Some Observations on the Use of Certain Analyzing Crystals for the Determination of Silicon and Aluminum
During the past several years, a number of analyzing crystals have been prepared in the U.S. Air Force Aerospace Research Laboratories. Crystals such as alkaline acid malonates, sucrose, pentaerythritol (PET), and several others have been grown with the sole purpose of application in X-ray fluorescence analysis of silicon, aluminum, and other elements of low atomic number. Crystals from natural sources, such as quartz, mica, and gypsum, have also been procured from different parts of the world for this purpose.
Among the analyzing crystals so prepared, pentaerythritol gave the highest count rate for silicon and aluminum. However, since this crystal is organic in nature, great care must be exercised in handling this type of crystal in order to obtain constant count rates. For the analysis of silicon and aluminum, α-quartz crystal gave a somewhat low count rate, but this crystal has certain advantages over the organic crystals, and can be used for analysis of materials having high silicon and aluminum content.
For instance, when using a conventional X-ray emission vacuum spectrograph of standard make, with the latest alterations, one of the PET crystals attained a count rate of 101,870 cps for pure silicon when the instrument was operated at 60 kVP and 50 mA with a negligible background count. With a constant potential attachment and operated at 60 kVP and 34 mA, 115,600 cps with negligible background count was obtained for silicon. The count rate for aluminum was of the same order of magnitude with somewhat higher background count.
Several sets of standards have been procured and small amounts of silicon and aluminum in these standards have been analyzed by the latest modified vacuum spectrograph. These results are under study and the limit of detection calculated. Procedures and results are described and discussed.
KeywordsCount Rate Pure Silicon Analyze Crystal High Count Rate Flow Proportional Counter
Unable to display preview. Download preview PDF.
- 1.Frank L. Chan, “X-Ray Camera Attachment,” U.S. Patent No. 3, 079, 500.Google Scholar
- 2.Frank L. Chan, “Spherical Goniometer Head and Adapter,” U.S. Patent No. 3, 160, 748.Google Scholar
- 3.Frank L. Chan, “Some Modifications of the X-ray Instruments and Their Utilization to the Study of Analytical Problems,” Norelco Reptr. 10: 133, 1963.Google Scholar
- 4.Frank L. Chan, “Determination of Zinc Sulfide and Cadmium Sulfide in Solid Solutions of Small Single Crystals Used for Semiconductors by X-ray and Chemical Methods,” in: W. M. Mueller, G. R. Mallett, and M. J. Fay (eds.), Advances in X-Ray Analysis, Vol. 5, Plenum Press, New York, 1962, p. 142.Google Scholar
- 5.Frank L. Chan, “A Study of Silicon Determination in Organic-silicon Compounds by X-ray Fluorescence with Vacuum Spectrograph,” paper presented at SAC Conference held at Nottingham, England, July 1965.Google Scholar
- 6.Joseph V. Smith (ed.), Index (Organic) to the X-ray Powder Data File, 1964, ASTM Special Technical Publication 48-N1, American Society for Testing and Materials, Philadelphia, Pa. [see also X-ray Powder Data File Set 1–5 (Revised) ASTM, 1960 ].Google Scholar
- 7.Harold P. Klug and Leroy E. Alexander, X-ray Diffraction Procedures, John Wiley & Sons, Inc., New York, 1954.Google Scholar
- 10.L. S. Birks, X-Ray Spectrochemical Analysis, Interscience Publishers, Inc., New York, 1959.Google Scholar