Ammonium Ion Binding to DNA G-Quadruplexes: Do Electrospray Mass Spectra Faithfully Reflect the Solution-Phase Species?
- First Online:
- Cite this article as:
- Balthasart, F., Plavec, J. & Gabelica, V. J. Am. Soc. Mass Spectrom. (2013) 24: 1. doi:10.1007/s13361-012-0499-3
- 1.4k Downloads
Key wordsG-quadruplex DNA Noncovalent complex Ion mobility spectrometry Gas-phase structure
Nucleic acid sequences containing series of adjacent guanines can form particular nucleic acid structures called G-quadruplexes [1, 2, 3], in which four guanines can form G-quartets. Sufficiently high ionic strength is required to favor the self-assembly of negatively charged strands. Moreover, the very nature of the cation also influences the G-quadruplex structure and stability [4, 5] because specific cation coordination occurs between the stacked G-quartets. G-quadruplex structures can differ by their strand orientation (parallel/antiparallel), by the syn/anti conformation of each base relative to the deoxyribose, and by the placement of their loops. These three structural characteristics are inter-dependent .
Like many other cation-bound supramolecular assemblies , G-quadruplexes can be studied by mass spectrometry (for a recent review, see ), with the restriction that solutions in which a physiological ionic strength (150 mM monovalent cations) would be ensured solely by Na+ or K+ cannot be used. Ammonium cations can be used instead , as NH4+ ions also coordinate between G-quartets and thereby stabilize G-quadruplexes in solution [10, 11, 12]. When sufficient internal energy is provided to the biomolecule, neutral ammonia is lost following proton transfer to the biomolecule. Mass spectrometry analysis of G-quadruplexes from ammonium solutions therefore entails the following tradeoff. On the one hand, it is desirable to preserve these internal ammonium cations in the gas phase because (1) detecting them is a strong indication that G-quadruplex structures existed in solution , and (2) preserving them is thought to be essential to preserve the structure in the gas phase [13, 14, 15]. On the other hand, it is desirable to get rid of ammonium ions bound to less specific sites like the phosphate groups, in order to obtain cleaner mass spectra and higher sensitivity.
In parallel G-quadruplex structures, the internal ammonium ions are much less labile in the gas phase than the other ammonium ions, so mass spectral peaks with the specific number of internal ammonium ions can be detected over a wide range of instrumental tuning settings [9, 16, 17, 18]. In antiparallel structures, internal ammonium ions are usually more labile than in parallel structures [9, 18], but the relative lability of inter-quartet versus externally bound ammonium ions has not been investigated systematically. One previous MS/MS study focused on base loss and strand separation , which are dissociation steps subsequent to ammonium loss.
Our objective is to address the following questions in the case of antiparallel G-quadruplexes produced from ammonium acetate solutions. To what extent do the mass spectra reflect the binding constants of ammonium ions in solution? What is the optimum instrumental tuning to determine the number of specific binding sites using the ammonium ion distribution detected in the mass spectra? Would similar conclusions be reached by different users on different instruments? What is the influence of ammonium ion loss on the gas-phase structure?
Ion mobility calibration was carried out using oligonucleotides of known collision cross section (CCS), as described previously (supporting information of ). The prediction error (the CCS interval in which one has 95 % chances of finding the true value using the calibration curve) is 1.6 %, and the confidence error (the CCS interval in which repeated measurements of molecules belonging to the dataset will be found) is 0.3 %. The prediction error must be considered when comparing our experiments with experiments carried out on a different instrument or with calculations, and the confidence error must be considered when comparing different systems measured on the same instrument. The CCS distributions were fitted using PeakFit 4.11. The instrumental ion mobility mobility resolution in the concerned m/z range is 22.5, determined as Ω/ΔΩFWHM of the G-quadruplex [(dTG4T)4●(NH4)3]5–. This parallel G-quadruplex was chosen because it is very rigid [13, 32, 33], and its CCS peak displays a resolution of 75 on the Synapt G2 HDMS.
3 Results and Discussion
3.1 Electrospray Mass Spectrometers Are Not Equally Suited to Preserve Specific Inner Ammonium Ions
The Solarix ESI-FTICRMS and the Synapt G1 provided the softest experimental conditions (see Figure 2a for a typical spectrum). The two other Q-TOF type instruments were found harsher than the Synapt G1, but for different reasons. With the particular source of the Q-TOF Ultima Global mass spectrometer (Figure 2b), the 3-NH4/2-NH4 ratio is similar to that obtained on the ESI-FTICRMS (Figure 2a), but simultaneously, a peak with zero ammonium ion adducts is also observed. Therefore, a major fraction of the ions is produced in soft conditions, while a small fraction of the ions acquires a significantly higher internal energy, and the whole ESI-MS ammonium ion binding distribution does not reflect the solution phase distribution. On the Synapt G2 HDMS, ammonium loss was more abundant in the ion mobility mode than on all other mass spectrometers.
3.2 Collision-Induced Dissociation Occurs Inside the Traveling Wave Ion Mobility Cell
In addition to ion heating before the IMS cell [34, 35], ion heating could also occur inside traveling wave IMS cells, as reported for small singly charged ions [35, 36]. However, as the extent of ion heating is supposed to decrease when the size of the system increases, increase when the charge increases, and decrease when the cell pressure increases, whether detectable ion heating inside the IMS cell would be observed for larger, multiply charged ions on a Synapt G2 mass spectrometer remained unknown. For [(dG4T4G4)2●(NH4)3]5–, lowering the IMS wave height from 40 to 25 V (Figure 2c and d, respectively), or increasing the IMS wave velocity from 1000 to 1600 m/s (see Supporting Information S2) leads to better preservation of the three ammonium ions, confirming that ion heating is partly occurring inside the IMS cell.
3.3 In Some G-quadruplexes, Inner Ammonium Ions Are Too Labile to be Fully Retained in the Gas Phase Structures
3.4 ESI-MS Spectra Suggest Additional Specific Ammonium Ion Binding Sites
In some spectra, additional ammonium ions compared with the number trapped between the G-quartets can also be detected. The question is whether these extra ammonium ions are located at specific binding sites or rather bound to the phosphodiester backbones of the strands. Specific binding sites would have higher binding strength in solution than ammonium counter-ion binding to the phosphate groups. If this higher binding strength in solution translates into a higher kinetic stability in the gas phase, the mass spectra could show a bias in the ammonium ion distribution towards a particular stoichiometry higher than the number of cations trapped between G-quartets. Note that ammonium ions in these external binding sites are invisible to NMR, either because of the short residence times, or because their partial hydration is making them equivalent to bulk ammonium ions.
3.4.1 Thrombin Binding Aptamer (TBA)
The TBA intramolecular G-quadruplex (Figure 3d) differs from the quadruplexes reported above. The peak corresponding to two ammonium ions bound is the base peak for charge state 3-. The existence of a second cation binding site for the TBA in solution is currently controversial [30, 37, 38, 39]. Our mass spectrometry results clearly show the preservation of an additional specific ammonium cation by the TBA in addition to the one likely bound between the G-quartets, suggesting that a fraction of the population in solution possessed this additional cation. Interestingly, molecular dynamics simulation of cation (Na+ and K+) movement in the TBA in solution showed that cations can be transiently captured between the loops before replacing the central cation [40, 41]. Similar results were found in preliminary simulations made on TBA in the presence of ammonium cations in solution [A.V. Golovin, personal communication]. However, the change in ammonium ion distribution according to the charge state prevents the quantification of this species by MS. The 2:1 complex is very labile in the gas phase (see below), even for the lowest charge state, so it is also possible that the 2:1 complex is partially dissociated in the gas phase even in the softest possible conditions.
3.4.2 Sequences Containing Diagonal Loops
In dG4(T4G4)3 (Figure 3c), one additional ammonium ion is detected for charge state 6-, but the lack of bias towards a specific stoichiometry prevents attributing it unambiguously to a specific binding site. In contrast, in (dG3T4G4)2 (Figure 3b), the peak with a total of three ammonium ions preserved is clearly more abundant than the peak with four ammonium ions. Note that for sensitivity reasons, the spectrum in Figure 3b was recorded with a drying gas temperature of 150 °C instead of 20 °C. The biased distribution towards the 3-NH4 peak is even more clearly observed on the Q-TOF and on the Synapt G1 instruments (Supporting Information S4). The mass spectra, therefore, hint at the presence of one additional specific binding site. Finally, the G-quadruplex [(dG4T4G4)2]5– (Figure 2) a peak with five ammonium ions is retained, hinting at the presence of two such additional specific binding sites. Interestingly, X-ray crystallography studies showed that in addition to binding between G-quartets, K+ cations can also bind to the loops of (dG4T4G4)2 . As this structure with two equivalent diagonal loops takes up two additional ammonium ions, whereas G-quadruplexes (dG3T4G4)2 and dG4(T4G4)3 have only one diagonal loop each, we propose that outer G-quartets are additional ammonium ion binding sites and that the bases of the diagonal loops participate in the cation binding. Guanines in the loops (G-quadruplexes (dG3T4G4)2 and TBA) also favor additional ammonium ion binding. These results also recall the capture of one additional ammonium ion between the outer G-quartets and heterocyclic ligands like telomestatin [43, 44]. ESI-MS is the only experimental method capable of probing these outer-quartet specific binding sites directly, and will be very useful to reveal the existence of such binding sites in new systems. However, the results will require further validation by molecular dynamics simulations or by ion titration experiments.
3.5 All Inner Ammonium Ions Are Necessary to Preserve the G-Quadruplex Fold in the Gas Phase
Collision cross sections and CCS peak resolution for the intact G-quadruplexes containing all inner ammonium ions
802 ± 3
801.9 ± 0.6
803.4 ± 0.4
740.2 ± 1.2
746.7 ± 0.7
753.2 ± 1.3
740 ± 3
747.3 ± 1.7
762 ± 3
896.6 ± 0.6
915.4 ± 1.2
940 ± 5
565.9 ± 1.0
583 ± 2
624.6 ± 1.7
For each bimolecular G-quadruplex, the loss of a single inner ammonium ion significantly changes the collision cross section. In general, in the lowest charge state the ammonium ion loss induces a slight crunch of the structure, whereas in the highest charge states it induces a dramatic expansion of the structure. For the most abundant charge state (5- for all dimers presented here), the direction of the change varies with the sequence. Moreover, the absence of change in CCS does not necessarily mean the absence of change in gas-phase structure. For example, for [(dG4T4G4)2]5– (Figure 4a), the first ammonium ion loss induces an expansion of the structure, then the two subsequent ammonium ion losses induce consecutive crunches of the structure. Incidentally, the 1-NH4 CCS is equal to the 3-NH4 CCS. Therefore, to ensure preserving the G-quadruplex structure from the solution to the gas phase, the species with all specific inner ammonium ions and at relatively low charge state must be selected (even when this species is particularly labile, such as the 3-NH4 species for (dG4T3G4)2, see Figure 4b.
In the case of intramolecular G-quadruplexes (Supporting Information Figure S6-2), for the two lowest charge states (including the most abundant charge state), the collision cross section does virtually not change upon ammonium ion loss. Ammonium loss involves proton transfer of the ammonium ions to the DNA and, hence, the repartition of the charges over the DNA structure at total constant charge changes upon ammonia loss. This apparently is not always accompanied by a change of collision cross section for the lowest charge states, presumably because other salt bridge interactions in the denatured state overcome the Coulombic repulsion and the denatured state remains compact.
3.6 Ion Mobility Spectrometry Helps to Assign the Number of Inner Ammonium Ions
On the contrary to inner ammoniums, additional ammonium ions are not essential to preserve the gas-phase structure [compare the CCS of the 3-NH4 and 2-NH4 species in (dG4T3G4)2, Figure 4b, where the three inner ammonium ions are essential, with (dG4T3G4)2, Figure 4c, where only two inner ammonium ions are essential]. Additional (i.e., outer-G-quartet) ammonium ions cause little changes of CCS [see for example the 3-NH4 species for (dG3T4G4)2 in Figure 4c]. At the highest charge state, the mean CCS and the peak width of the intact complex increase with the bias voltage, but in lower proportion than when ammonium ion loss occurs. This suggests that the solution structure is preserved in the gas phase when all inner ammonium ions are preserved. Moreover, the most significant change in CCS is observed for the highest charge state (compare Figure 4c and d). IMS experiments on the high charge state may not be ideal to derive structure from the CCS, but give the best indication that a specific number of ammonium ions are involved in the formation of a specific structure like a G-quadruplex. The number of ammonium ions needed to keep the CCS of the high charge state ions close to the CCS of the lower charge state ions is equal to the number of inner ammonium ions in the solution structure.
Instrument choice is crucial. A good system to test instrumental conditions is the (dG4T4G4)25– bimolecular G-quadruplex, for which soft conditions should show the 3-NH4 peak (all inner ammonium ions preserved) as the base peak. The 5-NH4 peak (with two additional ammonium ions presumably bound between the external G-quartets and the diagonal loops) should also be visible.
The gas-phase lability should be tested on a range of relative internal energies, and only clear biases in the ammonium ion distribution can be interpreted as an indication of specific ammonium ion binding. This helps assigning the number of binding sites in solution. With regard to determination of ammonium ion equilibrium binding constants, when some lability is evident at conditions close to the softest achievable, the ammonium ion distribution will likely not reflect the solution-phase distribution.
Ion mobility spectrometry helps assigning the number of inner ammonium ions as the minimum number of ammonium ions needed to obtain a near-constant CCS at all charge states. Indeed, all inner ammonium ions are necessary and sufficient to preserve the structure from the solution to the gas phase, for the most abundant charge state produced from solutions at physiological ionic strength (150 mM NH4OAc).
This work was supported by the Fonds de la Recherche Scientifique-FNRS (research associate position and FRFC grant 2.4528.11 to V.G.), the EU COST action MP0802, the Slovenian Research Agency [ARRS, P1-0242 and J1-4020], EU FP7 projects with acronyms EAST-NMR  and Bio-NMR , and the Cooperation Agreement between Wallonia-Brussels and the Republic of Slovenia.