Variable Temperature Sample Holder for Raman Spectroscopy
This note describes a simple, convenient sample holder for laser Raman spectroscopy of liquids and powders at any temperature between −196° and +200°C. It was designed for the Spex Ramalog system, which uses 90° viewing. With this system a liquid or powdered sample is contained in a thinwalled melting point tube which is oriented crosswise to the slit, i.e., parallel to the plane of the slit jaws and perpendicular to the long axis of the slit.1 The laser beam impinges on the sample vertically, and the tattered radiation is viewed horizontally.
KeywordsRaman Spectroscopy Sample Tube Metal Support Variable Flow Rate Helical Spring
Unable to display preview. Download preview PDF.
- 1.S. V. Freeman and D. O. Landon, Anal. Chem. 41, 398 (1969).Google Scholar