Demonstration of No-Vent Fill and Solidification of Nitrogen for the Nicmos Dewar
In order to minimize parasitic heat loads and achieve a five-year mission lifetime, the NICMOS solid nitrogen dewar has been designed with a single fluid line serving as both fill and vent for the cryogen tank. A no-vent fill of liquid nitrogen (LN2) is necessary to achieve a reasonable fill time with this configuration. High density of the solid nitrogen (SN2) is also desired in order to maximize dewar efficiency. A test was conducted at Ball Electro-Optics and Cryogenics Division in December of 1992 to demonstrate the capability to perform a no-vent fill of LN2 and to determine the attainable solid nitrogen density. The test was successful in achieving a complete no-vent fill of the 67 L test tank with LN2. The loaded LN2 was then solidified using cold helium gas circulated through the tank wall heat exchanger. A series of additional loading steps achieved a bulk cryogen fill density of 94% of the maximum SN2 density at 60K. The no-vent fill procedure should be applicable to other solid cryogen systems in which the cryogen can exist in a liquid phase, and could be a candidate method for on-orbit replenishment of certain spaceborne liquid cryogen systems.
KeywordsFoam Platinum Helium Stratification Boulder
Unable to display preview. Download preview PDF.
- 2.D. J. Chato, “Ground Testing of the Nonvented Fill Method of Orbital Propellant Transfer: Results of Initial Test Series”, AIAA Paper 91-2326, AIAA/SAE/ASME/ASEE 27th Joint Propulsion Conference, Sacramento, CA, June 24–27, 1991.Google Scholar
- 5.P. A. Beck, Internal Communication, Beech Aircraft Corporation, Boulder Division, January 25, 1982.Google Scholar