The Ultrasonic Measurement of Stress on Ferrous Plate using Pulsed Electromagnet Magnetostrictive Emat Probes
Last year we finished the design and assembly of a semi-automatic system for the measurement of stress in textured ferritic steel, Magneto- strictive EMATs, which have no contributions generated by Lorentz force, were used on ferrous alloys since the absence of the Lorentz force mechanism precludes any velocity errors that might be produced by interference of Lorentz force and magnetostrictively generated signals. Phase shifts due to lift-off were solved by driving the transmitter EMAT’s in series. To make reliable wave-speed measurements in an industrial environment with an economical sampling period of 100 nanoseconds, a Fourier Transform-Phase-Slope technique was implemented which not only solves the zero-crossing problem[1,2], but achieves accuracy approaching the Cramer-Rao bound, Initial data are improved over the previous year’s work, however, further refinements are required to archive the expected error bound of ± 2 ksi. These refinements take the form of reducing the effects of temperature induced velocity changes during the measurement process.
KeywordsRadar Hull Acoustics EMAT
Unable to display preview. Download preview PDF.
- 1.S. J. Wormley and R. B. Thompson, “Development of a Breadboard Instrument for the Ultrasonic Measurement of Stress”, Review of Progress in Quantitative NDE 8A, D. O. Thompson and D. E. Chimenti, Eds. (Plenum Press, NY, 1989) pp. 1119–1125.Google Scholar
- 2.S. J. Wormley, R. B. Thompson and Yan Li, “The Ultrasonic Measurement of Stress on Ferrous Plate”, Review of Progress in Quantitative NDE 9B, D. O. Thompson and D. E. Chimenti, Eds. (Plenum Press, NY, 1990), pp. 1797–1804.Google Scholar
- 6.S. J. Wormley, R. B. Thompson, E. P Papadakis, Y. Li and K. Forouraghi, “A Functional Prototypic Instrument for the Ultrasonic Measurement of Formability”, Review of Progress in Quantitative NDE 10A, D. O. Thompson and D. E. Chimenti, Eds. (Plenum Press, NY, 1991).Google Scholar
- 7.S. J. Wormley, K. Forouraghi, Y. Li, R. B. Thompson and E. P. Papadakis, “Application of a Fourier Transform-Phase-Slope Technique to the Design of an Instrument for the Ultrasonic Measurement of Texture and Stress”, Review of Progress in Quantitative NDE 9A, D. O. Thompson and D. E. Chimenti, Eds. (Plenum Press, NY, 1990), pp. 951–958.Google Scholar
- 9.D. R. Hull, H. E. Kautz, and A. Vary, “Measurement or Ultrasonic Velocity Using Phase-Slope and Cross-Correlation Methods”, Materials Evaluation 43, October (1985), pp. 1455–1460.Google Scholar
- 10.P. M. Woodword, “Probability and Information Theory, With Applications To Radar”, 1953, New York: Pergaman.Google Scholar
- 11.A. H. Quazi, “An Over View on the Time Delay Estimate in Active and Passive Systems for Target Localization”, IEEE Transactions on Acoustics, Speech, and Signal Processing, June 1981 volume ASSP-29 3.Google Scholar
- 12.Operator’s Manual, (1990), Internal Document for the Prototypic Instrument under funding provided by the United States Department of Commerce under Grant ITA 87-02.Google Scholar