A low-frequency power-type pressure-gradient receiver for oceanological investigations
- 21 Downloads
Technical solutions for the construction of a two-component power-type pressure-gradient receiver are developed. The manufactured prototype with dimensions of 111 × ∅52 mm provides an acousticpressure sensitivity in a plane wave of 60–70 μV/Pa at a frequency of 100 Hz. A mathematical model for an approximate calculation of the sensitivity to the acoustic pressure and optimizing the parameters of the structure was developed. The performance characteristics of the hydroacoustic pressure-gradient receivers, which provide oceanological investigations at frequencies that are substantially lower than 1 kHz, were improved.
Unable to display preview. Download preview PDF.
- 1.Gordienko, V.A., Vektorno-fazovye metody v akustike (Vector–Phase Methods in Acoustics), Moscow: FizMatLit, 2007, р.23.Google Scholar
- 3.Korenbaum, V.I., Tagil’tsev, A.A., Gorovoi, S.V., Kostiv, A.E., and Shiryaev, A.D., Instrtum. Exp. Tech., 2017 (in press).Google Scholar
- 4.Korenbaum, V.I., Tagil’tsev, A.A., Gorovoi, S.V., and Fershalov, Yu.Ya., RF Patent 2568411, Byull. Izobret., 2015, no.32.Google Scholar
- 5.Skrebnev, G.K., Kombinirofannye gidroakusticheskie priemniki (Combined Hydoacoustic Receivers), St.Petersburg: Elmor, 1997.Google Scholar
- 6.Bobber, R.J., Underwater Electroacoustic Measurements, Washington: US Governm., 1971.Google Scholar
- 9.Korenbaum, V.I., Gorovoy, S.V., Tagiltcev, A.A., and Kostiv, A.E., Proc. Meetings on Acoustics, 2014, vol. 21, p. 070007.Google Scholar
- 10.Gordienko, V.A., Gordienko, E.L., Dryndin, A.V., and Likhachev, S.M., Akust. Zh., 1994, vol. 40, no. 2, p. 243.Google Scholar