Optimization of the angle between detector modules in a dual-head cardiac SPECT
- 114 Downloads
In recent years, dedicated cardiac single photon emission computed tomography (SPECT) systems have been undergoing a profound change in design with multiple detectors and various angles between the modules to improve the sensitivity and the resolution by reducing the distance between the heart and the detector. The performance of a dual-head cardiac SPECT for small-animal imaging was characterized as a function of the angle between two detector heads by using GATE simulations, and simulation data were validated with experimental results. Each detector head consists of 50 × 50 × 6 mm3 NaI(Tl) optically coupled to a Hamamatsu H8500 position sensitive photomultiplier (PSPMT) and a low-energy high-resolution parallel-hole collimator (LEHR, septal thickness: 0.2 mm, diameter: 1.9 mm). The distance between the collimator surface and the center of rotation was set as 20, 20, 20, 25, or 31.5 mm for 70°, 80°, 90°, 100°, or 110°, respectively, based on a 40-mm field of view (FOV). A point source and a rat cardiac phantom of Tc-99m in scattering media were simulated. Projection data were acquired for 180 angular views in steps of 2° and were reconstructed by using a filtered back-projection algorithm. Results demonstrated that the angle between the detector heads did not make a big difference in the image quality when scattering media were not presented, but the dual heads in the 80° geometry provided the best spatial resolution in the cardiac phantom study. The peak-to-valley ratio between the myocardial wall and the cavity was measured as 1.87, 11.01, 3.28, 3.40, or 2.46 for 70°, 80°, 90°, 100°, or 110°, respectively. Experiments were performed with a dual-head SPECT in the 80° geometry, and the results agreed well with these from the simulations. In this study, the impact of the angle between dual detector heads on the imaging performance was evaluated, and the optimal angle was derived for a dedicated cardiac SPECT.
KeywordsCardiac SPECT GATE Monte carlo simulation
Unable to display preview. Download preview PDF.
- M. I. Travin, Semin. Nucl. Med. 3, 182 (2011).Google Scholar
- R. L. Eisner et al., J. Nucl. Med. 11, 1717 (1986).Google Scholar
- J. T. Bushberg et al., The Essential Physics of Medical Imaging (Lippincott Williams & Wilkins, North American Edition, 2011), p. 712.Google Scholar