency, and bi-manual dexterity are evaluated subjectively. Objective measurement of motion and forces can improve the evaluation of surgical performance and enhance microsurgical training and clinical care. Future surgeons will need to be trained in both conventional and robot-assisted microsurgery techniques to be capable of dealing with the wide variety of surgical problems that they will encounter throughout their career. Therefore, both robot-assisted and conventional microsurgical experience should be incorporated into surgical training .
Intra-operative imaging techniques
Robot assistance in microsurgery can be further enhanced by incorporating novel optical imaging techniques such as three-dimensional imaging, high-resolution stereotactic operation, spectral imaging, and real-time navigation. Intra-operative imaging such as near-infrared fluorescence imaging particularly offers anatomical guidance, such as better vascular and lymph and nerve visualization, relevant to (super)microsurgery .
As a subset of artificial intelligence, machine learning uses decision-making computer algorithms to understand and respond to specific data . Already applied in a wide spectrum of robotic applications, machine learning has only been considered for robotic surgery recently . The development of these cognitive surgical robots endows robotic systems with the ability of self-learning and cognitive skills, thereby enabling automation, which can improve and standardize our microsurgical techniques . Improving quality and efficiency in microsurgery extends beyond the surgical procedure. Combining patient-related data with robot registration can provide better understanding of the consequences of decisions and interventions during surgery, ultimately leading to better patient care.
The initial idea of developing a robot that is dedicated to (super)microsurgery resulted in a platform that is feasible to perform (super)microsurgical anastomosis. Robot assistance enables tremor filtration and enhances dexterity and can train more surgeons for (super)microsurgery. It has the potential to increase the quality of current microsurgical and supermicrosurgical procedures and create new possibilities in the field of (super)microsurgery. The increasing importance of quality, patient safety, and cost efficiency will propel the development of new robotic technology for microsurgery. Further research and clinical testing are needed to use robot assistance to its full potential and open new possibilities in (super)microsurgery.
This study did not receive any specific grant from funding agencies in the public, commercial and not-for-profit sectors.
Compliance with ethical standards
Conflict of interest
Tom van Mulken is Chief Medical Officer of the spin off company MicroSure and shareholder in the company. René van der Hulst is shareholder of MicroSure. Raimondo Cau is Chief Technical Officer and shareholder of MicroSure. The other authors declare that they have no conflict of interest.
Ethical approval for the pre-clinical study on rats was obtained from the Maastricht University Institutional Review Board for Animal Use in Research (Maastricht, the Netherlands; project reference: DEC 2014-078).
- 1.Riviere CN, Khosla PK (eds) (1998) Active handheld instrument for error compensation in microsurgery. Intelligent systems and advanced manufacturing. SPIEGoogle Scholar
- 2.Takacs A, Nagy DA, Rudas IJ, Haidegger T (2016) Origins of surgical robotics: from space to the operating room. Acta Polytech Hung 13(1):13–30Google Scholar
- 14.Cau R (2014) Design and realization of a master-slave system for reconstructive surgery, EindhovenGoogle Scholar
- 22.Kassahun Y, Yu B, Tibebu AT, Stoyanov D, Giannarou S, Metzen JH et al (2016) Surgical robotics beyond enhanced dexterity instrumentation: a survey of machine learning techniques and their role in intelligent and autonomous surgical actions. Int J Comput Assist Radiol Surg 11(4):553–568CrossRefGoogle Scholar
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.