Is It Smart to Use Smartphones to Measure Illuminance for Occupational Health and Safety Purposes?
This study reports on the accuracy of smartphone-based illuminance measurement applications and whether they may be employed for occupational lighting measurements. A sample of nine mobile phones on three platforms (Android, iOS and Windows) was assembled and 14 apps were selected for testing. Testing conditions comprised four pre-specified illuminance levels (300 lx, 500 lx, 750 lx and 1000 lx) accomplished with three light sources presenting different Correlated Color Temperatures (2700 K; 4000 K; 6500 K). The results reveal extremely variable and sometimes large deviations from the reference levels and suggest smartphones are not appropriate for use in occupational lighting assessments.
KeywordsLight Measurement Applications Mobile phones Accuracy
We would like to thank those who volunteered their smartphones for testing.
- 1.CEN: EN 12464-1:2011 Light and lighting—lighting of work places part 1: indoor work places (2011)Google Scholar
- 2.ISO: ISO 8995-1:2002(en) Lighting of work places—part 1: indoor (2002)Google Scholar
- 4.ISO: ISO/CIE 19476:2014 Characterization of the performance of illuminance meters and luminance meters (2014)Google Scholar
- 5.Statista (The Statistics Portal): Smartphones industry: statistics and Facts. https://www.statista.com/topics/840/smartphones/
- 6.Grupo Marktest: Smartphone continua a crescer em Portugal. http://www.marktest.com/wap/a/n/id~2046.aspx
- 8.Johnson, N., Saxena, P., Williams, D., Bangole, O.C., Kamrul, H.M., Ahamed, S.I., Smith, R.O., Tomashek, D.: Smartphone-based light and sound intensity calculation application for accessibility measurement. In: RESNA Annual Conference, Denver (2015)Google Scholar
- 10.Negar, N., Williams, D., Schwartz, J., Ahamed, S.I., Smith, R.O.: Smartphone-based light intensity calculation application for accessibility measurement. In: RESNA Annual Conference, 2014. Indianapolis (2014)Google Scholar
- 14.Kardous, C.A., Shaw, P.B.: Do sound meter apps measure noise levels accurately? Sound Vib. 49, 10–13 (2015)Google Scholar
- 18.Wolfgang, R., Corleto, L.Di, Burgess-Limerick, R.: Can an ipod touch be used to assess whole-body vibration associated with mining equipment? Ann. Occup. Hyg. 58, 1200–1204 (2014)Google Scholar
- 21.Goldschmidt, J.: Luxmeter App versus measuring device: are smartphones suitable for measuring illuminance. https://www.dial.de/en/blog/article/luxmeter-app-versus-measuring-deviceare-smartphones-suitable-for-measuring-illuminance/
- 22.NetMarketShare (Market Share Statistics for Internet Technologies): Mobile/Tablet Operating System Market Share. https://www.netmarketshare.com/operating-system-market-share.aspx?qprid=8&qpcustomd=1
- 23.Simões, P.: Apple voltou ao topo das vendas de smartphones no final de 2016. https://pplware.sapo.pt/apple/apple-topo-vendas-smartphones-final-2016/