Optical Particle Sizing pp 431-441 | Cite as

# Application of Top-Hat Laser Beam to Particle Sizing in LDV System

## Abstract

Efforts have been made to obtain information on mixing to relate with ignition in the peripheral region of spray jets with clouds dilute enough to be measured by an laser Doppler velocimeter system. Investigators have attempted simultaneous size measurements within the same optical system, or combined with another light source, by Mie-scatter intensity, Doppler visibility or Doppler phase difference [1–11]. The authors have investigated heat transfer in two-phase mist flows and analyzed flow behaviour and the variation of droplet size distribution in boundary layers with a two-beam LDV system [12,13]. The LDV system with particle sizing proposed previously [14,15] has worked enough in detecting the scatter intensity with discrimination of the particle paths in the measuring volume. However, the electronic circuits had to be more sophisticated, and the ambiguity of size estimation can not be avoided in highly turbulent flows.

## Keywords

Spatial Filter Droplet Size Distribution Particle Path Light Scatter Intensity Green Beam## Preview

Unable to display preview. Download preview PDF.

## References

- [1]Holve, D. and Self, S.A.,1979, Optical particle sizingfor in situ measurements part 1, Appl. Optics, 18-10, pp. 1632-1645.Google Scholar
- [2]Kline, R., Gouesbet, G. and Ruck, B., 1982, Simultaneous, local measurements of particle velocity, particle size and particle concentration, Sensor 82 Transducer Technology and Temperature Measurement vol. 3, pp. 15-41.Google Scholar
- [3]Hadded, O., Bates, C.J., Yeoman, M.L. and White, H.J., 1984, Evaluation of drop size and two-component velocitymeasurement technique in annular two phase-flow, 2nd Int. Symp. on Appl. of LDA to Fluid Mech., pp. 16.2, 1984.Google Scholar
- [4]Yule, A.J., Chiger, N.A., Atakan, S. and Ungut, A., J. 1977, Particle size and velocity measurement by laser anemometry, Energy, 1, 4, pp. 220–228.CrossRefGoogle Scholar
- [5]Gouesbet, G and Ledoux, M., 1984, Supermicronic and submicronic optical sizing including a discussion of densely laden flow, Optical Engineering vol.23-5, pp. 631.Google Scholar
- [6]Ruck, B. and Pavlowski, B., 1984, Combined measurement of particle size amd velocity distribution in ducts, 2nd Int. Symp. on Appl. of Laser Anemometry to Fluid Mechanics, pp. 18.3.Google Scholar
- [7]Durst, F. and Umhauer, H., 1975, Local measurements of particle velocity size distribution and concentration with a combined laser Doppler particle sizing system, Proc. of the LDA-Symp. Copenhagen, pp. 430.Google Scholar
- [8]Durst, F. and Zare, M., 1975, Laser Doppler measurements in two-phase flow, Proc. of the LDA-Symp. Copenhagen, pp. 403.Google Scholar
- [9]Bauckhage, K and Schoene, A., 1985, Measurement of velocities and diameters of small particles by using fast digital circuits for the on-line evaluation of laser-Doppler signals, Part. Caract. 2, pp. 113.Google Scholar
- [10]Bachalo, W.D., Houser, M.J. and Smith, J.N., 1986, Evolutionary behaviour of sprays produced by pressure atomizers, AIAA 24th Aerospace Sciences Meeting, AIAA-86-0296.Google Scholar
- [11]Saffman, M., Buchave, P. and Tanger, H., 1984, Simultaneous measurement of size, concentration and velocity of spherical particles by a laser Doppler method, Int. Symp. on Appl. of Laser Anemometry on Fluid Mech./Lisbon, 8.1.Google Scholar
- [12]Hishida, K., Maeda, M. and Ikai, S., 1980, Heat transfer from a flat plate in two-component mist flow, Trans. ASME J. Heat Transfer vol. 102, pp. 513–518.CrossRefGoogle Scholar
- [13]Hishida, K., Maeda, M. and Ikai, S., 1982 Heat transfer in two-component mist flow; boundary layer structure on an isothermal plate, Proc. 7th IHTC, Munich/FRG vol. 5, pp. 301–306.Google Scholar
- [14]Maeda, M., Hishida, K. and Furutani, T., 1980. Optical measurements of local gas and particle velocity in a upward flowing dilute gas-solids suspension, ASME Polyphase Flow and Transport Technology, BK# H00158, pp. 211-216.Google Scholar
- [15]Maeda, M., and Hishida, K., 1982, Velocity and turbulent intensity measurements of gas and spherical particles in two-phase flows by a modified LDA system, Sensor 82 Transducer Technology and Temperature Measurement vol. 3, pp. 43.Google Scholar
- [16]Hishida, K., Maeda, M., Imaru, J., Hironaga, K. and Kano, H., 1982. Proc. Int. Symp. on Appl. of LDA to Fluid Mech., pp. 5.6, 1982.Google Scholar
- [17]Hishida, K., Maeda, M., Tajima, K. and Kano, H., 1984, Measurements of two-phase turbulent flow by LDA with particle size discrimination, 2nd Int. Symp. on Appl. of LDA to Fluid Mech., pp. 18.4.Google Scholar
- [18]Maheu, B., Letoulouzan, J.N. and Gouesbet, G., 1984, Four flux models to solve the scattering transfer equation in terms of Lorenz-Mie parameters, Appl. Optics 23-19, pp. 3353.Google Scholar
- [19]Belvaux, Y. and Virdi, S.P.S., 1975, A method for obtaining a uniform non-Gaussian laser illumination, Opt. Commun. 15-(2), pp. 193.Google Scholar
- [20]Veldkampf, W.B. and Kastner, C.J., 1982, Appl. Opt. 21(2), pp. 345.CrossRefGoogle Scholar
- [21]Allano, D., G. Gouesbet, Grehan, G. and Lisiecki, D., 1984, Droplet sizing using a top-hat laser beam technique, J. Phys. D: Appl. Phys., 17, pp. 43–58.CrossRefGoogle Scholar
- [22]Quintanilla, M. and de Frutos, A.M., 1981, Holographic filter that transforms a Gaussian into an uniform beam, Appl. Optics, 20-5, pp. 879.Google Scholar
- [23]Ruck, B and Schmitt, F., 1982, Verfahren und Vorrichtung zum Erzeugen einer zylinderformigen Intensitatsverteilung eines Laserstrahl, Patentanmeldung/Deutsches Patentamt, P 33 01 232 6, 82/15369-HfGoogle Scholar
- [24]Knuhtsen, J., Olldag, E. and Buchhave, P., 1982, Fibre-optic laser Doppier anemometer with Bragg frequency shift utilizing polarization-preserving single-mode fibre, J. Phys. Sci. Instrum., 15 pp. 1188-1191.Google Scholar