Neutron Fourier diffractometer FSD for residual stress studies in materials and industrial components
- 64 Downloads
Neutron diffraction study of residual stresses in materials became widely used in the world due to high penetrating power of neutrons. Therefore, to study residual stresses, the FSD (Fourier stress diffractometer) was developed at the IBR-2 reactor channel (Frank Laboratory of Neutron Physics, Joint Institute for Nuclear Research, Dubna, Russia), which, due to a special correlation technique (a fast Fourier chopper for modulating the primary neutron beam intensity and the RTOF method for data acquisition) makes it possible to obtain high-esolution diffraction spectra Δd/d = 4 × 10-3. This diffractometer was developed taking into account world experience in the study of residual stresses in materials; experience in the development of such devices in Russia and abroad was also used. The FSD diffractometer itself and its current state are described.
KeywordsSurface Investigation Neutron Technique Neutron Beam Gauge Volume Neutron Guide
Unable to display preview. Download preview PDF.
- 5.J. Schreiber, V. Richter, G. Bokuchava, et al., in Proc. of Eur. Conf. on Hard Materials and Diamond Tooling (Eur. Powder Metallurgy Association, 2002), p. 114.Google Scholar
- 6.V. L. Aksenov, A. M. Balagurov, G. D. Bokuchava, et al., in Proc. of the Nation. Conf. RSNE’97 (Dubna, 1997), Vol. 1, p. 69.Google Scholar
- 9.G. D. Bokuchava, V. L. Aksenov, A. M. Balagurov, et al., Appl. Phys. A: Mater. Sci. Process. 74(Suppl. 1), s86 (2002).Google Scholar
- 13.P. Hiismaki, V. A. Trounov, O. Antson, et al., in Proc. of the Conf. on Neutron Scattering in the Nineties (IAEA, Vienna, 1985), p. 453.Google Scholar
- 15.A. A. Bogdzel’, G. D. Bokuchava, V. A. Butenko, et al., Preprint No. R10-2004-21 (JINR, 2004).Google Scholar
- 18.Handbook on Construction Materials, Ed. by B. N. Arzamasov et al. (Mosc. Gos. Tekh. Univ., Moscow, 2005) [in Russian].Google Scholar