A polarisation modulation scheme for measuring vacuum magnetic birefringence with static fields

A novel polarisation modulation scheme for polarimeters based on Fabry-Perot cavities is presented. The application to the proposed HERA-X experiment aiming to measuring the magnetic birefringence of vacuum with the HERA superconducting magnets is discussed.

Here The ellipticity ψ induced on a linearly polarised beam of light with wavelength λ passing through a medium with birefringence n and length L, and whose axes are defined by the external magnetic field, is ψ = π L λ n sin 2φ a e-mail: guido.zavattini@unife.it where φ is the angle between the magnetic field and the polarisation direction. The birefringence predicted by L EHW is [7][8][9][10][11] n = 3A e B 2 4 × 10 −24 B 2 .
Several experiments are underway, of which the most sensitive at present are based on polarimeters with very high finesse Fabry-Perot cavities and variable magnetic fields [12][13][14]. The Fabry-Perot cavity is necessary to increase the optical path L within the magnetic field region, whereas the variable magnetic field is necessary to induce a time dependent effect. Both of these aspects significantly increase the sensitivity of the polarimeters.
Ideas to use high field superconducting magnets, such as those used in the LHC and HERA accelerators, have also been suggested but their use is limited by the difficulty in modulating, in one way or another, their magnetic fields. To work around this problem, proposals of rotating the polarisation have been considered [15], but the presence of the Fabry-Perot cavity, whose mirrors always present an intrinsic birefringence whose induced ellipticity is orders of magnitude larger than the ellipticity due to vacuum magnetic birefringence, have made this idea unfeasible.
In this note, a novel modulation scheme is presented that might profitably be employed with large superconding magnets.

Preliminary considerations
In a recent workshop in Hamburg [16], a new scheme, presented in this paper, has been suggested to measure the magnetic birefringence of vacuum predicted on the basis of the 1936 effective Lagrangian L EHW . One of the presenta-tions [17], proposed the idea (called HERA-X: Heisenberg-Euler-biRefringent-ALPS-eXperiment) of making use of the powerful infrastructure of the ALPSIIc set-up [18]: about 5000 T 2 m, which could go up to about 7700 T 2 m if the peak field of 6.6 T is employed. In this configuration the magnetic birefringence would be n (HERA-X) ≈ 10 −22 for the 5.3 T magnetic field. With this birefringence, the maximum ellipticity is ψ (HERA-X) = 5 × 10 −14 for λ = 1064 nm and L = 177 m.
In the usual setups, the magnetic field is modulated to gain sensitivity. In the particular case of the HERA superconducting magnets the electric current can be modulated at about a millihertz frequency [19]. Let's analyse the measurement scheme of Fig. 1, featuring two crossed polarisers, a variable magnetic field (fixed direction) at a frequency ν B , and an ellipticity modulator at a frequency ν m . The purpose of the ellipticity modulator is twofold: it allows heterodyne detection for improving sensitivity by linearising the signal and shifting it to a high frequency; it permits the distinction between an ellipticity signal and a rotation signal [12]. In this scheme the intensity collected at the photodiode PDE is, at the lowest useful order, The interesting signal is found, in a Fourier transform of the signal from the photodiode, at the two sidebands ±ν B from the carrier frequency ν m of the ellipticity modulator.
The resulting peak shot-noise sensitivity in such a scheme is where e is the electron charge, I 0 is the intensity reaching the analyser, and q is the quantum efficiency of the photodiode. With I 0 100 mW and q = 0.7 A/W, the shot-noise peak sensitivity is S shot 2 × 10 −9 1/ √ Hz. Despite the exceptional parameters of the magnetic field of HERA-X, Fig. 1 A simple heterodyne ellipsometer. PDE extinction photodiode, PDT transmission photodiode the integration time T to achieve a unitary signal-to-noise ratio remains too long, even supposing to work at shot-noise sensitivity: T ∼ S shot ψ (HERA-X) 2 ∼ 10 9 s.
As mentioned above, further amplification is required. This can be achieved with a Fabry-Perot cavity, which can be thought of as a lengthening of the optical path by a factor N = 2F/π , where F is the finesse of the cavity. The proposed finesse for HERA-X is F = 60,000. With such a finesse, the ellipticity ψ increases by a factor N = 38,000 and the integration time therefore diminishes by a factor N 2 . Assuming shot-noise sensitivity, on paper, this device should easily allow the measurement.
A problem remains, however, regarding the actual sensitivity that one may reasonably think to achieve at low frequencies with such a long cavity. Let us consider the experiments on this subject realised so far with a scheme similar to the one proposed with HERA-X [12,[20][21][22][23]. In Fig. 2 we show the noise densities in birefringence π d measured in these apparatuses as a function of the frequency of the effect. In this formula S ψ is the ellipticity sensitivity of each experiment, λ is the wavelength, F is the finesse and d the cavity length. Note that the cavity length d has been used instead of the length L of the magnetic region; what is plotted is therefore the best sensitivity in birefringence that could be obtained by the experiments. In the figure we did not report a much worse sensitivity value of the Q &A experiment [13]. The data are fitted with a power function. The message put forward by Fig. 2 is that increasing the effective length (finesse and magnetic field length) does not guarantee the shortening of the necessary integration time to reach a unitary signal-to-noise ratio; seeking the highest finesse possible is not necessarily the optimal choice. Increasing the birefringence modulation frequency seems to be more effective. Furthermore, with lower finesses, the cavity will have a shorter decay time and therefore a higher cutoff frequency allowing higher modulation frequencies. Figure 2 suggests, therefore, that the finesse of the cavity should be the highest for which the polarimeter is still limited by intrinsic noises (shot-noise, Johnson-noise, etc.).
The figure suggests that it is unlikely that, at 1 mHz, a sensitivity better than S

Method
In this note, we present a novel modulation scheme that would bring in several advantages. This idea has never been tested in a laboratory, but is likely to be more effective than the one described above. In this way one can work at higher frequencies for the best sensitivity. In this scheme the magnetic field does not need to be modulated. The scheme consists in introducing a pair of co-rotating half-wave-plates L 1 and L 2 inside the Fabry-Perot cavity, as schematically shown in Fig. 3. The polarisation within the magnetic field would rotate at twice the frequency of the wave-plates and should allow to increase substantially the modulation frequency of the effect. An important feature of this scheme is that the polarisation direction of the light on the Fabry-Perot mirrors would remain fixed, thereby eliminating the contribution of the ellipticity due to the intrinsic birefringence of the mirrors. Furthermore, the polarisation direction on each mirror Fig. 3 Proposed modulation scheme. L 1,2 rotating half-wave-plates, PDE extinction photodiode, PDT transmission photodiode could be chosen; the input polariser defines the polarisation direction on the first mirror whereas on the second mirror the polarisation direction is defined by the relative angle between the axes of the two wave-plates.
Let us indicate with ν L the rotation frequency of the waveplates, that we suppose to rotate synchronously but not necessarily aligned one to the other. The Jones representation of the electric field at the exit of the cavity is where δ is the round-trip phase acquired by the light between the two cavity mirrors, R and T are the reflectivity and transmissivity of the mirrors, I is the identity matrix, is the magnetic birefringence of vacuum generating an ellipticity ψ in the polarisation of the light, and are the rotating wave-plates. Here represents the wave-plate and the rotation matrix, with φ the variable azimuthal angle of the wave-plates: φ(t) = 2πν L t. The angle φ 2 − φ 1 is the constant relative phase between the slow axes of the two rotating wave-plates, and α 1,2 allow for small deviations from π of the retardation of the two imperfect wave-plates. The electric field after the analyser is then 1 and A = 0 0 0 1 are the ellipticity modulator, placed at 45 • with respect to the output polarisation, and the analyser set to maximum extinction, respectively. In the expression for H, η(t) = η 0 cos 2πν m t. The rotation matrix between the cavity and the ellipticity modulator ensures that the modulator and the analyser are correctly oriented. To first order in α 1 , α 2 , and ψ, the intensity detected by the photodiode PDE is given by The interesting result from this formula is that the signal of the magnetic birefringence of vacuum is found at the frequencies ν m ± 4ν L deriving from the product η(t) ψ sin(4φ(t) + 4φ 1 ), while the signals due to a small retardance difference from λ/2 of the wave-plates appear at ν m ± 2ν L . Higher order imperfections in the phase delay of the rotating waveplates may be present and may be described by the following expansion: The various orders of imperfection can be estimated from the specifications of the producers. A typical value for α The main contribution for α (1) 1,2 may come from the parallelism of the surfaces of the wave-plate (wedge) coupled to the distance of the beam from the center of rotation. The typical parallelism of a wave-plate is 2 × 10 −6 rad. With an off-center rotation of ≈1 mm this gives an estimated value of α (1) 1,2 2×10 −9 . The effect of such a defect, though, would generate signals at ν m ± 3ν L and ν m ± ν L and not at ν m ± 4ν L . To generate a spurious signal at ν m ± 4ν L the term α (2) 1,2 is necessary. Assuming an ellipticity value to be measured due to vacuum magnetic birefringence of ψ = 2.5 × 10 −11 [formula (2)], it is not unreasonable to imagine that α (2) 1,2 ψ ≈ α (1) 1,2 /100. In all cases, though, there will not be a contribution from these terms if the rotation axis of each wave-plate coincides with the beam position, condition which can be obtained with a careful alignment of the optics. The reduction of these systematic effects is to be performed with the magnets turned off.
In the above formulas we have not considered the intrinsic birefringence of the mirrors [24]. In this scheme, by choosing appropriately φ 1 and φ 2 it should be possible to minimise the effect of this birefringence by independently aligning, on each mirror, the polarisation of the light to the birefringence axes of the mirrors [12].
Clearly the presence of the two half-wave-plates inside the cavity introduces some losses. Therefore there is an upper limit to the finesse one can obtain due to the absorption of the wave-plates. With a correct antireflective coating, waveplates can be obtained with a total absorption/reflection of 0.1 % each. Unwanted effects due to the reflected light from the wave-plates inside the cavity can be eliminated by misaligning the wave-plates very slightly so as to send the reflected light against the baffles which would be present inside the vacuum tube. Considering that the finesse F is where R + T + P = 1, and assuming that the transmission of the mirrors T are such that T P = 4×10 −3 (four passages through the waveplates), the absorption of the wave-plates limits the finesse to In this case the predicted QED ellipticity signal would be Assuming for HERA-X the best birefringence sensitivity as shown in Fig. 2, which is independent from the finesse, this would mean a value of S (100 Hz) n 2.5 × 10 −20 1/ √ Hz @ 100 Hz (with ν L = 25 Hz). Using F = 800 the sensitivity in ellipticity is The corresponding integration time to reach S/N = 1 would therefore be T = S ψ ψ (WavePlates) 2 10 5 s. ( Such a sensitivity remains to be demonstrated in the exceptional conditions of the proposed HERA-X experiment, but with such a low finesse, near shot-noise ellipticity sensitivities have been demonstrated. The minimum output power from the cavity to avoid being limited by shot-noise is ≈10 mW. With a finesse of F = 800 the circulating power inside the cavity is about 5 W distributed over a surface of about 0.1 cm 2 determined by the beam radius. This is well below the damage threshold of the wave-plates of 1 kW/cm 2 which therefore guarantees a correct and stable operation of the wave-plates. Furthermore very long Fabry-Perot cavities have been shown to be stable at frequencies of a few tens of hertz by LIGO and VIRGO reaching shot-noise performances [25]. The numbers seem to be within reach and we believe that this scheme could be a viable solution when using high field static magnetic fields generated by superconducting magnets.

Conclusion
In this note we have proposed a new scheme for a sensitive polarimeter dedicated to measuring vacuum magnetic bire-fringence based on a Fabry-Perot cavity which would allow the use of static magnetic fields generated by superconducting magnets. The modulation of the birefringence, necessary to reach high sensitivities, is performed by two co-rotating half-wave-plates inside the cavity, thus satisfying two conditions: rotating polarisation of the light inside the magnetic field; fixed polarisation direction on the Fabry-Perot mirrors. Furthermore the polarisation direction on the two mirrors can be controlled independently.
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecomm ons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. Funded by SCOAP 3 .