Grating-based X-ray dark-field imaging: a new paradigm in radiography
- 1.8k Downloads
Grating-based X-ray dark-field contrast is an emerging new imaging modality that is demonstrating particularly high potential for radiography. The signal in dark-field X-ray imaging is determined by small-angle X-ray scattering at structures typically below the spatial resolution of the imaging setup. Thus, this technique not only yields complementary information but also visualizes information that lies under the resolution limit for conventional, absorption-based radiography. Grating-based X-ray dark-field imaging has been shown to be feasible with both synchrotron radiation and conventional X-ray tubes. Lung, breast, and bone imaging have been identified as the applications promising the main impact, but other applications are on the horizon. Specifically, dark-field radiography has been used to detect pulmonary emphysema and assesses its regional distribution in mice and holds promise to improve the visualization of micro-calcifications in mammography and yields information about bone microstructure. Further technical developments are required to make the technique suitable for clinical use.
KeywordsX-ray phase-contrast imaging X-ray dark-field imaging Grating-based imaging Grating interferometry, Lung imaging Pulmonary emphysema
Clinical X-ray imaging, so far, depends solely on the X-ray absorbing properties of tissues and materials. Since the X-ray attenuation coefficient is roughly proportional to the fourth power of the atomic number Z  tissues composed of heavier atoms (e.g., bones and teeth) show a high contrast on conventional X-ray images. Soft tissue, mainly composed of light elements, yields only low contrast. X-ray phase-sensitive imaging methods have the potential to not only overcome the limitation of low soft tissue contrast but also visualize biophysical tissue properties that are inaccessible for conventional radiography. These imaging methods, similar to the concepts known from light microscopy , use the wave nature of X-rays. Passing through matter, X-rays not only get absorbed but are also refracted, i.e., are deviated from their original propagation direction by a certain angle. Measurement of the deviation angle yields complementary information in comparison to the conventional attenuation-based imaging. Different approaches make it possible to acquire X-ray phase information [3, 4, 5, 6, 7] and have been in use at large scale synchrotron facilities since the first groundbreaking experiments reported by Bonse and Hart in 1965 . However, only in 2006, the Talbot-Lau X-ray interferometer has been reported [9, 10] which allows to acquire X-ray phase information with conventional polychromatic laboratory sources. Since then, considerable efforts have been devoted to applying this method to different biomedical applications. The present review presents an overview over the physical principles of dark-field radiography as well as recent results of experiments applying the technique to biomedical imaging.
Physical and Technical Principles of Grating-Based X-Ray Dark-Field Imaging
Currently, the most spread method for obtaining X-ray phase information with conventional laboratory sources is based on the use of a three-grating Talbot-Lau interferometer. This method utilizes three gratings between the X-ray source and detector. The first (absorption) grating, a so-called source grating, is placed as close as possible to the X-rays source emission point and divides the source into a number of line sources, thus creating spatial coherence in the beam. A certain distance downstream of the source grating a phase grating is placed. It gives rise to an interference pattern (i.e., typically a combination of dark and light stripes) further downstream. This interference pattern is registered with and without the sample in the beam, and deviations in the pattern reveal phase information. However, since the interference pattern is typically too small to be directly resolved by clinically used detectors, the third grating (a so-called analyzer grating) is placed just in front of the detector. This last absorption grating is moved in a number of steps perpendicular to the beam propagation direction and grating lamellae over one grating period, acquiring a detector image for each new grating position. Thus, a so-called intensity stepping curve  is acquired, which gives the registered intensity in a pixel as a function of the analyzer grating position. This way, information about the interference pattern can be obtained, without directly resolving it. By comparing the stepping curves acquired with and without the sample, it is possible to calculate conventional transmission, differential phase-contrast, and dark-field images if the stepping was performed with at least three steps . Thus, three image contrasts can be obtained simultaneously, and the images can be directly compared pixel by pixel.
In particular, conventional transmission is obtained as the mean value of the stepping curve performed with the sample, normalized by the corresponding value acquired without the sample. Differential phase-contrast is calculated as the shift of the stepping curve, acquired with the sample, compared to the reference curve. And the dark-field contrast is obtained as the amplitude of the stepping curve with the sample divided by the amplitude without the sample and normalized by the transmission signal [9, 10, 11].
The three imaging signals originate from different physical processes. While conventional transmission reveals the amount of photons absorbed in the sample, the differential phase-contrast gives the first derivative of the wave phase shift caused by the sample. If the phase shift is too small to be directly resolved by the grating interferometer, a dark-field signal is generated. The origin of the dark-field can be considered as small-angle scattering, which leads to a damping of the amplitude of the stepping curve. Hence, the dark-field can visualize structures that are smaller than the resolution of the system. This way, for example, the alveoli in the lung can be clearly visualized in dark-field radiography.
To avoid the differential depiction, the phase shift information can be derived from the differential phase-contrast signal by integration. Different methods [13, 14] for noise reduction integration algorithms have been reported both for projection and tomographic acquisitions. However, their application to radiographs remains challenging.
A further grating-based approach used to retrieve phase information with conventional polychromatic laboratory sources is based on the introduction of a periodic illumination pattern on the detector that can be directly resolved. Typically, a simple absorption grating is illuminated to produce such a pattern [15, 16]. Alternatively, the interference pattern can be achieved by introducing Moiré fringes on the detector through slightly tilted gratings or a different structure [17•, 18].
To retrieve the three imaging modalities, a radiographic image is acquired with and without the sample in the beam. The images are subsequently considered in Fourier space. Due to the introduced pattern on the detector, two distinct peaks are obtained in Fourier space, corresponding to the 0th and the 1st Fourier component. A mask is applied in Fourier space to take the 0th order or the 1st order peak, respectively. Subsequently, the information is transformed back to the real space. This approach allows obtaining transmission as well as phase-contrast and dark-field images. Although this method is relatively easy to realize experimentally, it suffers from significantly lower sensitivity than the approach with the three-grating interferometer.
Pulmonary emphysema is commonly associated with chronic obstructive lung disease and thus most commonly seen in patients with a history of smoking. If detected early, the progression of pulmonary emphysema can be delayed by smoking cessation. Interventional techniques such as bronchial valves or lung volume reduction surgery can relieve symptoms by removing or decreasing ventilation to lung parenchyma with advanced destruction. Lung transplantation remains the only curative treatment, but this procedure is associated with high mortality and morbidity and is thus only suitable for a minority of patients with end-stage pulmonary emphysema.
The diagnosis and severity assessment of pulmonary emphysema on chest radiographs is challenging, particularly in the early stages of the disease. Conventional chest radiographs can only visualize indirect signs of the increased pulmonary volume seen in emphysema such as a flattened diaphragm, widely spaced ribs as well as increased chest diameter and increased retrosternal air space on the lateral view. Consequently, chest radiograms have been shown to be reasonably accurate for advanced emphysema but only moderately sensitive in mild to moderate emphysema with substantial inter-observer variability [23, 24, 25, 26].
A pneumothorax occurs when air enters into the pleural space leading to a partial or complete collapse of the lung on the affected side. On conventional chest radiographs, small pneumothoraces can easily be overlooked. Due to the low density of lung tissue, the difference in attenuation between air and lung is minimal. Frequently, the pneumothorax can only be recognized due to the absence of vascular markings and the fine pleural line separating lung and pneumothorax. On dark-field radiographs, the presence and extent of a pneumothorax can be easily recognized by the sharp contrast between the absent dark-field signal in the pneumothorax and the strong dark-field signal of the lung. Weber and colleagues have explored this potential application in their study, where bilateral pneumothoraces were incidentally detected in one of the mice imaged post mortem .
Other Pulmonary Pathologies
To date, no other pulmonary pathologies have been investigated with grating-based dark-field radiography. Many other pulmonary pathologies such as pulmonary infections and fibrosing interstitial lung disease likely interfere with the scattering characteristics of the affected parenchyma and could thus be detected on dark-field chest radiographs. Whether or not dark-field radiography can provide a diagnostic benefit for the detection, and characterization of these pathologies remains to be investigated.
Mammography has been demonstrated to reduce breast cancer mortality  although the precise strength of this benefit and its relationship with potential harms of breast cancer screening such as overtreatment and psychological distress remains controversial . Nevertheless, annual or biennial mammography for breast cancer screening is generally recommended for women aged 50–69 years . The detection of breast cancer on mammograms is challenging with a high rate of false-positive findings prompting unnecessary biopsies.
Schleede and colleagues imaged a mammography phantom with grating-based imaging at a compact synchrotron light source and reported an increase in image contrast attainable through phase-contrast and dark-field imaging over conventional attenuation-based projections . Stampanoni and colleagues first used grating-based radiography to image full-breast human mastectomy samples bearing breast cancers [39••]. They found that the phase-contrast and dark-field signal can be used to create color-coded fusion images that enhance tumor visibility. The authors subsequently determined the value of their approach in a larger study using fresh mastectomy samples from 33 patients and six breast radiologists as readers . The results of this study suggested that the addition of phase-contrast and dark-field information to absorption images improved the overall image sharpness, allowed for better delineation of breast lesions and increased the visibility of calcifications.
Anton and colleagues imaged human mastectomy samples and detected an area of increased dark-field signal, which corresponded to a tumor region containing small calcification grains . A subsequent publication by the same research group confirmed that micro-calcifications with a grain size of 1–40 μm were responsible for the changes in dark-field signal observed in dark-field breast radiography . Interestingly, both tumors imaged in this study were invisible on conventional radiography, but one of them could be detected on the dark-field image due to the signal alterations from micro-calcifications within the tumor.
These findings may be clinically relevant, since ductal carcinoma in situ (a precursor lesion which may progress to invasive breast cancer) can often only be detected on mammography due to microcalcifications. However, no reliable data exist whether or not dark-field mammography can differentiate microcalcifications of benign and malignant etiology.
Several groups have made some effort to evaluate the benefit of dark-field radiography for bone imaging. The main advantage of x-ray dark-field imaging for bone applications is seen in the fact that dark-field imaging can reveal information about substructures of the bone that are not directly resolved on the transmission image. Thus, Wen and colleagues reported images of mineralized collagen fibrils and porosities of the mineralized matrix . Further studies demonstrated that X-ray dark-field radiography can yield information about bone micro-architecture, without the need to directly resolve the structures [43, 44, 45]. It was confirmed using high resolution CT scans that the directional dark-field signal originates from X-ray scattering on trabecular micro-architecture in bone. A further recent publication by Thüring and colleagues presented a radiograph of a human hand cadaver specimen  acquired with a grating-based setup. In their study, the dark-field image was visually rated as superior to the absorption image for the delineation of bone and the depiction of the joint spaces and provided superior visualization of periarticular soft tissue calcifications.
Emerging Biomedical Applications
Based on the very promising results in lung imaging, it has been hypothesized that microbubbles, which are typically used as a contrast agent for ultrasound imaging, can be used to produce contrast in dark-field radiography. A first proof-of-principle study by Velroyen et al.  demonstrated that considerable signal can indeed be generated. However, further studies demonstrating similar results in situ and in vivo are needed to confirm the utility of microbubbles as contrast agents for dark-field radiography. This approach may allow for performing angiographic examinations without iodinated contrast agents.
Challenges and Future Perspectives
Toward Dark-Field X-ray Tomography
Dark-field tomography can be performed very similar to the conventional absorption CT . However, there is a significant difference in information that is obtained. Conventional X-ray transmission tomography yields the distribution of the linear attenuation coefficient, i.e., how much radiation is locally absorbed by the sample. Similar to the attenuation coefficient of a material, a linear diffusion coefficient is introduced for dark-field imaging. The diffusion coefficient is a material dependent variable describing the specific scattering width (angle distribution) per unit length of the sample . Thus, similar to the transmission CT, the dark-field tomography gives the distribution of the diffusion coefficient, i.e., the local scattering power. For example, a dark-field CT of the lung would not visualize lung density but rather the frequency of air/tissue interfaces as sources of small-angle X-ray scattering. This means that alveoli do not have to be directly resolved, to retrieve information about their size. However, dark-field tomography has not yet been performed in vivo, and it remains to be demonstrated whether it is feasible with a radiation dose and acquisition time suitable for the in vivo setting.
Challenges for the Clinical Application of X-ray Dark-Field Imaging
X-ray dark-field imaging is still in the preclinical phase of development and has not been used in living humans. The transition of this technique for the clinical use in patients will require a number of further technical developments. First, the size of currently available gratings limits the field of view that can be imaged with a single acquisition. In the laboratory environment, larger specimens are often imaged by combining images derived from multiple acquisitions, but this requires acquisition times that are impractical for clinical use. Thus, larger gratings have to be developed to provide a field of view that is sufficient for clinical radiography applications. Second, the currently available technical equipment for grating-based X-ray imaging is still in a prototype stage. Therefore, the radiation dose associated with grating-based radiography is still substantially higher than would be acceptable for clinical use. Further technical developments will thus need to focus on optimizing the dose efficiency of grating-based X-ray imaging setups. As the third challenge, the preclinical studies on dark-field radiography have been performed with relatively low photon energies in the order of 20–40 keV. This photon energy may be compatible with clinical requirements for mammography, where only few centimeters of tissue need to be penetrated, and low photon energies are beneficial due to their higher soft tissue contrast . However, for most other clinical applications, the technique will need to be adapted to considerably higher photon energies. This is certainly the case for lung imaging, where the X-ray beam has to penetrate the human chest. As the first step in this direction, the feasibility of X-ray phase-contrast imaging at 82 keV has recently been demonstrated [49•].
X-ray dark-field is an emerging imaging modality that has shown a high potential for substantially expanding the diagnostic information that can be gained from radiographs. During the past years, a number of applications have been identified that could benefit considerably from this new imaging modality. Among the most promising are lung imaging, mammography, and bone imaging. Even if the progress in the field of X-ray phase-contrast and dark-field imaging has recently been considerable, no studies have thus far been performed on living humans. Numerous technical challenges still have to be overcome before this imaging modality can be introduced into the clinical world.
We acknowledge financial support through the DFG Cluster of Excellence Munich-Centre for Advanced Photonics (MAP), the DFG Gottfried Wilhelm Leibniz program, and the European Research Council (ERC, FP7, StG 240142). AY acknowledges the support of his studies through the TU München graduate school.
Papers of particular interest, published recently, have been highlighted as • Of importance •• Of major importance
- 2.Murphy DB. Fundamentals of light microscopy and electronic imaging. New York: Wiley; 2001.Google Scholar
- 17.• Momose A, Yashiro W, Harasse S, Kuwabara H. Four-dimensional X-ray phase tomography with Talbot interferometry and white synchrotron radiation: dynamic observation of a living worm. Opt Express. 2011;19(9):8423–8432. This publication demonstrates the feasibility to acquire phase information with very short exposure times.Google Scholar
- 19.•• Schleede S, Meinel FG, Bech M, et al. Emphysema diagnosis using X-ray dark-field imaging at a laser-driven compact synchrotron light source. Proc Natl Acad Sci USA. 2012;109(44):17880–17885. This publication demonstrated for the first time that grating-based X-ray imaging can be used to detect pulmonary emphysema and map its regional distribution on radiographs.Google Scholar
- 21.Weber T, Bayer F, Haas W, et al. Investigation of the signature of lung tissue in X-ray grating-based phase-contrast imaging. 2012. https://archive.org/details/arxiv-1212.5031. Accessed 5 Jan 2014.
- 22.•• Bech M, Tapfer A, Velroyen A, et al. In-vivo dark-field and phase-contrast x-ray imaging. Sci Rep. 2013;3:3209. This was the first study to perform grating-based dark-field radiography in vivo.Google Scholar
- 27.• Meinel FG, Schwab F, Schleede S, et al. Diagnosing and mapping pulmonary emphysema on X-ray projection images: incremental value of grating-based X-ray dark-field imaging. PLoS One. 2013;8(3):e59526. This analysis demonstrated for the first time that grating-based dark-field radiography is superior to conventional radiography for the detection of pulmonary emphysema and the assessment of its regional distribution.Google Scholar
- 28.•• Yaroshenko A, Meinel FG, Bech M, et al. Pulmonary emphysema diagnosis with a preclinical small-animal X-ray dark-field scatter-contrast scanner. Radiology. 2013;269(2):427–433. This publication demonstrated that dark-field radiography for pulmonary emphysema diagnosis can be performed using a conventional polychromatic X-ray source.Google Scholar
- 30.•• Meinel FG, Yaroshenko A, Hellbach K, et al. Improved diagnosis of pulmonary emphysema using in vivo dark-field radiography. Invest Radiol. 2014;49. This study demonstrated for the first time that dark-field radiography can detect pulmonary emphysema in vivo.Google Scholar
- 32.• Meinel FG, Schwab F, Yaroshenko A, et al. Lung tumors on multimodal radiographs derived from grating-based X-ray imaging—A feasibility study. Phys Med. 2013;30(3):352–357. First report on imaging pulmonary nodules with grating-based phase-contrast and dark-field radiography.Google Scholar
- 33.Nelson HD, Tyne K, Naik A, et al. Screening for breast cancer: an update for the U.S. preventive services task force. Ann Intern Med. 2009;151(10):727–737, W237–742.Google Scholar
- 34.Gotzsche PC, Jorgensen KJ. Screening for breast cancer with mammography. Cochrane Database Syst Rev. 2013;6:CD001877.Google Scholar
- 39.•• Stampanoni M, Wang Z, Thuring T, et al. The first analysis and clinical evaluation of native breast tissue using differential phase-contrast mammography. Invest Radiol. 2011;46(12):801–806. Study demonstrating the incremental diagnostic value of phase-contrast and dark-field imaging in mammography.Google Scholar
- 44.Schaff F, Malecki A, Potdevin G, et al. Correlation of x-ray vector radiography to bone micro-architecture. Nat Sci Rep. 2013;3694:3695.Google Scholar
- 49.• Willner M, Bech M, Herzen J, et al. Quantitative X-ray phase-contrast computed tomography at 82 keV. Opt Express. 2013;21(4):4155–4166. Study demonstrating the feasibility of phase-contrast and dark-field imaging at clinically relevant energies.Google Scholar
- 50.Auweter SD, Herzen J, Willner M, et al. X-ray phase-contrast imaging of the breast–advances towards clinical implementation. Br J Radiol. 2014;87(1034):20130606.Google Scholar