A Two-Element Annular Array of Short Fixed Focal Length for Dynamic Focusing
A two-element annular array for dynamic focusing has been designed. It is combined with a lens of short, fixed focal length and is to be used in a mechanically scanned transducer for simultaneous 2D cardiographic echo imaging and 2D doppler flow chart mapping. The goal has been to minimize the number of elements, since it is difficult to make annular arrays with many elements having high sensitivity and equal impulse responses. High sensitivity is needed for the Doppler application and equal impulse responses are needed for dynamic focusing in the echo imaging application. A study of the on-axis spatial impulse response has led to a formula for the required number of annular elements. It has also suggested that one should combine the array with a lens of short fixed focal length to obtain best results in dynamic focusing. To test this new design principle experimentally we have made a two-element transducer and measured the impulse response and sensitivity of each element as well as the transient beam pattern. The measurements are in good agreement with calculations based on the transmission line model and the spatial impulse response method. The array produces a narrow beam close to the diffraction limit in the desired depth range.
KeywordsImpulse Response Focal Length Beam Diameter Beam Profile Transmission Line Model
Unable to display preview. Download preview PDF.
- 1.L. Halte, B.A. Angelsen, “Doppler ultrasound in Cardiology — Physical Principles and Clinical Applications”, New York, Lea and Fabiger 1982.Google Scholar
- 2.P.R. Stepanishen, “The time-dependent force and radiation impedance on a poston in a rigid infinite planar baffle”, JASA 49, 841–849, (1971).Google Scholar
- 3.P.R. Stepanishen, “Transient Radiation from Postons in an Infinite Planar Baffle”, JASA 49, 1627–1638, (1971).Google Scholar
- 4.P.R. Stepanishen, “Wide Bandwidth Acoustic Near and Far Field Transients from Baffled Postons” Proc. 1977 IEEE Ultrason. Symp., 113–118.Google Scholar
- 5.J.W. Strutt (Lord Rayleigh), “Theory of Sound”, Vol. 2, Dover, New York, 1945.Google Scholar
- 6.N. Sponheim, “Focusing of annular arrays”. Submitted to IEEE ultrasonics, Ferroelectrics and Frequency Control.Google Scholar
- 7.N. Sponheim, “Transient Ultrasonic fields of efficient broadband piezoelectric transducers”. Dr.ing.-thesis, The Norwegian Institute of Technology, 1985.Google Scholar
- 8.D.R. Dietz, S.J. Norton, M. Linzer, “Wideband annular array response”, Proc. 1978 IEEE Ultrason. Symp., 206–211.Google Scholar
- 12.J. Stamnes, “Waves in focal regions”, Chapter 12.2, Adam Hilger, Bristol, 1986.Google Scholar
- 13.E. Hecht, A. Zajac, “Optics”, Addison-Wesley, USA, 1979. JASA 72, 1121–1123, (1982).Google Scholar
- 14.H. Engan, “An ultrasound power meter”, JASA 72, 1121–1123, (1982).Google Scholar