Continuous-Wave Photoacoustic-Based Sensor for the Detection of Aqueous Glucose: Towards Non-invasive and Continuous Glycemia Sensing
Measurement of blood glucose levels (BGLs) is a basic procedure that diabetic patients need to perform several times a day. The conventional standard protocol for on-site measurement, despite several advantages such as portability, low cost, fast response time, and ease of operation, is based on the finger-prick technique to extract blood samples. This process is invasive and cannot provide continuous monitoring.
Towards the achievement of non-invasive and continuous BGL monitoring, we have developed two measurement methods based on the continuous-wave photoacoustic (CW-PA) protocol and we performed preliminary in vitro tests with aqueous solutions. The first method relies on the measurement of the frequency shift induced by the change in the composition of the propagation medium. This method is equivalent to an acoustic velocity measurement and provides high sensitivity but no selectivity to glucose compound. The second approach utilizes simultaneous optical excitation at two wavelengths for compound-selective measurements. After correcting the frequency shift mentioned previously, this protocol allows measurements equivalent to a differential absorption coefficient one at the two wavelengths used. It then combines the advantages of absorption spectroscopy without the limitation from scattering due to the use of acoustic detection. Furthermore, the combination of the two methods can be generalized to systems involving more than one changing parameter by using not only two optical wavelengths for the excitation sequence but also several pairs of wavelength sequentially.
These methods then represent an important step forward the non-invasive, selective, and continuous measurements of glucose compound concentrations from a complex mixture, typically blood.
Keywordsphotoacoustic method continuous blood glucose level
Unable to display preview. Download preview PDF.
- 2.World Health Organization (2011), http://www.who.int/mediacentre/facsheets/fs312/en/
- 4.World Health Organization, International Diabetes Federation (2004), http://www.who.int/diabetes/publications/diabetes_booklet/en/index.html
- 5.The Expert Committee on the Diagnosis and Classification if Diabetes Mellitus. Follow-up report on the diagnosis of diabetes mellitus, Diabetes Care 26(11), 3160–3167 (2003)Google Scholar
- 6.American Diabetes Association. Standards of medical care in diabetes, Diabetes Care 28 (supp.1), s4–s36 (2005)Google Scholar
- 7.Consensus Development Panel. Consensus statement on self-monitoring of blood glucose. Diabetes Care 10, 95–99 (1987)Google Scholar
- 8.Kirk, J.K., Stegner, J.: Self-monitoring of blood glucose: practical aspects. J. Diabetes Sci. Technol. 4(2), 435–439 (2010)Google Scholar
- 9.Rubin, A.L.: Diabetes for dummies, 3rd edn. Wiley Publishing, Inc. (2008)Google Scholar
- 12.Ginsberg, B.H.: An overview of minimally invasive technologies. Clin. Chem. 38(9), 1596–1600 (1992)Google Scholar
- 17.Steil, G.M., Rebrin, K., Goode Jr., P.V., Mastrototaro, J.J., Purvis, R.E., Van Antwerp, W.P., Shin, J.J., Talbot, C.D.: Closed loop system for controlling insulin infusion. US Patent US006558351B1 (2003)Google Scholar
- 18.Chee, F., Fernando, T.: Closed-loop control of blood glucose. LNCIS, vol. 368. Springer Ed. (2007) ISBN 3-540-74030-9Google Scholar
- 19.Heo, Y.J., Shibata, H., Okitsu, T., Kawanishi, T., Takeuchi, S.: Long-term in vivo glucose monitoring using fluorescent hydrogel fibers. PNAS 1104954108 (2011)Google Scholar
- 20.Voskerician, G., Anderson, J.: Foreign body reaction. Wiley Encyclopedia of Biomedical Engineering (2006) ISBN 9780471740360Google Scholar
- 21.Ma, K., Yuen, J.M., Shah, N.C., Walsh Jr., J.T., Glucksberg, M.R., Van Duyne, R.P.: In vivo, transcutaneous glucose sensing using surface-enhanced spatially offset Raman spectroscopy: multiple rats, improved hypoglycemic accuracy, low in-cident power, and continuous monitoring for greater than 17 days. Anal. Chem. 83(23), 9146–9152 (2011)CrossRefGoogle Scholar
- 27.MacKenzie, H.A., Ashton, H.S., Shen, Y.C., Lindberg, J., Rae, P., Quan, K.M., Spiers, S.: Blood glucose measurements by photoacoustics. Biomedical Opt. Biomedical Opt. Spectroscopy and Diagnostics/Therapeutic Laser App. 22, 156–159 (1998)Google Scholar
- 28.Zhao, Z.: Pulse photoacoustic techniques and glucose determination in human blood and tissue. Oulu University Press, Oulu (2002) ISBN 951-42-6689-7Google Scholar
- 31.Spanner, G., Niessner, R.: New concept for the non-invasive determination of physiological glucose concentrations using modulated laser diodes. Fresenius J. Anal. Chem. 354, 306–310 (1996)Google Scholar
- 37.Camou, S., Ueno, Y., Tamechika, E.: Towards non-invasive and continuous blood sugar sensor: detection of aqueous glucose based on CW-photoacoustic proto-col. In: Proceeding of IEEE Sensors 2010, Hawaii, USA (2010)Google Scholar
- 41.Zips, A., Faust, U.: Determination of biomass by ultrasonic measurements. Appl. Environ. Microbiol. 55(7), 1801–1807 (1989)Google Scholar
- 42.Hall, J.W., Pollard, A.: Near-infrared spectrophotometry: a new dimension in clinical chemistry. Clin. Chem. 38(9), 1623–1631 (1992)Google Scholar
- 46.Spanner, G., Niessner, R.: Noninvasive determination of blood constituents using an array of modulated laser diodes and a photoacoustic sensor head. Fresenius J. Anal. Chem. 355, 327–328 (1996)Google Scholar
- 48.Allen, K.: Principles and limitations of pulse oximetry in patient monitoring. Nurs. Times 12-18, 100(41), 34–37 (2004)Google Scholar
- 49.Mendelson, Y.: Pulse oximetry: theory and applications for non-invasive monitoring. Clin. Chem. 38(9), 1601–1607 (1992)Google Scholar