The Design and Construction of a New Quartz Fiber Balance
The problems concerned in the commercial production of a quartz fiber ultramicrobalance are considered in relation to a number of existing designs. It is concluded that none of these lend themselves to this type of production, and a design finally adopted is described.
Performance tests show that the balance enables weighings to be carried out with a standard deviation of 0.08 µg, up to a maximum load of 250 mg in each pan. The factors influencing the accuracy attainable are discussed.
KeywordsQuartz Dust Torque Refraction Suffix
Unable to display preview. Download preview PDF.
- 1.B. B. Cunningham, Nucleonics 5, 62 (1949).Google Scholar
- 3.H. Asbury, R. Belcher, and T. S. West, Mikrochimica Acta, 598 (1956).Google Scholar
- 4.A. G. Day, Technical Report L/T256, Electrical Research Association, Greenford, Middx., England, 1951.Google Scholar
- 5.R. G. Olt, H. R. Dufour, M. I. Gray, and J. H. Wright, “A Remote Controlled Quartz Fiber Microbalance— Fabrication of Quartz Components.n A.E.C. Research and Development Report MLM-1023, 1954.Google Scholar
- 6.T.T. O’Donnell, “Drawing and Working Quartz Fibers,” National Academy of Sciences—National Research Council, Washington, D. C., 1958.Google Scholar
- 7.C. L. Wilson and H. M. El-Badry, Symposium on Microbalances, R. Inst. of Chemistry, London, 1950.Google Scholar