Dynamic Distortion Correction for Endoscopy Systems with Exchangeable Optics
Endoscopic images are strongly affected by lens distortion caused by the use of wide angle lenses. In case of endoscopy systems with exchangeable optics, e.g. in bladder endoscopy or sinus endoscopy, the camera sensor and the optics do not form a rigid system but they can be shifted and rotated with respect to each other during an examination. This flexibility has a major impact on the location of the distortion centre as it is moved along with the optics. In this paper, we describe an algorithm for the dynamic correction of lens distortion in cystoscopy which is based on a one time calibration. For the compensation, we combine a conventional static method for distortion correction with an algorithm to detect the position and the orientation of the elliptic field of view. This enables us to estimate the position of the distortion centre according to the relative movement of camera and optics. Therewith, a distortion correction for arbitrary rotation angles and shifts becomes possible without performing static calibrations for every possible combination of shifts and angles beforehand.
Unable to display preview. Download preview PDF.
- 1.Konen W, Breiderhoff B, Scholz M. Real-time image mosaic for endoscopic video sequences. Proc BVM. 2007; p. 298–302.Google Scholar
- 2.Behrens A. Creating panoramic images for bladder fluorescence endoscopy. Acta Polytech: J Adv Eng. 2008;3:50–54.Google Scholar
- 3.Stehle T, Truhn D, Aach T, et al. Camera calibration for fish-eye lenses in endoscopy with an application to 3D reconstruction. Procs IEEE ISBI. 2007 April; p. 1176–1179.Google Scholar
- 4.Kannala J, Brandt SS. A generic camera model and calibration method for conventional, wide-angle, and fish-eye lenses. IEEE Trans PAMI. 2006;28(8):1335–1340.Google Scholar
- 5.Hartley R, Kang SB. Parameter-free radial distortion correction with center of distortion estimation. IEEE Trans PAMI. 2007;29(8): 1309–1321.Google Scholar
- 6.Mühlich M, Aach T. High accuracy feature detection for camera calibration: A multi-steerable approach. LNCS. 2007;(4713):284–293.Google Scholar
- 7.Stäche NC, Zimmer H, Gedicke J, et al. Robust high-speed melt pool measurements for laser welding with sputter detection capability. LNCS. 2007;4713:476–485.Google Scholar