Protocol to perform pressurized blister tests on thin elastic films
This work aims to identify common challenges in the preparation of the blister test devices designed for the measurement of the energy release rate for brittle thin films and to propose easy-to-implement solutions accordingly. To this end, we provide a step-by-step guide for fabricating a blister test device comprised of thin polystyrene films adhered to glass substrates. Thin films are first transferred from donor substrates to an air-water interface, which is then used as a platform to locate them on a receiver substrate. We embed a microchannel at the back of the device to evacuate the air trapped in the opening, through which the pressure is applied. We quantify the height and the radius of the blister to estimate the adhesion energy using the available expressions correlating the normal force and the moment with the shape of the blister. The present blister test provided an adhesion energy per unit area of G = 18±2 mJ/m^2 for polystyrene on glass, which is in good agreement with the measurement of G = 14±2 mJ/m^2 found in our independent cleavage test.
KeywordsTips and Tricks
- 13.M. Dupeux, A. Bosseboeuf, Application of the Blister Test to Adhesion Energy Measurements in Metal/Ceramic Film-on-Substrate Systems, (Springer Netherlands, Dordrecht, 1998) pp. 319--327Google Scholar
- 26.J. Brandrup, E.H. Immergut, E.A. Grulke, A. Abe, D.R. Bloch, Polymer Handbook, fourth edition (Wiley, New York, 1989)Google Scholar
- 35.E. Rio, F. Boulogne, Withdrawing a solid of a bath: How much liquid is coated?, to be published in Adv. Colloid Interface Sci. (2017) DOI:10.1016/j.cis.2017.01.006