The shear strength of Ti–HA composite coatings for intraosseous implants
- 39 Downloads
Plasma spraying of composite coatings is developed and investigated. Three-dimensional capillary porous titanium (3DCP Ti) coatings with a thickness of 1 mm are sprayed using a wire. Hydroxyapatite (HA) coatings with a thickness of 0.08–0.35 mm are sprayed on 3DCP Ti coatings at a temperature of 300–550°C. The joint between the coating and plastic is analyzed at shear. The plastic simulates bone tissue that grows into the coating surface. The heating of the 3DCP Ti coating to 550°С when the HA coating is being sprayed increases the shear strength of the coating with respect to the plastic to 9.8 MPa. Modeling approximations are proposed for the shear of the joint between the coating and the plastic.
Keywordsimplants plasma spraying composite coatings three-dimensional capillary porous titanium hydroxyapatite shear strength plastic model
Unable to display preview. Download preview PDF.
- 3.Sahay, V., Lare, P.J., and Hahn, H., Physical and mechanical characterization of porous coatings for medical and dental devices, Proc. Third National Thermal Spray Conf. “Thermal Spray, Research, and Application, Long Beach, CA, May 20–25, 1990”, Metals Park, 1990, pp. 425–430.Google Scholar
- 4.Harris, D.H., Bioinert CP–Ti and Ti–6Al–4V Coatings by the Arc-Plasma Spray Process, Dayton, OH: Titan. Dev. Assoc., 1990, pp. 636–645.Google Scholar
- 5.Borsari, V., Giavaresi, G., Fini, M., Torricelli, P., Salito, A., Chiesa, R., and Giardino, R., Physical characterization of different-roughness titanium surfaces, with and without hydroxyapatite coating, and their effect on human osteoblast-like cells, J. Biomed. Mater. Res., Part B, 2005, vol. 75, no. 2, pp. 359–368.Google Scholar
- 6.Borsari, V., Fini, M., Giavaresi, G., Tschon, M., Chiesa, R., Chiusoli, L., and Giardino, R., Comparative in vivo evaluation of porous and dense duplex titanium and hydroxyapatite coating with high roughnesses in different implantation environments, J. Biomed. Mater. Res., 2009, vol. 89, no. 2, pp. 550–560.CrossRefGoogle Scholar
- 7.Kalita, V.I., Gnedovets, A.G., Mamaev, A.I., Mamaeva, V.A., Malanin, D.A., and Pisarev, V.B., Plasma deposited bioactive porous coatings for intrabone implants, Proc. 17th Int. Symp. on Plasma Chemistry (ISPC–17), Toronto, August 7–12, 2005, Toronto: Univ. of Toronto Press, 2005,pp. 1105–1106.Google Scholar
- 8.Gnedovets, A.G., Kalita, V.I., Komlev, D.I., Yerokhin, A.L., and Matthews, A., Plasma spraying of capillary- porous coatings: experiments, modeling and applications, Int. Vacuum Congr. IVC-16/ICSS-12/NANO-8, Venice, Trento: Italian Vac. Soc., 2004, pp. 398–399.Google Scholar
- 9.Kalita, V.I., Bocharova, M.A., Gnedovets, A.G., Trushnicova, A.S., Yerokhin, A.L., Matthews, A., and Shaternicov, B.N., Structure and mechanical properties of novel plasma sprayed titanic capillary-porous for intrabone implants, Int. Vacuum Congr. IVC-16/ICSS-12/NANO-8, Venice, Trento: Italian Vac. Soc., 2004, pp. 58–59.Google Scholar
- 10.Kalita, V.I., Bocharova, M.A., Trushnikova, A.S., and Shaternikov, B.N., Surface structure of titanium materials intended for intraosseous implants, Russ. Metall. (Engl. Transl.), 2005, no. 3, pp. 282–288.Google Scholar
- 12.Van Oirschot B.A., Eman R.M., Habibovic P., Leeuwenburgh S.C., Weinans H., Alblas J., de Boer J. Comparing the osteophilicity of bone implant surface modifications in a cassette model on the decorticated goat spinal transverse process. Surface Modifications for Endosseous Implant Materials, 2015, pp. 21–43.Google Scholar
- 15.Rakngarm Nimkerdphol, A., Otsuka, Y., and Mutoh, Y., Effect of dissolution/precipitation on the residual stress redistribution of plasma-sprayed hydroxyapatite coating on titanium substrate in simulated body fluid (SBF), J. Mech. Behav. Biomed. Mater., 2014, vol. 36, pp. 98–108.CrossRefGoogle Scholar