Surface Properties and In Vitro Bioactivity of Fluorapatite/TiO2 Coatings Deposited on Ti Substrates by Nd:YAG Laser Cladding
- 133 Downloads
Fluorapatite (FA)/TiO2 composite coatings were deposited on Ti–6Al–4V substrates with an Nd:YAG laser cladding process. Two TiO2 powder phases, namely anatase (A) and rutile (R), were used. After cladding, the FA/TiO2(R) specimen had a rougher surface morphology than that of FA/TiO2(A). Both coatings had a cellular-like main microstructure near the interface of the coating (CL) and transition layers (TL). However, a fine metallurgical bonding state was found that existed between CL and TL of the FA + TiO2(R) specimen. The X-ray diffraction (XRD) analysis results show that the coatings of both specimens were composed principally of FA, CaTiO3, and Al2O3 phases. With the high-energy–density laser cladding process, a portion of the FA in the original coating material remained, while all of the TiO2 powder was decomposed and reacted with the Ca in the FA (Ca-rich phase) to produce CaTiO3. Upon immersion of the clad specimens in simulated body fluid, apatite grew more rapidly on the FA/TiO2(R) coating than on the FA/TiO2(A) coating. The Ca/P ratio of the FA/TiO2(R) specimen approached the ideal bioactivity value after just 2 days of immersion. In contrast, that of the FA/TiO2(A) specimen did not reach the ideal value until 7 days of immersion. Furthermore, a peak corresponding to hydroxycarbonated apatite (HCA) appeared in the XRD patterns of both specimens. For the FA/TiO2(R) specimen, this HCA peak appears after a shorter immersion time. The FA/TiO2(R) specimen with a rougher surface morphology had better in vitro bioactivity than that of FA/TiO2(A).
KeywordsLaser cladding Fluorapatite TiO2 Ti–6Al–4V Hydroxycarbonated apatite
The authors gratefully acknowledge the financial support provided to this study by the Chimei Foundation Hospital, Republic of China (Taiwan), under Grant 110980191.
- 2.Jarcho, M. (1981). Calcium phosphate ceramics as hard tissue prosthetics. Clinical Orthopaedics and Related Research, 157, 259–278.Google Scholar
- 28.Kou, S. (1987). Basic concepts of solidification welding metallurgy (pp. 129–177). New York: John Wiley & Sons.Google Scholar
- 31.Yang, C. Y., Lee, T. M., Lui, Y. Z., Yang, C. W., Lui, T. S., Kuo, A., & Huang, B. W. (2010). The influence of plasma-sprayed parameters on the characteristics of fluorapatite coatings. Journal of Medical and Biological Engineering, 30, 91–98.Google Scholar
- 39.Matsusaka, S., Ohmori, A., & Toki, Y. (2001). Thermal spray 2001: New surfaces for a new millennium (pp. 87–92). Material Park, OH: ASM International.Google Scholar
- 46.Chien, C. S., Liao, T. Y., Hong, T. F., Kuo, T. Y., Chang, C. H., Yeh, M. L., & Lee, T. M. (2014). Surface microstructure and bioactivity of hydroxyapatite and fluorapatite coatings deposited on Ti–6Al–4V substrates using Nd-YAG laser. Journal of Medical and Biological Engineering, 34, 109–115.CrossRefGoogle Scholar
- 49.Zhou, P., & Akao, M. (1997). Preparation and characterization of double layered coating composed of hydroxyapatite and perovskite by thermal decomposition. BioMedical Materials and Engineering, 7, 67–81.Google Scholar
- 52.Denissen, H. W., de Nieuport, H. M., Kalk, W., Schaeken, H. G., & van de Hooff, A. (1991). Fluorapatite and hydroxyapatite heat-treated coatings for dental implants (pp. 130–140). Amsterdam: Elsevier Science.Google Scholar
- 53.Wolke, J., Klein, C., & de Groot, K. (1991). Bioceramics for maxillofacial applications. In Bioceramics for the human body (pp. 166–180). Amsterdam: Elsevier Science.Google Scholar
- 56.Doi, Y. (1997). Sintered carbonate apatites as bone substitutes. Cells Materials, 7, 111–122.Google Scholar