Imaging of polar cap patches with a low-cost airglow camera: pilot observations in Svalbard, Norway
KeywordsIonosphere Polar cap patches Airglow Optical observations Satellite navigation
Electron Multiplying Charged-Coupled Device
Full Width Half Maximum
interplanetary magnetic field
medium-scale traveling ionospheric disturbances
optical mesosphere thermosphere imagers
Super Dual Auroral Radar Network
During the night, weak and almost invisible light, referred to as airglow or nightglow, is released from the Earth’s upper atmosphere. There are two major emission lines of airglow from atomic oxygen; at 557.7 nm and 630.0 nm. Both of these have been used to study various phenomena in the mesosphere and thermosphere. The optical intensity of the 630.0 nm emission of oxygen (so-called red-line oxygen emission) from the lower thermosphere at an altitude of 200–300 km is known to be proportional to the densities of O+ and O2 (e.g., Sobral et al. 1993). The 630.0 nm emission can, therefore, be used as a proxy for variations in the electron density slightly below the peak height of the F-region ionosphere. In the last 2 decades, the oxygen red-line emission observed with all-sky airglow imagers (ASIs) equipped with cooled charge-coupled device (CCD) cameras has been widely used to observe various kinds of ionospheric phenomena (Shiokawa et al. 2009, and references therein). High signal-to-noise ratio images of the ionosphere from ASIs have enabled us to observe the detailed two-dimensional structure and temporal evolution of plasma bubbles (e.g., Otsuka et al. 2002) in the low-latitude region, as well as traveling ionospheric disturbances of various scales in the mid-latitude region (Shiokawa et al. 2003a, b).
In the last decade, similar airglow observations at high-latitudes have made it possible to visualize the dynamic behavior of polar cap patches (Hosokawa et al. 2006; Moen et al. 2007). Polar cap patches are known as regions of high-density plasma in the F-region, which are generated near the dayside cusp and subsequently propagate towards the nightside across the central polar cap (Crowley 1996, and references therein). The dynamic characteristics and generation processes of polar cap patches have been investigated for almost 3 decades since their discovery in early 1980s (Weber et al. 1984). In recent years, it has been commonly recognized that the two-dimensional imaging capability of ASIs is useful for visualizing the large-scale spatial structure of polar cap patches (Hosokawa et al. 2014, and references therein). For instance, Hosokawa et al. (2009b) demonstrated that the overall motion of patches is controlled by the magnitude and orientation of the interplanetary magnetic field (IMF). Hosokawa et al. (2009a) combined observations of plasma irregularities by the coherent HF radar of SuperDARN using ASI data from Resolute Bay, Canada, and showed that plasma irregularities are distributed in the entire region of the patches. More recently, Hosokawa et al. (2013a, b) identified mesoscale structures within or in the vicinity of polar cap patches by utilizing the high spatial and temporal resolution of the electron-multiplying charge-coupled device (EMCCD) ASI system in Longyearbyen, Norway.
As mentioned previously, ASIs are one of the essential tools for monitoring the spatial and temporal evolution of various ionospheric phenomena at all latitude regions. However, it is still very difficult to make a dense network of ASIs and to monitor the large-scale structure of such ionospheric phenomena, because ASI systems are relatively expensive. In addition, conventional ASIs equipped with a cooled CCD detector tend to be large (e.g., Shiokawa 1999); thus, the engineering work at the time of deployment and during maintenance would be difficult, especially in the severe conditions of the high Arctic and Antarctic regions. If a low-cost, compact ASI can be developed with the confirmed capability of observing the 630.0 nm airglow, it would contribute to building a network of 630.0 nm airglow observations over a large area. Recently, Hosokawa et al. (2014) combined data from two ASIs in Resolute Bay and Longyearbyen, respectively, and demonstrated that the measurements of the 630.0 nm airglow taken with a large field-of-view (FOV) provide a powerful monitoring system for the propagation characteristics of polar cap patches. In this sense, there is a strong demand to construct a network of 630.0 nm airglow measurement using low-cost, compact ASI systems.
For this purpose, we developed an ASI system for observation of 630.0 nm airglow emission. The ASI is composed of a low-cost camera, fish-eye lens, and an optical filter. We have been operating this low-cost ASI in Longyearbyen, Norway, since October 2013 to evaluate its performance. In this paper, we investigate the ASI data obtained during an interval of polar cap patches on the night of December 4, 2013. In particular, we compare the data from the low-cost ASI with those from an EMCCD ASI operating at the same observatory. Through this comparison, we quantitatively evaluate the feasibility of making 630.0 nm airglow measurements with the low-cost ASI system.
In this study, we employ two ASI systems that have been operative in Longyearbyen (78.1° N, 15.5° E), Norway. The low-cost ASI (hereafter called WATEC ASI) consists of a small CCD camera (WATEC-910HX, manufactured by Watec Co. Ltd., Japan), a narrow pass-band optical filter (Edmund Co. Ltd., with a central wavelength of 632.0 nm and full-width half maximum (FWHM) of 10 nm), and a fish-eye lens (YV2.2X1.4A-SA2, Fujinon Co. Ltd.). We accumulated 256 frames (the exposure for one single frame is 1/60 s) before streaming from the camera as a video signal, and thus, the exposure time of the streaming images is ~ 4.3 s. These raw images from the WATEC ASI were digitized (8 bit: 0–255) every 4 s with a video encoder (AXIS Q7404, AXIS Co. Ltd.) and recorded by a computer as 640 × 480 pixels JPEG files. There is an offset of 2 nm between the central wavelengths of the optical filter (632.0 nm) from the 630.0 nm airglow line. However, the filter provides good coverage of the 630.0 nm emission because of its wide bandwidth (~ 10 nm). The WATEC ASI is operative when the sun is below the horizon, even during the full moon period. The technical details of the WATEC ASI system, including the photos of the ASI system, are fully detailed by Ogawa (2019). The cost of the entire WATEC system is ~ 1000 USD, which is approximately 1% of that of the large cooled CCD ASI system.
The other optical system used in this study is an ASI equipped with a cooled EMCCD detector (hereafter called EMCCD ASI), which is composed of a Nikon ASI-2 system (Taguchi et al. 2004) and an EMCCD camera (C9100-13, Hamamatsu Photonics), which has an imaging part of 512 × 512 pixels. Narrow pass-band interference filters (Andover Co. Ltd.) are used for observing oxygen emissions at 557.7 nm and 630.0 nm. The background continuum emission is also sampled at a wavelength of 572.5 nm (where no distinct airglow line exists) to derive the absolute optical intensity. The imager was deployed in Longyearbyen, Norway, in October 2012 (Taguchi et al. 2012), and has since been operating without any technical problems. The EMCCD ASI observes the 630.0 nm airglow images with an exposure time of 4 s to detect weak airglow signatures of polar cap patches. Details of this ASI system are documented in Taguchi et al. (2012), and some recent examples of polar cap patch measurements can be found in Hosokawa et al. (2016a, b). In this study, we employ the 630.0 nm airglow images from the EMCCD ASI to evaluate the performance of the WATEC ASI.
We note that, at the time of comparison, the central wavelength of the optical filter for the EMCCD ASI is 630.0 nm and FWHM is 3 nm, which is narrower than that of the optical filter used for the WATEC ASI, and as such may introduce a difference in the absolute optical intensity values from the two ASI systems. The WATEC imager cannot use the narrow pass-band filter that was used for the EMCCD imager. The reason is as follows: to prevent a decrease in transmittance, the WATEC imager does not produce parallel ray path in the optics. Therefore, the light passes the filter obliquely. As the incident angle increases, the transmission region shifts to the shorter wavelength, while the shape of the transmittance band does not change significantly. As described in Ogawa (2019), the incident angles of the ray path are mostly within about 18 deg. The use of relative wider pass-band filter allows us to observe the 630.0 nm emission even when the incident angle reaches this value (i.e., 18°).
Figure 2a shows a raw 630.0 nm all-sky image simultaneously obtained by the WATEC ASI with an exposure time of 4 s. Although the rough outline of the patch can be identified, it is almost impossible to visualize the detailed spatial structure, for example, the finger-like structure seen in Fig. 2b. To improve the signal-to-noise ratio of the image, we integrated consecutive images and derived average images. Figure 2c shows an average of seven consecutive images, the total exposure time being ~ 30 s. The reduction of the noise makes it possible to see the large-scale structure of the patch. Figure 2d shows an average of 15 images, thereby making it possible to see slight indications of the finger-like structure along the trailing edge. Ogawa (2019) estimated the average noise level of the raw images from the 630.0 nm WATEC ASI to be 30 R. Because thermal noise is known to be inversely proportional to the square root of the number of images for integration, the noise in Fig. 2c would be reduced to ~ 1/4 of that in Fig. 2a, which is ~ 8 R. The typical luminosity of polar cap patches is 100–300 R (e.g., Hosokawa et al. 2009b), and thus, the signal-to-noise ratio from the WATEC ASI would range from 12 to 37 when we employ 15 images for averaging.
Figure 2e and f shows averages of 30 and 75 images, respectively. The signal-to-noise ratio is further improved, but the structure of the patch is somewhat blurred. This is primarily because patches normally move with a background convection speed of 500–1000 m s−1 (Hosokawa et al. 2009b). Hence, there is a trade-off between the signal-to-noise ratio of the images and the discreteness of the observed patch structure, which makes it difficult to select the best integration time in the post-processing procedure. During the current interval, however, the averages of 7 or 15 images (i.e., 30 s or 1 min total integration times) would be appropriate for visualizing the spatial structure of the patch in various scale sizes. An animation of the all-sky images from the simultaneous measurements of EMCCD and WATEC ASIs is provided as Additional file 1. The animation better demonstrates that the quality of the WATEC ASI data is sufficient for visualizing the propagation characteristics and large-scale spatial structure of polar cap patches.
In this paper, we have shown that the low-cost, compact airglow imager can be used for the study of polar cap patches. The utilization of such an ASI system would contribute to the expansion of the existing network of ASIs in the polar cap regions. Currently, several imagers have been operative at polar cap latitudes, for example, at Longyearbyen and Ny Alesund in Norway, Resolute Bay and Eureka in Canada, and a few sites in northern Greenland. To date, however, we have not been able to perform global imaging of the polar cap ionosphere by combining these ASIs, because the coverage of their FOV is still limited. If we deploy the WATEC ASI systems at several sites in northern Canada, Alaska, and Greenland, then a network of 630.0 nm airglow measurements with a much larger coverage would be established. In the southern hemisphere, there are a few on-going projects of unmanned networks of Geospace monitoring systems, which will be installed in the inland region of Antarctica (Ogawa 2019, and references therein). Although it is necessary to further reduce power consumption and establish a data transfer procedure in this location, if we install the WATEC ASI in such an unmanned observatory, a network of 630.0 nm airglow observations would be possible even at the polar cap latitudes in the southern hemisphere. This network would be used to carry out the interhemispheric comparison study of the polar cap patch and arc characteristics.
Another possible use of the WATEC ASI is for airglow measurements at mid-latitudes, low-latitudes, and in the equatorial regions. In these regions, various phenomena occur in the F-region ionosphere during nighttime, such as plasma bubbles and MSTIDs. In earlier literature, larger ASI systems such as OMTIs (Shiokawa et al. 2009) have been used for measuring such features. Kubota et al. (2000) carried out a campaign measurement of MSTIDs by distributing five large ASI systems in mainland Japan. They succeeded in visualizing the large-scale structure of southwestward propagating wave fronts of MSTID across the Japanese continent. However, the source and destination of such features are still unknown due to the limited coverage of ASIs in the mid-latitude region. In the case of plasma bubbles, most of the airglow measurements are regional. If an array of the WATEC ASIs is established in the southeast Asia and Oceania region, it would be possible to monitor the temporal evolution of a plasma bubble from its initiation. For such future measurements in the mid- and low-latitude regions, we have installed the WATEC ASIs into several test sites in mid- and low-latitude regions, and tried to evaluate their capability of detecting MSTIDs and plasma bubbles. Once the feasibility of such airglow measurements is confirmed, the WATEC ASI will open a new door for the global imaging of the ionosphere with ground-based airglow measurements.
We have employed a low-cost CCD camera (WAT-910HX, Watec Co. Ltd) for airglow measurements in the polar regions. The camera is composed of a fish-eye lens and an optical filter whose central wavelength is 632.0 nm, FWHM is ~ 10 nm, and transmittance is 85%. The aim is to observe the 630.0 nm airglow from the F-region ionosphere in a two-dimensional fashion. The low-cost ASI was installed in Longyearbyen (78.1°N, 15.5°E), Norway, in October 2013, and operated continuously during the winter of 2013/2014. In this paper, we attempted to use the low-cost ASI to observe faint airglow variations caused by polar cap patches, which are regions of weak airglow enhancement in the polar cap regions. In Longyearbyen, airglow measurements with an ASI equipped with EMCCD detector (EMCCD ASI) have been carried out; thus, we are able to compare the images from these two ASI systems and evaluate the performance of the low-cost ASI quantitatively. On the night of December 4, 2013, a series of polar cap patches were observed by the EMCCD ASI. The optical intensity of the patches was as large as 400 Rayleigh. Simultaneous to this, the low-cost ASI also detected regions of enhanced airglow intensity passing through their FOV. The signal-to-noise ratio of the raw images obtained every 4 s was evidently lower than those from the EMCCD ASI, but integration of 7–15 consecutive images made it possible to capture the two-dimensional structure of the polar cap patches. However, the estimated values of the absolute optical intensity from the low-cost ASI were found to be slightly larger than those from the EMCCD ASI when the optical intensity was low. This offset may be explained by the fact that the bandwidth of the optical filters used for the low-cost ASI is three times wider than that of the EMCCD ASI. The results of the current study demonstrate that the 630.0 nm airglow measurements using a low-cost and compact CCD camera can be used for studies of various phenomena in the F-region ionosphere.
We would like to thank Fred Sigernes for their help at UNIS, Longyearbyen. We also thank Shin Suzuki, Kazuo Shiokawa, and Akira Kadokura for their helpful comments for making the EMCCD ASI and its calibration, and Takeshi Aoki, and staff of Manufacturing Design Center for facilitating our work in the University of Electro-Communications.
KH designed and conducted the current research and prepared the manuscript. YO developed and calibrated the low-cost all-sky imager used for the observation. ST operated and maintained the EMCCD ASI in Longyearbyen and discussed the results of the data analysis. All authors read and approved the final manuscript.
This work was supported by JSPS-Kakenhi (22340143, 26302006, and 17H02967).
Ethics approval and consent to participate
Consent for publication
The authors declare that they have no competing interests.
- Crowley G (1996) Critical review of ionospheric patches and blobs, in Review of Radio Science 1993–1996. Oxford University Press, New York, p 619Google Scholar
- Hosokawa K, Shiokawa K, Otsuka Y, Ogawa T, St-Maurice J-P, Sofko GJ, Andre DA (2009a) The relationship between polar cap patches and field-aligned irregularities as observed with an all-sky airglow imager at Resolute Bay and the PolarDARN radar at Rankin Inlet. J Geophys Res. https://doi.org/10.1029/2008JA013707 CrossRefGoogle Scholar
- Kubota M, Shiokawa K, Ejiri MK, Otsuka Y, Ogawa T, Sakanoi T, Fukunishi H, Yamamoto M, Fukao S, Saito A (2000) Traveling ionospheric disturbances observed in the OI 630-nm nightglow images over Japan by using a multi-point imager network during the FRONT campaign. Geophys Res Lett 27:4037–4040CrossRefGoogle Scholar
- Ogawa, Y., Y. Tanaka, A. Kadokura, K. Hosokawa, Y. Ebihara, T. Motoba, B. Gustavsson, U. Brandstroem, Y. Sato, S. Oyama, T. Raita, F. Sigernes, S. Nozawa, K. Shiokawa, M. J. Kosch, K. Kauristie, C. Hall, S. Suzuki, Y. Miyoshi, A. Gerrard, H. Miyaoka, and R. Fujii (2019), Development of low-cost multi-wavelength imager system for studies of auroras and airglows, submitted to Polar ScienceGoogle Scholar
- Shiokawa K, Katoh Y, Satoh M, Ejiri MK, Ogawa T (2000) Integrating-sphere calibration of all-sky cameras for nightglow measurements. Adv Space Res 26(1025–1028):2000Google Scholar
- Sobral JHA, Takahashi H, Abdu MA, Muralikrishna P, Sahai Y, Zamlutti CJ, dePaula ER, Batista PP (1993) Determination of the quenching rate of the O(1D) by O(3P) from rocket-borne optical (630 nm) and electron density data. J Geophys Res 98:7791–7798. https://doi.org/10.1029/92JA01839 CrossRefGoogle Scholar
- Taguchi M, Okano S, Sakanoi T, Koizumi N, Aso T, Ejiri M (2002) A new meridian imaging spectrograph for the auroral spectroscopy. Adv Polar Upper Atmos Res 16:99–110Google Scholar
- Taguchi M, Ejiri M, Tomimatsu K (2004) A new all-sky optics for aurora and airglow imaging. Adv Polar Upper Atmos Res 18:140–148Google Scholar
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.