Scintillation and Semiconductor Detectors
As stated in Chapter 7, the detection efficiency of γ- and x-rays in gas detectors is very low, because these penetrating radiations travel through the low-density gas with little interaction. To improve counting efficiency for these radiations, solid and liquid scintillation detectors with high density are used. These detectors have the unique property of emitting scintillations or flashes of light after absorbing γ- or x-radiations. The light photons produced are converted to an electrical pulse by means of a photomultiplier (PM) tube (described later). The pulse is then amplified by a linear amplifier, sorted by a pulse-height analyzer (PHA) and then registered as a count. Different solid or liquid detectors are used for different types of radiation. For example, sodium iodide detectors containing a trace of thallium (NaI[T1]) are used for γ- and x-ray detection, whereas organic detectors such as anthracene and plastic fluors are used for β-particle detection.
KeywordsSemiconductor Detector Light Photon Scintillation Camera Organic Scintillator Bismuth Germanate
Unable to display preview. Download preview PDF.
- Bushberg JT, Seibert JA, Leidholdt EM Jr, Boone JM. The Essential Physics of Medical Imaging. Baltimore: Williams & Wilkins; 1994.Google Scholar
- Hendee WR, Ritenour ER. Medical Imaging Physics. St. Louis: Mosby; 1992.Google Scholar
- Hine GJ. Sodium iodide scintillators. In: Hine GJ, eds. Instrumentation in Nuclear Medicine. New York: Academic Press; 1967;1:95–117.Google Scholar
- Peng CT, Horrocks DL, Alpen EL, eds. Liquid Scintillation Counting. New York: Academic Press; 1980; I, II.Google Scholar
- Rollo FD, ed. Nuclear Medicine Physics, Instrumentation and Agents. St. Louis: Mosby; 1977.Google Scholar
- Sorensen JA, Phelps ME. Physics in Nuclear Medicine. 2nd ed. New York: Grune & Stratton; 1987, Chapters 12 and 13.Google Scholar