Low-Cost Blood Pressure Monitor Device for Developing Countries
- 1.8k Downloads
Taking the Blood Pressure (BP) with a traditional sphygmomanometer requires a trained user. In developed countries, patients who need to monitor their BP at home usually acquire an electronic BP device with an automatic inflate/deflate cycle that determines the BP through the oscillometric method. For patients in resource constrained regions automated BP measurement devices are scarce because supply channels are limited and relative costs are high. Consequently, routine screening for and monitoring of hypertension is not common place. In this project we aim to offer an alternative strategy to measure BP and Heart Rate (HR) in developing countries. Given that mobile phones are becoming increasingly available and affordable in these regions, we designed a system that comprises low-cost peripherals with minimal electronics, offloading the main processing to the phone. A simple pressure sensor passes information to the mobile phone and the oscillometric method is used to determine BP and HR. Data are then transmitted to a central medical record to reduce errors in time stamping and information loss.
KeywordsBlood pressure developing countries electronic medical records hypertension mHealth resource-constrained healthcare low-cost devices
Unable to display preview. Download preview PDF.
- 2.Goldman, L., Ausiello, D.: Cecil Medicine. Saunders Elsevier, PA (2008)Google Scholar
- 8.Hug, C.W., Clifford, G.D.: An analysis of the errors in recorded heart rate and blood pressure in the ICU using a complex set of signal quality metrics. In: CIC, pp. 641–644 (2007)Google Scholar
- 13.Freescale product code: MPXx5050Google Scholar
- 14.Microchip product code: en533924Google Scholar
- 15.Ball-Llovera, A., et al.: An experience in implementing the oscillometric algorithm for the noninvasive determination of human blood pressure. In: Proceedings of the 25th Annual International Conference of the IEEE, vol. 4, pp. 3173–3175 (2003)Google Scholar
- 21.Dalvik Virtual Machine, http://code.google.com/p/dalvik/
- 23.EWH-BP-Project, http://code.google.com/p/ewh-bp-project/
- 24.Sana Mobile, http://www.sanamobile.org/
- 25.Goonasekera, C.D., Dillon, M.J.: Random zero sphygmomanometer versus automatic oscillometric blood pressure monitor; is either the instrument of choice? J. Hum. Hypertens 9(11), 885–889 (1995)Google Scholar
- 29.Canalys, Android smart phone shipments grow 886% year-on-year in Q2 (2010), http://www.canalys.com/newsroom/android-smart-phone-shipments-grow-886-year-year-q2-2010