- First Online:
- Cite this article as:
- Gridnev, D.G., Kanonidi, K.K., Kanonidi, K.D. et al. Seism. Instr. (2008) 44: 32. doi:10.3103/S0747923908010040
- 22 Downloads
The microbarograph of a new design has been offered, and the general principle of its operation is considered. The microbarograph is distinguished from well-known ones; it incorporates a compensatory spring eliminating a zero-point drift of the measuring spring, two additional springs that upgrade the angular displacement sensitivity of the microbarograph, and a control and calibration block that makes it possible to perform the remote calibration of the microbarograph and to control it.