Performance Evaluation of a New Mobile POC Lipids Analyzer
- 11 Downloads
Hyperlipidemia, excessive amounts of fat and fatty substances in the blood, is a major risk factor for cardiovascular disease. The measuring of fat or fatty substance in blood is usually obtained through a blood lipid test which includes total cholesterol (TC), triglycerides (TG), low-density lipoprotein (LDL) and high-density lipoprotein (HDL). In the Medical Device industry, Point-of-care (POC) methods provide fast, portable and easy to use devices suitable for rapid determination of Lipid levels. Here, we evaluated the analytical performance of PixoTest POCT Lipid System (iXensor Co., Ltd.), a newly developed POC lipid panel analyzer.
Materials and Methods
PixoTest POCT Analyzer accuracy evaluation with the Cobas c111 Analyzer (Roche Diagnostics Ltd.) as a reference method was carried out according to CLSI EP 09-A3 guidelines. This study enrolled one hundred and twenty (120) subjects and conducted over three (3) clinical sites (40 subjects each), administered through nine (9) trained operators (total) performed under manufacturer’s protocol using three (3) reagent lots per blood sample (capillary & venous). Linear regression, bias plot and other statistical analyses were extrapolated from the test data in this study.
Passing-Bablok regression analysis of the data between PixoTest POCT System and Roche Cobas c111 Analyzer with the venous blood showed a slope of 1.03, an intercept of − 0.1 and correlation coefficient of 0.99. The results for capillary blood showed a slope of 1.01, an intercept of − 0.04 and correlation coefficient of 0.99. Passing-Bablok regression analysis of the data (n = 120) from the comparison between the PixoTest POCT System and Roche Cobas c111 Analyzer with the TC venous blood of total 3 lots at 3 sites showed a slope of 0.99, an intercept of − 0.337 and correlation coefficient of 0.995 and the TC capillary blood showed a slope of 0.99, an intercept of − 0.105 and correlation coefficient of 0.984. TG venous blood of total 3 lots at 3 sites showed a slope of 0.99, an intercept of 0.0246 and correlation coefficient of 0.998 and TG capillary blood showed a slope of 0.99, an intercept of − 0.329 and correlation coefficient of 0.998. HDL venous blood showed a slope of 0.99, an intercept of − 0.0297 and correlation coefficient of 0.987 and the HDL capillary blood showed a slope of 0.99, an intercept of − 0.0403 and correlation coefficient of 0.989.
The evaluation results of PixoTest POCT system’s accuracy, linearity, and correlation with Roche Cobas c111 Analyzer, the PixoTest POCT System displayed great performance results for all evaluated data points. Thus, the PixoTest POCT System is capable of providing accurate and reliable lipid measurements in the POC setting.
KeywordsPoint-of-care (POC) testing Lipids Total cholesterol (TC) Triglycerides (TG) High-density lipoprotein (HDL) PixoTest
- 1.World Health Organization. Health topics: Cardiovascular diseases. http://www.who.int/topics/cardiovascular_diseases/en/.
- 2.Scarborough, P., Wickramasinghe, K., Bhatnagar, P., et al. (2011). Trends in coronary heart disease 1961–2011. London: British Heart Foundation.Google Scholar
- 4.Frantz, I. D., Dawson, E. A., Ashman, P. L., Gatewood, L. C., Bartsch, G. E., Kuba, K., et al. (1989). Test of effect of lipid lowering by diet on cardiovascular risk. The Minnesota Coronary Survey. Arteriosclerosis, Thrombosis, and Vascular Biology, 9, 129–135.Google Scholar
- 6.Department of Health. (2009). Putting prevention first. NHS Health Check: Vascular risk assessment and management. Best practice guidance. London: Department of Health.Google Scholar
- 10.Panz, V. R., Raal, F. J., Paiker, J., Immelman, R., & Miles, H. (2005). Performance of the CardioChek PA and Cholestech LDX point-of-care analysers compared to clinical diagnostic laboratory methods for the measurement of lipids. Cardiovascular Journal South Africa, 16(2), 112–117.Google Scholar
- 11.Batki, A. D., Holder, R., Thomason, H. L., & Thorpe, G. H. G. Boehringer Mannheim Accutrend GC meter for the determination of blood glucose and cholesterol. Medical Devices Agency evaluation report 1995 MDA/95/11.Google Scholar
- 12.NHS Purchasing and Supply Agency. (2009). Buyer’s guide. Point-of-care testing for cholesterol measurement.Google Scholar
- 13.Expert Panel on Detection, Evaluation, and Treatment of High Blood Cholesterol in Adults. (2001). Executive summary of the Third Report of the National Cholesterol Education Program (NCEP) Expert Panel on Detection, Evaluation, and Treatment of High Blood Cholesterol in Adults (Adult Treatment Panel III). JAMA, 285(19), 2486–2497. https://doi.org/10.1001/jama.285.19.2486.CrossRefGoogle Scholar