Design of Noninvasive Hemodynamic Monitoring Equipment Using Impedance Cardiography
- 608 Downloads
Hemodynamic parameters play a very important role in supporting physicians to diagnose and treat cardiovascular diseases. This paper presents the results of the study on designing of noninvasive hemodynamic monitoring equipment using impedance cardiography (ICG). This is a measurement method that shows many advantages and is interested in the present. The equipment has ability of measurement and monitoring hemodynamic parameters include: heart rate HR; thoracic fluid content TFC; left ventricular ejection time LVET; stroke volume SV; cardiac output CO and ICG waveform. Results of measurement between the research equipment BK-NICO and the standard equipment of NICCOMO for 104 patients show that the hemodynamic parameters measured by the two equipments are quite small difference. Specifically, the average deviation of the parameters obtained, HR is 0.16 bpm, TFC is 0.73 1/kΩ, LVET is 5.95 ms, SV is 0.17 ml, CO is 0.01 lpm. With the results of the research, the device is fully capable of deploying the application into practice in the hospitals to replace the current invasive measurement method.
KeywordsNoninvasive hemodynamic monitor Impedance cardiography Stroke volume Cardiac output Thoracic fluid content Left ventricular ejection time
We would like to thank the Ministry of Science and Technology (MOST), Hanoi University of Science and Technology (HUST), Viet Duc hospital, Hanoi Heart hospital for their support and encouragement. The system was designed and developed by the faculties and students at the Biomedical Electronics Center (BMEC).
Conflict of Interest The authors declare that they have no conflict of interest.
- 3.Webster, J.G.: Medical Instrumentation: Application and Design: Measurement of Flow and Volume of Blood. Wiley, New York (2010)Google Scholar
- 4.Ipate, M.C., Dobre, A.A., Morega, A.M.: The stroke volume and the cardiac output by the impedance cardiography. U.P.B. Sci. Bull. 74(3) (2012) Google Scholar
- 6.Lavdaniti, M.: Invasive and non-invasive methods for cardic output measurement. Int. J. Caring Sci. 1(3), 112–117 (2008)Google Scholar
- 7.Horster, S., Stemmler, H.J., Strecker, N., Brettner, F., Hausmann, A., Cnossen, J., Parhofer, K.G., Nickel, T., Geiger, S.: Cardiac output measurements in septic patients: comparing the accuracy of USCOM and PiCCO. Crit. Care Res. Pract. (2012)Google Scholar
- 10.Carvalho, P., Paiva, R.P., Henriques, J., Antunes, M., Quintal, I., Muehlsteff, J.: Robust characteristic points for ICG: definition and comparative analysis. In: Proceedings of the International Conference on Bio-inspired Systems and Signal Processing, Rome, Italy (2011)Google Scholar