Fabrication of osteoconductive Ca3–x M2x (PO4)2 (M = Na, K) calcium phosphate bioceramics by stereolithographic 3D printing
- 61 Downloads
Osteoconductive ceramic implants based on Ca3–x M2x (PO4)2 (M = Na, K) double phosphates and having the Kelvin structure, tailored macropore size (in the range 50–750 μm), and a total porosity of 70–80% have been produced by stereolithographic 3D printing. We demonstrate that, to maintain the initial geometry of a model and reach sufficiently high strength characteristics of macroporous ceramics (compressive strength up to 9 MPa and fracture toughness up to 0.7 MPa m1/2), the polymer component should be removed under specially tailored heat treatment conditions. Based on our results on polymer matrix destruction kinetics, we have found heat treatment conditions that ensure a polymer removal rate no higher than 0.1 wt%/min and allow one to avoid implant cracking during the firing process.
Keywordsbioceramics double calcium phosphates 3D printing stereolithography osteoconductivity Kelvin structure heat treatment strength
Unable to display preview. Download preview PDF.
- 6.Safronova, T.V. and Putlyaev, V.I., Medical inorganic materials research in Russia: calcium phosphate materials, Nanosist.: Fiz., Khim., Mat., 2013, vol. 4, no. 1, pp. 24–47.Google Scholar
- 7.Znamierowska, T., Uklad Ca3(PO4)2–CaKPO4–CaNaPO4, Zesz. Nauk. Politech. Slask. Ser.: Chem., 1982, vol. 709 , pp. 35–56.Google Scholar
- 8.Evdokimov, P.V., Putlyaev, V.I., Ivanov, V.K., Garshev, A.V., Shatalova, T.B., Orlov, N.K., Klimashina, E.S., and Safronova, T.V., Phase equilibria in the tricalcium phosphate–mixed calcium sodium (potassium) phosphate systems, Russ. J. Inorg. Chem., 2014, vol. 59, no. 11, pp. 1219–1227.CrossRefGoogle Scholar
- 9.Evdokimov, P.V., Synthesis of Ca3–xM2x(PO4)2 (M = Na, K) double phosphates for macroporous bioceramics with specialized architectures, Cand. Sci. (Chem.) Dissertation, Moscow: Moscow State Univ., 2015.Google Scholar
- 12.Eggli, P.S., Mueller, W., and Schenk, R.K., Porous hydroxyapatite and tricalcium phosphate cylinders with two different pore size ranges implanted in the cancellous bone of rabbits, Clin. Orth. Rel. Res., 1988, vol. 232, pp. 127–137.Google Scholar
- 13.Lord Kelvin (Thomson, W.), On the division of space with minimum partitional area, Phil. Mag., Ser. 5, 1887, vol. 24, no. 151, pp. 503–514. doi 10.1080/14786448708628135Google Scholar
- 15.Evdokimov, P.V., Putlyaev, V.I., Ievlev, V.M., et al., Osteoconductive ceramics with a specified system of interconnected pores based on double calcium alkali metal phosphates, Dokl. Chem., 2015, vol. 460, no. 2, pp. 61–65.Google Scholar
- 16.Evans, A.G. and Charles, E.A., Fracture toughness determinations by indentation, J. Am. Ceram. Soc., 1976, vol. 59, nos. 7–8, pp. 371–372.Google Scholar
- 17.Putlyaev, V.I., Evdokimov, P.V., Garshev, A.V., et al., Strength characteristics of resorbable osteoconductive ceramics based on calcium alkali metal double phosphates, Izv. Vyssh. Uchebn. Zaved., Fiz., 2013, vol. 56, no. 10, pp. 72–77.Google Scholar