Alternatives to polysomnography (PSG): A validation of wrist actigraphy and a partial-PSG system
- 1.4k Downloads
The objective of this study was to assess the validity of a sleep/wake activity monitor, an energy expenditure activity monitor, and a partial-polysomnography system at measuring sleep and wake under identical conditions. Secondary aims were to evaluate the sleep/wake thresholds for each activity monitor and to compare the three devices. To achieve these aims, two nights of sleep were recorded simultaneously with polysomnography (PSG), two activity monitors, and a partial-PSG system in a sleep laboratory. Agreement with PSG was evaluated epoch by epoch and with summary measures including total sleep time (TST) and wake after sleep onset (WASO). All of the devices had high agreement rates for identifying sleep and wake, but the partial-PSG system was the best, with an agreement of 91.6 % ± 5.1 %. At their best thresholds, the sleep/wake monitor (medium threshold, 87.7 % ± 7.6 %) and the energy expenditure monitor (very low threshold, 86.8 % ± 8.6 %) had similarly high rates of agreement. The summary measures were similar to those determined by PSG, but the partial-PSG system provided the most consistent estimates. Although the partial-PSG system was the most accurate device, both activity monitors were also valid for sleep estimation, provided that appropriate thresholds were selected. Each device has advantages, so the primary consideration for researchers will be to determine which best suits a given research design.
KeywordsActigraphy Accelerometry Wireless sleep monitoring
The authors wish to express appreciation to Michele Lastella for his assistance in data collection.
- CamNTech Ltd. (2008). The Actiwatch user manual v7.2. Cambridge, UK: Author.Google Scholar
- de Souza, L., Benedito-Silva, A. A., Pires, M. L. N., Poyares, D., Tufik, S., & Calil, H. M. (2003). Further validation of actigraphy for sleep studies. Sleep, 1, 81–85.Google Scholar
- Evans, D. S., Snitker, S., Wu, S.-H., Mody, A., Njajou, O. T., Perlis, M. L., … Hsueh, W.-C. (2011). Habitual sleep/wake patterns in the Old Order Amish: Heritability and association with non-genetic factors. Sleep, 34, 661–669.Google Scholar
- Kosmadopoulos, A., Sargent, C., Zhou, X., Darwent, D., & Roach, G. D. (2012). Two for one? Validating an energy expenditure monitor against polysomnography for sleep/wake measurement. In X. Zhou & C. Sargent (Eds.), Sleep of different populations (pp. 16–20). Adelaide, Australia: Australasian Chronobiology Society.Google Scholar
- Kushida, C. A., Chang, A., Gadkary, C., Guilleminault, C., Carrillo, O., & Dement, W. C. (2001). Comparison of actigraphic, polysomnographic, and subjective assessment of sleep parameters in sleep-disorderd patients. Sleep Medicine, 2, 389–396. doi: 10.1016/S1389-9457(00)00098-8 PubMedCrossRefGoogle Scholar
- Landis, J. R., & Koch, G. G. (1977). The measurement of observer agreement for categorical data. Biometrics, 33, 159–174. Retrieved from www.jstor.org/stable/2529310
- Oakley, N. R. (1997). Validation with polysomnography of the Sleepwatch sleep/wake scoring algorithm used by the Actiwatch activity monitoring system (Technical Report). Bend, OR: Mini-Mitter Co.Google Scholar
- Rechtschaffen, A., & Kales, A. (1968). A manual of standardized terminology, techniques, and scoring system for sleep stages of human subjects. Los Angeles, CA: University of California, Brain Information Service/Brain Research Institute.Google Scholar
- Sargent, C., Halson, S., & Roach, G. D. (2012). Sleep or swim? Early-morning training severely restricts the amount of sleep obtained by elite swimmers. European Journal of Sport Science. doi: 10.1080/17461391.2012.696711
- Sim, J., & Wright, C. C. (2005). The Kappa statistic in reliability studies: Use, interpretation, and sample size requirements. Physical Therapy, 85, 257–268. Retrieved from www.physicaltherapyjournal.com/