Correction for Non-Linearity of Proportional Counter Systems in Electron Probe X-Ray Microanalysis
While the theoretical basis for the correction of non-linearity of detector systems is well known, methods for the determination of dead-time effects must be adapted to electron probe microanalyzer systems. Two such methods, one employing both X-ray and current measurements and the other employing simultaneous X-ray measurements on two spectrometers, are described. The effect of pulse-height shrinkage at high counting rates on the linearity of the detector system is discussed. When the proposed corrections for the dead-time of X-ray detector systems employing proportional counters are applied to the X-ray intensity measurements obtained with the electron probe microanalyzer, count rates as high as 50,000 counts/sec can be used.
KeywordsCounting Rate Electron Probe Dead Time Pulse Height High Counting Rate
Unable to display preview. Download preview PDF.
- 1.H. Friedman, Electronics 18: 132, 1945.Google Scholar
- 5.N. Hole, Arkiv. Mat ., Astro. Fys. 33A (11), 1947.Google Scholar
- 6.R. Jost, Heiv. Phys. Acta 20: 173, 1947.Google Scholar
- 7.M. Blackman and J. L. Michiels, Proc. Phys. Soc. (London) 60: 594, 1948.Google Scholar
- 8.R. D. Evans, The Atomic Nucleus, MacGraw-Hill Book Company, Inc., New York, 1955.Google Scholar
- 9.D. B. Wittry, in: X-Ray and Electron Probe Analysis, ASTM Special Technical Publication No. 349, ASTM, 1963.Google Scholar
- 11.S. L. Bender and E. J. Rapperport, “The Electron Microprobe,” in: Proceedings of the Symposium on Electron Probe Microanalysis, Electrochemical Society, John Wiley & Sons, Inc., New York, 1966.Google Scholar
- 12.K. F. J. Heinrich, in: W. M. Mueller, G. R. Mallett, and M. J. Fay (eds.), Advances in X-Ray Analysis, Vol. 7, Plenum Press, New York, 1963, p. 325.Google Scholar
- 13.W. Petzold, Z. Angew. Phys. 15: 158, 1963.Google Scholar
- 14.K. F. J.Heinrich, to be published.Google Scholar