Steady State Technique for Low Temperature Heat Capacity of Small Samples
Reported here is an elegant method based on steady state or ac calorimetry technique, using laser beam as heat source, to measure absolute heat capacity of tiny samples of mass l-100mgm. The results on high purity copper, gold, nickel and isotopes of molybdenum are given to show that the absolute accuracy of this method is ~ ± 2%, at least comparable, if not better than conventional heat pulse techniques, which require bulky samples ~ 100 times larger. The possible variations of this method will also be discussed.
KeywordsHeat Capacity Heat Capacity Measurement Copper Block Focal Length Lens Mechanical Stage
Unable to display preview. Download preview PDF.
- 1.P. H. Keesom and N. Pearlman, Methods in Experimental Physics, Ed. K. Lark’-Horovitz and V. A. Johnson, Academic Press, N. Y. (1959), Vol. 6, Part A, Sec. 188.8.131.52.Google Scholar
- 2.C. A. Luengo, Ph.D. Thesis, Universidad Nacional de Cuyo, San Carlos de Bariloche Argentina (1972).Google Scholar
- 3.Y. A. Kraftmakher, Zh. Prikl. Mekhan. i. Tekhn. Fiz. 5, 176 (1962).Google Scholar
- 5.R. D. Hempstead, Ph.D. Thesis, University of Illinois, Urbana (1970).Google Scholar
- 7.D. S. Simons, Ph.D. Thesis, University of Illinois, Urbana (1973).Google Scholar
- 8.L. L. Sparks and W. J. Hall, Cryogenic Temperature Tables III, NBS Report 9721 (1969), pp. 16.Google Scholar
- 9.G. T. Furukawa, W. G. Saba and M. L. Reilly, National Standard Ref. Data Series, NBS18, April 1968, pp. 20.Google Scholar
- 11.F. Heiniger, E. Bucher and J. Müller, Phys. Kondens. Materie. 5, 243 (1966).Google Scholar
- 12.R. Viswanathan, H, L. Luo and J. J. Engelhardt, Proc. 13th International Conf. on Low Temp. Phys. (LT-13), 1972 (in press).Google Scholar
- 15.R. Viswanathan and C. T. Wu, (to be published).Google Scholar