Automatic Integrators and Gas Chromatography
It is generally desirable to simplify any technical operation. Simplification can mean such things as using less equipment, shortening the time involved, eliminating steps, or automating routine functions. Because of the number and complexity of samples analyzed in many gas-chromatography laboratories, simplification of the analysis is worthwhile. Automatic integration of peak areas can be one way to achieve this simplification.
KeywordsPoor Resolution Electronic Integrator Digital Integrator Automatic Integrator Unresolved Peak
Unable to display preview. Download preview PDF.
- 1.“New Printing Integrator Speed Analysis by Gas Chromatography,” Perkin-Elmer Instrument News, Vol. 9, No. 4, p. 1, 1958.Google Scholar
- 3.S. Dal Nogare, C.E. Bennett, and J. C. Harden, “A Simple Electromechanical Integrator,” in: V.J. Coates, H.J. Noebels, and I.S. Fagerson, eds., Gas Chromatography, Academic Press, New York, 1958.Google Scholar
- 4.K.W. Gardiner, R.F. Klaver, F. Baumann, and J.F. Johnson, “Gas Chromatographic Chart Integrators,” in: N. Brenner, J.E. Callen, and M.D. Weiss, eds., Gas Chromatography, Academic Press, New York, 1962.Google Scholar
- 9.R.D. Johnson, D.D. Lawson, and A. J. Havlik, “Voltageto Frequency Integrators in Gas Chromatography,” J. Gas Chromatog. 3:303, 1965.Google Scholar
- 10.W. L. Perrine, “A Precision Integrator for Gas Chromatography,” in: H. J. Noebels, R. F. Wall, and N. Brenner, eds., Gas Chromatography, Academic Press, New York, 1961.Google Scholar