This paper suggests a new optical system for measuring shock wave characteristic that is installed in a shock tube. This shock tube has an innovative pneumatic rapid opening valve designed to open from the center of the tube to the corners. In this measurement system, a plate has placed in front of the shock wave. The plate displacement has measured during a collision by measuring the amount of light reflectance change. Then these optical data have converted to a voltage. After calibration with the help of a piezoresistive sensor in the shock tube and showing its voltage-pressure relation, the shock wave pressure diagram has been demonstrated, which contains the wave’s positive and negative parts. Also, the delay time in this system is investigated. In addition to the shock wave velocity, the system can measure the timing and velocity opening of the pneumatic valve replaced in the shock tube. The velocity is also calculated theoretically and compared with the experimental results. The unique feature of this automatic valve is its opening from the center of the tube to the corners, like a camera aperture, which is best suited for opening the valve and resulting wave uniformity. Also, the alignment of the driver and the driven prevented gas perturbation due to the redirection or rotation. This problem has seen in most recently used valves. Finally, the planar shock wave formation has been investigated.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Stewart JB, Pecora C (2015) Explosively driven air blast in a conical shock tube. Rev Sci Instrum 86(3):035108. https://doi.org/10.1063/1.4914898
Wisniewiski D (2011) Second generation shock tube calibration system. Int J Acoust Vib 17:133–138. https://doi.org/10.20855/ijav.2012.17.3304
Campbell MF, Tulgestke AM, Davidson DF, Hanson RK (2014) A second-generation constrained reaction volume shock tube. Rev Sci Instrum 85:055108. https://doi.org/10.1063/1.4875056
Tranter RS, Giri BR (2008) A diaphragmless shock tube for high temperature kinetic studies. Rev Sci Instrum 79:094103. https://doi.org/10.1063/1.2976671
Shi JC, Shang YL, Ye W, Zhang RT, Luo SN (2018) Shock-tube experiments and chemical kinetic modeling study of CH4 sensitized by CH3NHCH3. Energy Fuels 32:5588–5595. https://doi.org/10.1021/acs.energyfuels.8b00860
Perkowski W, Irzycki A, Kawalec M, Łukasik B, Snopkiewicz K (2015) Measurement of pressure in front of shock wave-assessment of methodology influence on the measurement results on the basis of experiments with the shock tube. J Kones Powertrain Transp 20. https://doi.org/10.5604/12314005.1137844
Varas JM, Philippens M, Meijer SR, van den Berg AC, Sibma PC, van Bree JLMJ, de Vries DVWM (2011) Physics of IED blast shock tube simulations for mTBI research. Front Neurol 2:58. https://doi.org/10.3389/fneur.2011.00058
Andreotti R, Colombo M, Guardone A, Martinelli P, Riganti G, Di Prisco M (2015) Performance of a shock tube facility for impact response of structures. Int J Non Linear Mech 72:0020–7462. https://doi.org/10.1016/j.ijnonlinmec.2015.02.010
Courtney E, Courtney A, Courtney M (2014) Shock tube design for high intensity blast waves for laboratory testing of armor and combat materiel. Def Techno 10. https://doi.org/10.1016/j.dt.2014.04.003
Hanson RK, Davidson DF (2014) Recent advances in laser absorption and shock tube methods for studies of combustion chemistry. Prog Energy Combust Sci 44. https://doi.org/10.1016/j.pecs.2014.05.001
Davis WC, Salyer TR, Jackson SI, Aslam TD (2006) Explosive-driven shock waves in argon. In Proceedings of the 13th International Detonation Symposium, IDS 2006, pp 1035–1044
Duff RE, Blackwell AN (1966) Explosive driven shock tubes. Rev Sci Instrum 37:579–586. https://doi.org/10.1063/1.1720256
Courtney AC, Andrusiv LP, Courtney MW (2012) Oxy-acetylene driven laboratory scale shock tubes for studying blast wave effects. Rev Sci Instrum 83:045111. https://doi.org/10.1063/1.3702803
Stotz I, Lamanna G, Hettrich H, Weigand B, Steelant J (2008) Design of a double diaphragm shock tube for fluid disintegration studies. Rev Sci Instrum 79:125106. https://doi.org/10.1063/1.3058609
Colombo M, di Prisco M, Martinelli P (2011) A new shock tube facility for tunnel safety. Exp Mech 51:1143–1154. https://doi.org/10.1007/s11340-010-9430-7
Tranter RS, Lynch PT (2013) A miniature high repetition rate shock tube. Rev Sci Instrum 84:094102. https://doi.org/10.1063/1.4820917
Neupane S et al (2018) Shock tube/laser absorption and kinetic modeling study of Triethyl phosphate combustion. J Phys Chem A 122(15):3829–3836
Hornig DF (1963) Shock waves in chemistry and physics. J Am Chem Soc 85:1554–1554. https://doi.org/10.1021/ja00893a051
Tranter RS, Brezinsky K, Fulle D (2001) Design of a high-pressure single pulse shock tube for chemical kinetic investigations. Rev Sci Instrum 72:3046–3054. https://doi.org/10.1063/1.1379963
Abe A, Sugahara K, Yamada Y (2015) Rapid opening valve assisted by magnetic force for a Diaphragmless shock tube. In: Bonazza R, Ranjan D (eds.) 29th international symposium on shock waves 1. Springer International Publishing, Cham, pp 337–342
Hariharan MS, Janardhanraj S, Saravanan S, Jagadeesh G (2011) Diaphragmless shock wave generators for industrial applications of shock waves. Shock Waves 21:301–306. https://doi.org/10.1007/s00193-010-0296-5
Oude Nijeweme DJ, Kok JBW, Stone CR, Wyszynski L (2001) Unsteady in-cylinder heat transfer in a spark ignition engine: experiments and modelling. Proc Inst Mech Eng Part D J Automob Eng 215:747–760. https://doi.org/10.1243/0954407011528329
Udagawa S, Ota M, Maeno K (2012) Propagation characteristics of the shock wave in small diameter tubes at atmospheric initial driven pressure. pp 529–534. https://doi.org/10.1007/978-3-642-25688-2_81
Heufer KA, Olivier H, Drumm S, Murrenhoff H (2012) A new fast acting valve for Diaphragmless shock tubes. In: Kontis K (ed.) 28th international symposium on shock waves. Springer, Berlin, Heidelberg, pp 535–540. https://doi.org/10.1007/978-3-642-25688-2_82
Sivaraman I (2012) Introduction to hydraulics and pneumatics
Oliver MR, Spooncer RC, Ghezelayagh MH (1986) An autoreferenced two-state optical fibre reflective sensor. In: Baker LR (ed.) International Society for Optics and Photonics. SPIE, Bellingham, pp 233–238. https://doi.org/10.1117/12.963612
Declaration of Interests
The authors declare that they have no known competing financial interests or personal relationships that could have appeared to influence the work reported in this paper.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Samimi, M., Zamani, J. & Sardarzadeh, F. Shock Wave Pressure and Velocity Measuring Using a Novel Optic Sensor in a Newly Designed Diaphragm-less Shock Tube. Exp Tech 45, 55–66 (2021). https://doi.org/10.1007/s40799-020-00409-1
- Shock wave
- Shock tube
- Rapid opening valve
- Optical measurement system