Sterilization Method Using Plasma Discharge Against Biofilm-Producing Pseudomonas aeruginosa on Surface of Contact Lens
- 95 Downloads
Pseudomonas aeruginosa is the main causative bacteria of contact lens (CL)-associated microbial keratitis. Generally, multi-purpose solutions (MPSs) are widely used for washing, disinfection, and storage of CLs. However, the sterilization effect of an MPS against biofilm-producing P. aeruginosa is insufficient. The aim of the present study is to evaluate the bactericidal effect against biofilm-producing P. aeruginosa on CLs using plasma discharge. The standard strain of P. aeruginosa, PAO1, was used to make a biofilm-formed contamination model. The bactericidal effect of five MPSs and plasma discharge against biofilm-producing P. aeruginosa was observed using the culture method, scanning electron microscopy, and fluorescence microscopy. Plasma discharge was produced for 3, 5, or 10 min. Temperature, pH, and the concentrations of ozone (O3), hydrogen peroxide (H2O2), and nitrogen dioxide (NO2) in the discharged water were monitored. None of the tested MPSs showed a bactericidal effect after 4 h. In contrast, biofilm-producing P. aeruginosa strains were sterilized within 10 min using plasma discharge. Moreover, it was confirmed using fluorescence microscopy that there was no viable bacterium on the surface of the contamination model irradiated for 10 min. Although the concentration of O3 was 5 ppm after 3 min of discharge, it gradually decreased subsequently. Both NO2 and H2O2 concentrations gradually increased. Water temperature rose from 18 to 45 °C, and pH changed from 6 to 2. The results show that plasma discharge may be effective for the sterilization of biofilm-producing P. aeruginosa on CLs.
KeywordsPlasma Pseudomonas aeruginosa Contact lens
- 4.International Standards Organization (2001). ISO 14729 ophthalmic optics -contact lens care products- Microbiological requirements and test methods for products and regimens for hygienic management of contact lenses, Geneva, Switzerland.Google Scholar
- 7.Yang, C. Y., Lin, R. M., Wang, B. C., Lee, T. M., Chang, E., Hang, Y. S., & Chen, P. Q. (1997). In vitro and in vivo mechanical evaluations of plasma-sprayed hydroxyapatite coatings on titanium implants: The effect of coating characteristics. Journal of Biomedical Materials Research, 37, 335–345.zbMATHCrossRefGoogle Scholar
- 21.Admoni, M. M., Bartolomei, A., Qureshi, M. N., Bottone, E. J., & Asbell, P. A. (1994). Disinfection efficacy in an integrated ultraviolet light contact lens care system. CLAO Journal, 20, 246–248.Google Scholar