A Combined Focusing X-Ray Diffractometer and Nondispersive X-Ray Spectrometer for Lunar and Planetary Analysis
An instrument is described which is intended to perform a dual purpose (elemental—structural) analysis consistent with the environmental conditions implied by lunar or planetary operation. The diffractometer section is based on a modified Seeman—Bohlin focusing principle in which a sharp-line focus target, a powdered sample, and a movable detector slit all lie on the focusing circle. The convolution of the projections on the focal circle, of a narrow receiving slit on the detector, the line focus target, combined with a high dispersion produce higher resolution and intensity than is common with Bragg focusing diffractometers with similar instrumental parameters. The range of d-spacings covered is from 1 to 7 Å (chromium target). The chemical analysis section of the instrument utilizes the fluorescent X-rays produced in the specimen by the primary beam. A proportional counter and pulse-height analyzer accomplish detection and energy discrimination. Resolution is low, but the analysis can distinguish between elements in the range of atomic numbers 11 to 29. Data from a breadboard model is presented. The entire unit, although primarily intended to meet the requirements of space, performs equally well as a routine laboratory analyzer. The horizontal, stationary nature of the specimen holder suggests several specific applications.
KeywordsTitanium Methane Quartz Chromium Boron
Unable to display preview. Download preview PDF.
- 1.W. Parrish, Lunar Diffractometer Geometry,in press (1965).Google Scholar
- 2.R. C. Speed, D. B. Nash, and N. L. Nickle, “ A Lunar X-ray Diffraction Experiment,” in: W. M. Mueller, G. R. Mallett, and M. J. Fay (eds.), Advances in X-Ray Analysis, Vol. 8, Plenum Press, New York, 1964, pp. 400–419.Google Scholar
- 3.G. W. Briridley, in: H. S. Peiser, H. P. Rooksby, and A. J. C. Wilson (eds.), Monochromators and Focusing Cameras from X-Ray Diffraction by Polycrystalline Materials, Institute of Physics, London, 1955.Google Scholar
- 4.K. Das Gupta and N. Pan, “A New Experimental Technique for X-Ray Diffraction Study,” Sci. Ind. Res. (India) 17B: 131–133, 1958.Google Scholar
- 5.A. J. C. Wilson, The Mathematical Theory of X-Ray Powdered Diffractometry, Philips Technical Library, Eindhoven, 1963.Google Scholar
- 6.G. Kunze, “Intensitäs-, Absorptions-und Verschiebungsfaktoren von Interferenzlinien bei Bragg—Brentano und Seeman—Bohlin Diffraktometern I,” Z. Angew. Phys., Dec. 24. 1964, p. 522.Google Scholar
- 7.X-Ray Spectrometer Final Report prepared by Philips Electronic Instruments, JPL Contract No. 950159.Google Scholar
- 8.Powder Diffraction File, J. V. Smith (ed.), American Society for Testing and Materials, 1964, Index (inorganic).Google Scholar
- 10.L. S. Birks, Electron Probe Micro-Analysis, Interscience Publishers, Inc., New York, 1963, p. 95.Google Scholar
- 11.A. E. Metzger, A Nondispersive X-Ray Spectrometer for Extraterrestrial Geochemical Analysis, JPL Space Programs Summary, No. 37–31, Vol. IV, 1965, pp. 273–277.Google Scholar