Results from a real-time dosimetry study during left atrial ablations performed with ultra-low dose radiation settings

Background Three-dimensional mapping systems and the use of ultra-low dose radiation protocols have supported minimization of radiation dose during left atrial ablation procedures. By using optimal shielding, scattered radiation reaching the operator can be further reduced. This prospective study was designed to determine the remaining operator radiation exposure during left atrial catheter ablations using real-time dosimetry. Methods Radiation dose was recorded using real-time digital dosimetry badges outside the lead apron during 201 consecutive left atrial fibrillation ablation procedures. All procedures were performed using the same X‑ray system (Siemens Healthineers Artis dBc; Siemens Healthcare AG, Erlangen, Germany) programmed with ultra-low dose radiation settings including a low frame rate (two frames per second), maximum copper filtration, and an optimized detector dose. To reduce scattered radiation to the operators, table-suspended lead curtains, ceiling-suspended leaded plastic shields, and radiation-absorbing shields on the patient were positioned in an overlapping configuration. Results The 201 procedures included 139 (69%) pulmonary vein isolations (PVI) (20 cryoballoon ablations, 119 radiofrequency ablations, with 35 cases receiving additional ablation of the cavotricuspid isthmus) and 62 (31%) PVI plus further left atrial substrate ablation. Mean radiation dose measured as dose area product for all procedures was 128.09 ± 187.87 cGy ∙ cm2 with a mean fluoroscopy duration of 9.4 ± 8.7 min. Real-time dosimetry showed very low average operator doses of 0.52 ± 0.10 µSv. A subanalysis of 51 (25%) procedures showed that the radiation burden for the operator was highest during pulmonary vein angiography. Conclusion The use of ultra-low dose radiation protocols in combination with optimized shielding results in extremely low scattered radiation reaching the operator.


Introduction and background
Despite the widespread use of nonfluoroscopic catheter visualization, fluoroscopy remains an essential part of catheter and wire localization. One of the drawbacks of the ever-increasing number of complex and prolonged ablation procedures is the higher radiation burden to the patient and the operator; it is well documented that even low-dose radiation can lead to potentially fatal radiation damage [13].
Increased awareness lead to the development of ultra-low dose programs with optimized image processing and system settings, resulting in average doses that equal as little as 1% of an ablation procedure performed 6 years ago [2,6].
Most dose reduction studies [2,3,5,9,19] used the dose area product (DAP) or some variation thereof as primary endpoint, whichmainlyserves tomeasure the patient's radiation exposure [13]. However, the DAP cannot precisely determine the radiationburdenofthe operator, since it depends on factors like shielding, angulation, and distance to the X-ray system.
Therefore, the purpose of this study was to determine the amount of radiation received by the operator as measured by dosimeters during atrial fibrillation ablation procedures.
The authors hypothesized that by using overlapping shielding and an up-todate ultra-low-dose protocol, scattered radiation reaching the operator would be vastly reduced.
They also sought to determine the moments during the ablation procedure in which radiation exposure to the operator is particularly high.

Study design
This study is an observational singlecenter study including 201 patients that underwent left atrial catheter ablation (radiofrequency or cryoballon) due to atrial fibrillation. All patients were consecutively enrolled at the Charité University Hospital, Campus Benjamin Franklin, Berlin, Germany. All patients were treated by the same two senior electrophysiologists using the same Siemens Artis dBc systems (Siemens Healthcare AG, Erlangen, Germany).
Demographic parameters, including body mass index (BMI), age, gender, and comorbidities were documented. Relevant procedural parameters for this study included total procedure time, fluoroscopy time, total radiation dose (measured as DAP), as well as acute procedural success and major periprocedural complications.

Standard radiation protection
Emphasis was put on general "as low as reasonably achievable" (ALARA) measurements, which the authors previously described [24]. These included avoiding magnification and unnecessary fluoroscopy/cine loops, maximal collimation, preferring anteroposterior (AP) and right anterior oblique (RAO) views over steep left anterior oblique (LAO) angulation, and minimizing the distance between patient and detector.
Standard shielding included the use of thyroid and body aprons with 0.5-Pb equivalent. As shown in . Fig. 1, table-suspended lead curtains and transparent, ceiling-suspended plastic shields were positioned in an overlapping configuration to ensure maximum protection against scattered radiation. Also, radiation-absorbing shields were placed on the patient.
Furthermore, an ultra-low-dose protocol was uniformly used for all ablation procedures. This consists of the use of a low frame rate (two frames per second, FPS), maximum copper filtration, low detector entrance dose (8 nGy/pulse for fluoroscopy), and short pulse width (4 ms). As previously described, the antiscatter grid was not used in any cases in order to further minimize radiation exposure [1]. For left atrial angiography, short (3-s) cine loops with 7.5 FPS were used during fast left ventricular pacing (200 bpm) via the ablation catheter. The pictures were recorded in biplane mode in RAO 30°or LAO 30°view. No preprocedural imaging was performed.

Measurement of radiation dose
Two parameters aiming to quantify radiation burden were obtained for each procedure. To evaluate dose to the patient, DAP (cGy ⋅ cm 2 ) was automatically measured by an ionization chamber. Measurement of DAP is routinely used to compare radiation burden of the patient and correlates well with skin dose [18].
Additionally, electronic dosimetry badges (RaySafe i3, Unfors, Hopkinton, MA, USA) were mounted outside the lead apron at the left side of the chest to measure the operator's dose (μSv) for each procedure. Both operators were blinded to the measured doses.
The RaySafe badges were collected and analyzed every week with DoseViewer software in order to prevent loss of sensitivity. For the last 51 (25%) procedures, radiation burden was analyzed for three different segments of the ablation procedure to assess the sequences with the highest radiation burden.
The procedures were divided into three parts as follows (see . Fig. 2

Results from a real-time dosimetry study during left atrial ablations performed with ultra-low dose radiation settings Abstract
Background. Three-dimensional mapping systems and the use of ultra-low dose radiation protocols have supported minimization of radiation dose during left atrial ablation procedures. By using optimal shielding, scattered radiation reaching the operator can be further reduced. This prospective study was designed to determine the remaining operator radiation exposure during left atrial catheter ablations using real-time dosimetry. Methods. Radiation dose was recorded using real-time digital dosimetry badges outside the lead apron during 201 consecutive left atrial fibrillation ablation procedures. All procedures were performed using the same X-ray system (Siemens Healthineers Artis dBc; Siemens Healthcare AG, Erlangen, Germany) programmed with ultra-low dose radiation settings including a low frame rate (two frames per second), maximum copper filtration, and an optimized detector dose. To reduce scattered radiation to the operators,
Mean radiation dose measured as DAP for all procedures was 128.09 + 187.87 cGy ⋅ cm 2 with a mean fluoroscopy duration of 9.4 + 8.7 min. The 10 highest recorded doses predominantly (90%) occurred during ablation procedures of patients with a BMI >30. We observed a significant positive relationship between the patients BMI and operator dose/DAP (R = 0.348, p < 0.001 and R = 0.506, p < 0.001).
As mentioned earlier, for the last 51 procedures, a subanalysis was performed to determine which part of the procedure showed the highest radiation exposure to the operator. . Figure 3 shows that most radiation (55%) towards the operator occurs during left atrial angiography.

Main findings
This single-center study demonstrated that, with the use of an ultra-low-dose program and optimized shielding, operator doses can be kept at very low levels during catheter ablation of atrial fibrillation. The operator's exposure was highest during left atrial angiography, making shielding and further dose reduction particularly valuable assets at this moment in the procedure. To our knowledge, this is the first analysis of the operator's exposure with the use of ultra-low-dose programs for catheter ablation of atrial fibrillation.

Previously published studies
Scant data exist on personal dosimetry regarding atrial fibrillation ablation. A 2019 study reported mean values of 3 μSv with a mean DAP of 429.6 cGy ⋅ cm 2 [10]. Those results were achieved by using a frame rate of 4 FPS. By using a lower frame rate of 2 FPS, the authors were able to achieve lower average doses while maintaining a low rate of complications with a reasonable procedural time. Reducing the frame rate is arguably one of the simplest and most effective ways to lower radiation burden, given that the relationship between dose and frame rate is linear. Despite this fact, only 50% of European ablation centers use a frame rate of less than 6 FPS for electrophysiology (EP) procedures [9]. The reason for this might be that a considerable number of EP labs use settings also used in coronary interventions. However, the demand for image quality in EP studies is usually far more modest, even in complex procedures. Therefore, the authors advise contacting the X-ray system's manufacturer to establish a low-dose radiation setup tailored to the operators' requirements.
Since scattered radiation emitted from the patient's body is the main source of the operator's radiation burden, methods to absorb scattered radiation have been tested: sterile radiation-absorbing drapes placed around the puncture site can reduce the operator's radiation exposure by more than half [12]. Mobile fullbody protection systems can provide protection against scattered radiation, except for the arms, and can reduce radiation to the operator close to background radiation values [8]. High costs, limited availability, and structural preconditions have hindered their widespread use.

Radiation put into context
Despite various efforts to minimize operator radiation, interventional cardiologists in most centers are still subjected to a relevant amount of radiation throughout their career, with an annual estimated dose of 5 mSv [13]. Reducing radiation burden minimizes the detrimental effects of ionizing radiation, some of which include cancer, inherited diseases, and the formation of cataracts [21]. . Figure 4 shows previously published radiation exposure for different diagnostic and therapeutic interventions, as well annual background radiation and radiation during an intercontinental flight.

Minimizing radiation during left atrial angiography
Knowledge of pulmonary vein anatomy and its variations (common ostia, accessory pulmonary vein, pulmonary vein fusion) is a key factor for a safe and efficient ablation procedure. This information is most frequently obtained by intraprocedural angiography. Preprocedural computed tomograhy left atrial angiography might be helpful to avoid intraprocedural angiography and thereby reduce operator X-ray exposure. However, this method is associated with a higher radiation burden to the patient [4,17]. The authors suggest implementing the following measures to ensure low radiation burden: 4 Using short cine loops with low frame rates (for example, 3 s and 7.5 FPS) with fast left ventricular pacing to lower use of contrast agent [23] 4 Strict collimation to the left atrium 4 Avoiding additional angiography of single pulmonary veins The feasibility and safety of numerous fluoroless approaches for catheterbased pulmonary vein ablation and LA visualization have been demonstrated in smaller studies [7]. These include magnetic resonance imagingguided ablation procedures [26], as well as intracardiac echocardiography [11] and transesophageal echocardiography [20] without any subsequent LA an-giography. However, limited experience and availability, high costs, and possible complications have hindered more widespread use.

Limitations
Thisstudyrepresentsa single-centeranalysis with two senior electrophysiologists performing all procedures. Therefore, the observed results may not be valid for other ablation centers with different dose settings. Although standardized shielding was aimed for, different placement of the shields may have altered the results.

Conclusion
This study demonstrates that by using an ultra-low-dose setting and optimal shielding, radiation reaching the operator can be largely reduced. Therefore, table-suspended lead curtains and ceiling-suspended shields should be positioned in an overlapping configuration to ensure maximum protection against scattered radiation, especially during left atrial angiography.
or other third party material in this article are included in the article's Creative Commons licence, unless indicated otherwise in a credit line to the material. If material is not included in the article's Creative Commons licence and your intended use is not permitted by statutory regulation or exceeds the permitted use, you will need to obtain permission directly from the copyright holder. To view a copy of this licence, visit http://creativecommons.org/licenses/by/4.0/.