Ultra-Low Power High-Input Impedance Subthreshold CMOS Neural Front-End Amplifier
- 2 Downloads
An ultra-low power, voltage-mode front-end amplifier (FEA) for neural applications featuring subthreshold design is presented. This has been a topic of much research in implantable medical prosthetic devices during the past few decades to monitor and treat the neural disorders such as hearing or sight dysfunctions, epilepsy, Parkinson’s disease, paralysis. The FEA performs a critical signal detection operation in neural monitoring systems to ensure the biosignal fidelity. A matched double-MOS feedback technique is used to compensate the input leakage currents generated by low noise amplifier in the form of integrated circuit (IC), which is the primary reason for immense signal leakage in the input bias network. Therefore, this loop topology ensures that FEA maintains high impedance across a wide range of input frequency. The proposed FEA is implemented by using an SK Hynix 0.18 µm CMOS process. This IC consumes 320 nW in the area of 0.016 mm2 and achieves the input impedance of 44.9 GΩ, and the input-referred noise of 153 nV/Hz1/2.
Unable to display preview. Download preview PDF.
- 2.Y.-S. Sung, W.-M. Chen, C.-Y. Wu, “The design of 8-channel CMOS area-efficient low-power current-mode analog front-end amplifier for EEG signal recording,” Proc. of IEEE Int. Symp. On Circuits and Systems, 22–25 May 2016, Montreal, Canada (IEEE, 2016), pp. 530–533. DOI: https://doi.org/10.1109/ISCAS.2016.7527294.Google Scholar
- 5.D. Y. Barsakcioglu, Yan Liu, P. Bhunjun, J. Navajas, A. Eftekhar, A. Jackson, R. Q. Quiroga, T. G. Constandinou, “An analogue front-end model for developing neural spike sorting systems,” IEEE Trans. Biomed. Circuits Syst. 8, No. 2, 216 (2014). DOI: https://doi.org/10.1109/TBCAS.2014.2313087.CrossRefGoogle Scholar
- 15.R. Shulyzki, K. Abdelhalim, A. Bagheri, M. T. Salam, C. M. Florez, J. L. P. Velazquez, P. L. Carlen, R. Genov, “320-channel active probe for high-resolution neuromonitoring and responsive neurostimulation,” IEEE Trans. Biomed. Circuits Syst. 9, No. 1, 34 (2015). DOI: https://doi.org/10.1109/TBCAS.2014.2312552.CrossRefGoogle Scholar