# Accurate Spectral Testing with Arbitrary Non-coherency in Sampling and Simultaneous Drifts in Amplitude and Frequency

## Abstract

Accurate spectral testing plays a crucial role in modern high-precision ADCs’ evaluation process. One of the challenges is to be able to test the continually higher-resolution ADCs accurately and cost-effectively. Due to its stringent test requirement, the standard test method for ADCs can be difficult to implement with low cost. This chapter proposes an algorithm that relaxes the requirements of precise control over source amplitude and frequency and of the need to achieve coherent sampling. The algorithm divides the output data into segments and estimates the drifting fundamental via Newton iteration. By removing the estimated drift fundamental and replacing with a coherent, non-drift, fundamental in time domain, accurate spectral results can be achieved. Various simulation results have validated the accuracy of the proposed algorithm. The proposed algorithm is capable of tolerating various test condition variations such as any level of non-coherency, various input frequency ranges, and different numbers of segmentations. In addition, several measurement results from different ADCs have verified the accuracy of the proposed algorithm, which is able to accurately obtain spectral performance of an 18-bit high-resolution ADC. Such an algorithm relaxes the standard test requirement such as precise control over source frequency and amplitude, which dramatically reduces the test setup complexity and cost.

## References

- 1.Y. Zhuang, L. Xu, D. Chen, Accurate spectral testing with arbitrary noncoherency in sampling and simultaneous drifts in amplitude and frequency. IEEE Trans. Instrum. Meas.
**55**(5), 1002–1012 (2017)CrossRefGoogle Scholar - 2.Y. Zhuang, D. Chen, ADC spectral testing with signal amplitude drift and simultaneous non-coherent sampling. J. Electron. Test. Theory Appl.
**33**(3), 305–313 (2017)CrossRefGoogle Scholar - 3.R.J. van de Plassche,
*CMOS Integrated Analog-to-Digital and Digital-to-Analog Converters*(Kluwer Academic Publishers, Boston, 2003)CrossRefzbMATHGoogle Scholar - 4.M.J.M. Pelgrom,
*Analog-to-Digital Conversion*, 2nd edn. (Springer, New York, 2012)Google Scholar - 5.A. Oppenheim,
*Discrete-Time Signal Processing*(Prentice-Hall, Englewood Cliffs, 1999)Google Scholar - 6.M. Burns, G.W. Roberts,
*An Introduction to Mixed-Signal IC Test and Measurement*(Oxford University Press, New York, 2000)Google Scholar - 7.IEEE Standard for Terminology and Test Methods for Analg-to-Digital Converters,
*IEEE Std.1241*, (2010)Google Scholar - 8.IEEE Standard for Terminology and Test Methods of Digital-to-Analog Converter Devices,
*IEEE Std.1658*, (2011)Google Scholar - 9.IEEE Standard for Digitizing Waveform Recorders,
*IEEE Std.1057*, (2007)Google Scholar - 10.F.J. Harris, On the use of windows for harmonic analysis with the discrete Fourier transform. Proc. IEEE
**66**(1), 51–83 (1978)CrossRefGoogle Scholar - 11.P. Carbone, E. Nunzi, D. Petri, Windows for ADC dynamic testing via frequency-domain analysis. IEEE Trans. Instrum. Meas.
**50**(6), 1571–1576 (2001)CrossRefGoogle Scholar - 12.S. Raze, D. Dallet, P. Marchegay, Non coherent spectral analysis of ADC using FFT windows: An alternative approach, in
*Proceedings of IEEE Workshop Intelligent Data Acquisition Advanced Computer System*, (2005), pp. 474–478Google Scholar - 13.D. Belega, M. Ciugudean, D. Stoiciu, Choice of the cosine-class windows for ADC dynamic testing by spectral analysis. Measurement
**40**(4), 361–371 (2007)CrossRefGoogle Scholar - 14.O.M. Solomon, The use of DFT windows in signal-to-noise ratio and harmonic distortion computations. IEEE Trans. Instrum. Meas.
**43**(2), 194–199 (1994)CrossRefGoogle Scholar - 15.T.Z. Bilau, T. Megyeri, A. Sarhegyi, J. Márkus, I. Kollár, Four-parameter fitting of sine wave testing result: Iteration and convergence. Comput. Stand. Interfaces
**26**(1), 51–56 (2004.) 731CrossRefGoogle Scholar - 16.R. Pintelon, J. Schoukens, An improved sine-wave fitting procedure for characterizing data acquisition channels. IEEE Trans. Instrum. Meas.
**45**(2), 588–593 (1996)CrossRefGoogle Scholar - 17.K.F. Chen, Estimating parameters of a sine wave by separable nonlinear least squares fitting. IEEE Trans. Instrum. Meas.
**59**(12), 3214–3217 (2010)CrossRefGoogle Scholar - 18.G. Simon, R. Pintelon, L. Sujbert, J. Schoukens, An efficient nonlinear least square multisine fitting algorithm. IEEE Trans. Instrum. Meas.
**51**(4), 750–755 (2002)CrossRefGoogle Scholar - 19.D. Belega, D. Dallet, D. Petri, Estimation of the effective number of bits of ADCs using the interpolated DFT method, in
*Proceedings of IEEE Instrumentation and Measurement Technology Conference*, (2010), pp. 30–35Google Scholar - 20.S. Sudani, M. Wu, D. Chen, A novel robust and accurate spectral testing method for non-coherent sampling, in
*Proceedings of International Test Conference*, (2011), pp. 1–10Google Scholar - 21.S.K. Sudani, D. Chen, FIRE: A fundamental identification and replacement method for accurate spectral test without requiring coherency. IEEE Trans. Instrum. Meas.
**62**(11), 3015–3025 (2013)CrossRefGoogle Scholar - 22.Y. Zhuang, D. Chen, Accurate spectral testing with non-coherent sampling for large distortion to noise ratios, in
*Proceedings of IEEE 34th VLSI Test Symposium (VTS)*, (2016), pp. 1–6Google Scholar - 23.Y. Zhuang, D. Chen, New strategies in removing non-coherency from signals with large distortion to noise ratios, in
*Proceedings of IEEE International Symposium Circuits Systems (ISCAS)*, (2016), p. 2901Google Scholar - 24.Y. Zhuang, D. Chen, New strategies in removing noncoherency from signals with large distortion-to-noise ratios. IEEE Trans. Circuits Syst. II, Express Briefs
**63**(12), 1136–1140 (2016)CrossRefGoogle Scholar - 25.M.H. Ackroyd, Short-time spectra and time-frequency energy distributions. J. Acoust. Soc. Am.
**50**, 1229–1231 (2005)CrossRefGoogle Scholar - 26.J.K. Hammond, P.R. White, The analysis of non-stationary signals using time-frequency methods. J. Sound Vib.
**190**(3), 419–447 (1996)CrossRefGoogle Scholar - 27.M. Landisman, A. Dziewonski, Y. Satô, Recent improvements in the analysis of surface wave observations. Geophys. J. Int.
**17**(4), 369–403 (1969)CrossRefGoogle Scholar - 28.L. Griffiths, Rapid measurement of digital instantaneous frequency. IEEE Trans. Acoust. Speech Signal Process.
**23**(2), 207–222 (1975)CrossRefGoogle Scholar - 29.L. Cohen, Time-frequency distributions—A review. Proc. IEEE
**77**(7), 941–981 (1989)CrossRefGoogle Scholar - 30.T. Dabóczi, ADC testing using a resonator-based observer: Processing very long time records and/or testing systems with limited stability. IEEE Trans. Instrum. Meas.
**62**(5), 1166–1173 (2013)CrossRefGoogle Scholar - 31.T. Dabóczi, Increasing the robustness of the resonator based ADC testing, in
*Proceedings of IEEE International Instrumentation and Measurement Technology Conference (IMTC)*, (2013), pp. 353–358Google Scholar - 32.L. Xu, D. Chen, Accurate spectral testing of analog-to-digital converters with frequency drift using phase correction and averaging, in
*Proceedings of IEEE International Symposium Circuits System*, (2015), pp. 2265–2268Google Scholar - 33.G.C. Cecchi, Error analysis of the parameters of a least-squares determined curve when both variables have uncertainties. Meas. Sci. Technol.
**2**(12), 1127–1128 (1991)CrossRefGoogle Scholar - 34.A. Bergamin, G. Cavagnero, G. Mana, Accuracy assessment of a least-squares estimator for scanning X-ray interferometry. Meas. Sci. Technol.
**2**(8), 725–734 (1991)CrossRefGoogle Scholar