Microwave Immunohistochemistry: Advances in Temperature Control
The advantages and limitations to employing a fiber-optic temperature probe for feedback control of a microwave oven designed for immunohistochemistry were tested. The probe was designed to be submerged in a drop of immunoreagent on a glass slide, and the measured temperature of the droplet was used to regulate the output power and the time needed for the magnetron to reach and maintain the desired incubation temperature. A three-step streptavidin-biotin-peroxidase microwave method, employing a range of antibodies, particularly anti-insulin on rat pancreas, was used as the model system to illustrate the effect of temperature and duration of incubation in microwave immunostaining. Insight into the factors affecting a droplet temperature under microwave irradiation and the relationship among temperature, duration, concentrations of antibodies and immunoreaction for immunostaining were obtained. The optimal temperature and duration for microwave immunostaining in our model system were 37°C for 3 min at 75% power level followed by a 2-min incubation inside the oven for all three stages of immunostaining. Use of droplet temperature-regulated microwave standardizes conditions used in microwave immunohistochemistry. Other potential applications where uniform heating of small amounts of reagent and control of reagent temperature are important may benefit from this technique.
KeywordsPower Level Water Load Droplet Temperature Evaporative Water Loss Color Plate
Unable to display preview. Download preview PDF.
- 1.Boon, M.E. and L.K. Kok (Eds.). 1992. Microwave irradiation in immunostaining, p.256–285. In Microwave Cookbook of Pathology: The Art of Microscopic Visualization, 3rd ed. Columbia Press, Leiden.Google Scholar
- 2.Boon, M.E. and L.K. Kok. 1992. Microwave oven and temperature control, p.74–115. In Microwave Cookbook of Pathology: The Art of Microscopic Visualization. 3rd ed. Columbia Press, Leiden.Google Scholar
- 3.Boon, M.E., L.P. Kok, H.E. Moorlag and A.J.H. Suurmeljer. 1989. Accelerated immunogold silver and immunoperoxidase staining of paraffin sections with the use of microwave irradiation. Factors influencing results. Am. J. Clin. Pathol. 97:137–143.Google Scholar
- 7.Choi, T-S., M.M. Whittlesey, S.E. Slap, V.M. Anderson and J. Gu. 1995. Advances in temperature control of microwave immunohistochemistry. Cell Vision 2:151–164.Google Scholar
- 8.Clearly, S.F. 1978. Survey of microwave and radio frequency biological effects and mechanisms. D.H.E.W. Publication (FDA) 78–8055:1–33.Google Scholar
- 12.Jennette, J.C. and M.R. Wick. 1990. Immunohistochemical techniques, p. 1–28. In J.C. Jennette (Ed.), Immunohistology in Diagnostic Pathology. CRC Press, Boca Raton, FL.Google Scholar
- 13.Klein, J. (Ed.). 1982. Lymphocyte-activating substances, p.347–414. In Immunology: The Science of Self-Nonself Discrimination. John Wiley & Sons, New York.Google Scholar
- 17.Leong A.S.-Y. 1994. Microwave technology for morphological analysis. Cell Vision 7:278–288.Google Scholar
- 18.Stapleton, M.J., M.C. Levin and S. Jacobson. 1994. DNA amplification within cells on slides: advances in evaporation and temperature control. Cell Vision 7:177–181.Google Scholar
- 19.Takes, P.A., J. Kohrs, R. Krug and S. Kewley. 1989. Microwave technology in immunohistochemistry: application to avidin-biotin staining of diverse antigen. J. Histotechnol. 72:95–98.Google Scholar