3D Misorientation of Cantilevers and Its Impact in Friction Force Microscopy
The use of AFM in nanotribology is limited by our capacity to fully estimate the cantilever’s properties. Especially, cantilever’s stiffness plays a major part in the estimation of the contact forces. In this paper, we look at the 3D misorientation of a perfect triangular cantilever in static. The analysis was carried using Finite Element Analysis to estimate the stiffness matrix at the tip end. Results show a small influence of the misorientation in bending. However misorientation along the torsion axis has a notable impact on the estimation of lateral forces and introduce a mechanical cross-talk that can influence machine’s measurement.
KeywordsAFM FEM friction
- Varenberg V, Etsion I and Halperin G, 2003, “An improved wedge calibration method for lateral force in atomic force microscopy”. Rev. Sei. Instrum. 74 3362–7.Google Scholar
- Cannara R J, Eglin E and Carpick R W, 2006, “Lateral force calibration in atomic force microscopy: A new lateral force calibration method and general guidelines for optimization”. Rev. Sei. Instrum. 77 053701.Google Scholar
- Piner R and Ruoff R S, 2002, “Cross talk between friction and height signals in atomic force microscopy” Rev. Sei. Instrum. 73 9 3392-4.Google Scholar
- Muller M, Schimmel T et al, 2005, “Finite element analysis of V-shaped cantilevers for atomic force microscopy under normal and lateral force loads”, Surf. Interface Anal 38 1090-5.Google Scholar