Impact of Clinically Relevant Elliptical Deformations on the Damage Patterns of Sagging and Stretched Leaflets in a Bioprosthetic Heart Valve
- 125 Downloads
After implantation of a transcatheter bioprosthetic heart valve its original circular circumference may become distorted, which can lead to changes in leaflet coaptation and leaflets that are stretched or sagging. This may lead to early structural deterioration of the valve as seen in some explanted transcatheter heart valves. Our in vitro study evaluates the effect of leaflet deformations seen in elliptical configurations on the damage patterns of the leaflets, with circular valve deformation as the control. Bovine pericardial tissue heart valves were subjected to accelerated wear testing under both circular (N = 2) and elliptical (N = 4) configurations. The elliptical configurations were created by placing the valve inside custom-made elliptical holders, which caused the leaflets to sag or stretch. The hydrodynamic performance of the valves was monitored and high resolution images were acquired to evaluate leaflet damage patterns over time. In the elliptically deformed valves, sagging leaflets experienced more damage from wear compared to stretched leaflets; the undistorted leaflets of the circular valves experienced the least leaflet damage. Free-edge thinning and tearing were the primary modes of damage in the sagging leaflets. Belly region thinning was seen in the undistorted and stretched leaflets. Leaflet and fabric tears at the commissures were seen in all valve configurations. Free-edge tearing and commissure tears were the leading cause of valve hydrodynamic incompetence. Our study shows that mechanical wear affects heart valve pericardial leaflets differently based on whether they are undistorted, stretched, or sagging in a valve configuration. Sagging leaflets are more likely to be subjected to free-edge tear than stretched or undistorted leaflets. Reducing leaflet stress at the free edge of non-circular valve configurations should be an important factor to consider in the design and/or deployment of transcatheter bioprosthetic heart valves to improve their long-term performance.
KeywordsBioprosthetic heart valve Pericardial tissue Leaflet durability Non-circular deformation Valve fatigue
Belly region thinning
Accelerated wear testing
Transcatheter heart valve
This work was supported by the FDA’s Office of Women’s Health grants and in part by an appointment to the ORISE Research Participation Program at the FDA/CDRH, administered by the Oak Ridge Institute for Science and Education through an interagency agreement between the U.S. Department of Energy and FDA/CDRH. We are thankful to Edwards Lifesciences, for assisting with purchase of the surgical valves for this study. We appreciate help from our colleagues, Jon Casamento, Terry Woods, and Shiril Sivan, and our interns Lena Karkar, Nick Lane, and Robyn Hall.
The mention of commercial products, their sources, or their use in connection with materials reported herein is not to be construed as either an actual or implied endorsement of such products by the Department of Health and Human Services. This material is declared a work of the U.S. Government and is not subject to copyright protection in the United Stated. Approved for public release; distribution is unlimited.
Conflict of interest
There are no conflicts of interest.
Statement of Human and Animal Studies
- 6.Caudron, J., J. Fares, C. Hauville, A. Cribier, J.-N. Dacher, C. Tron, F. Bauer, P.-Y. Litzler, J.-P. Bessou, and H. Eltchaninoff. Evaluation of multislice computed tomography early after transcatheter aortic valve implantation with the Edwards SAPIEN bioprosthesis. Am. J. Cardiol. 108(6):873–881, 2011. https://doi.org/10.1016/j.amjcard.2011.05.014.CrossRefGoogle Scholar
- 8.Delgado, V., A. C. T. Ng, N. R. van de Veire, F. van der Kley, J. D. Schuijf, L. F. Tops, A. de Weger, et al. Transcatheter aortic valve implantation: role of multi-detector row computed tomography to evaluate prosthesis positioning and deployment in relation to valve function. Eur. Heart J. 2010. https://doi.org/10.1093/eurheartj/ehq018.Google Scholar
- 9.Duraiswamy, N., J. D. Weaver, Y. Ekrami, S. M. Retta, and W. Changfu. A parametric computational study of the impact of non-circular configurations on bioprosthetic heart valve leaflet deformations and stresses: possible implications for transcatheter heart valves. Cardiovasc. Eng. Technol. 7(2):126–138, 2016. https://doi.org/10.1007/s13239-016-0259-9.CrossRefGoogle Scholar
- 13.Haziza, F., G. Papouin, B. Barratt-Boyes, G. Christie, and R. Whitlock. Tears in bioprosthetic heart valve leaflets without calcific degeneration. J. Heart Valve Dis. 5(1):35–39, 1996.Google Scholar
- 15.Hilbert, S. L., V. J. Ferrans, H. A. McAllister, and D. A. Cooley. Ionescu-Shiley bovine pericardial bioprostheses. Histologic and ultrastructural studies. Am. J. Pathol. 140(5):1195–1204, 1992.Google Scholar
- 16.Ishihara, T., V. J. Ferrans, S. W. Boyce, M. Jones, and W. C. Roberts. Structure and classification of cuspal tears and perforations in porcine bioprosthetic cardiac valves implanted in patients. Am. J. Cardiol. 48(4):665–678, 1981. https://doi.org/10.1016/0002-9149(81)90145-4.CrossRefGoogle Scholar
- 22.Lee, M. C., Y. C. Fung, R. Shabetai, and M. M. LeWinter. Biaxial mechanical properties of human pericardium and canine comparisons. Am. J. Physiol. 253(1 Pt 2):H75–H82, 1987.Google Scholar
- 25.Midha, P. A., V. Raghav, J. F. Condado, I. U. Okafor, S. Lerakis, V. H. Thourani, V. Babaliaros, and A. P. Yoganathan. Valve type, size, and deployment location affect hemodynamics in an in vitro valve-in-valve model. JACC 9(15):1618–1628, 2016. https://doi.org/10.1016/j.jcin.2016.05.030.Google Scholar
- 27.Sacks, M. S. The biomechanical effects of fatigue on the porcine bioprosthetic heart valve. J. Long Term Eff. Med. Implants 2001. https://doi.org/10.1615/JLongTermEffMedImplants.v11.i34.100.Google Scholar
- 34.Schultz, C. J., A. Weustink, N. Piazza, A. Otten, N. Mollet, G. Krestin, R. J. van Geuns, P. de Feyter, Patrick W. J. Serruys, and P. de Jaegere. Geometry and degree of apposition of the corevalve revalving system with multislice computed tomography after implantation in patients with aortic stenosis. J. Am. Coll. Cardiol. 54(10):911–918, 2009. https://doi.org/10.1016/j.jacc.2009.04.075.CrossRefGoogle Scholar
- 36.Shabetai, R. The Pericardium. Berlin: Springer, 2012.Google Scholar
- 37.Souteyrand, G., K. Wilczek, A. Innorta, L. Camilleri, P. Chodor, J.-R. Lusson, P. Motreff, J.-C. Laborde, P. Chabrot, and N. Durel. Distortion of the corevalve during transcatheter aortic valve-in-valve implantation due to valve dislocation. Cardiovasc. Revasc. Med. 14(5):294–298, 2013. https://doi.org/10.1016/j.carrev.2013.05.004.CrossRefGoogle Scholar
- 43.Vesely, I., J. E. Barber, and N. B. Ratliff. Tissue damage and calcification may be independent mechanisms of bioprosthetic heart valve failure. J. Heart Valve Dis. 10(4):471–477, 2001.Google Scholar
- 46.Ye, J., A. Cheung, M. Yamashita, D. Wood, D. Peng, M. Gao, C. R. Thompson, et al. Transcatheter aortic and mitral valve-in-valve implantation for failed surgical bioprosthetic valves: an 8-year single-center experience. JACC 8(13):1735–1744, 2015. https://doi.org/10.1016/j.jcin.2015.08.012.Google Scholar