Two types of flowmeters have been calibrated volumeixically in liquid hydrogen. Two turbine-type flowmeters (a 1½-in.-diameter 5 to 110 gpm, and a 1-in.-diameter 2 to 40 gpm) have been calibrated within their respective ranges. A 1-in. orifice flowmeter has also been calibrated within the same range.
Reynolds Number Liquid Hydrogen Product Division Respective Range Cryogenic Engineer
These keywords were added by machine and not by the authors. This process is experimental and the keywords may be updated as the learning algorithm improves.
This is a preview of subscription content, log in to check access