An Experimental and Computational Study on the Effect of Caval Valved Stent Oversizing
- 183 Downloads
Heterotopic implantation of transcatheter tricuspid valve is a new treatment option for tricuspid regurgitation. Transcatheter tricuspid valves are implanted onto the cavoatrial junction in order to avoid the challenging task of anchoring the valve onto the complex tricuspid valve annulus. However, little is known about optimum extent of oversizing of the valved stent in a vena cava. In this study, we implanted valves of the same diameter onto the larger sized inferior vena cava (IVC) and a smaller sized superior vena cava (SVC). The valve in the IVC was oversized by 10.7% while the valve in the SVC was oversized by 21.6%. Finite element analysis was performed (i) to assess the strain on the nitinol stent during manufacturing and deployment; (ii) the stents were deployed in a patient-specific vena cava model and the intramural stress of the vena cava was calculated computationally. These valves were fabricated and placed in a silicone model of a patient-specific right atrium which was part of a mock circulatory system that emulated the patho-physiological flow rate and pressure of a patient with tricuspid regurgitation. Flow measurements were conducted by particle image velocimetry (PIV). It was found that the maximum crimping strain on the nitinol stent was 6.85% which was lower than the critical threshold of 10%. The maximum stress on the vena cava was located at the spot where the hooks met the wall. The maximum stress on the IVC was 0.5098 MPa while the maximum stress on the SVC was 0.7 MPa. The maximum Reynolds shear stress (mRSS) in the vena cava was found to be higher in the IVC than SVC with the highest mRSS being 1741 dynes/cm2 found in the region of high flow during the peak flow phase. The overtly oversized valve in the SVC did not cause flow disturbances and exhibited mostly laminar flows. The mRSS at the downstream of the vena cava valve and the middle of the atrium remained at low magnitudes. However, velocity fluctuations were high in the IVC in all the time points measured. In conclusion, oversizing the valve may assist anchorage; yet, careful consideration should be taken in choosing the extent of oversizing as it may lead to adverse effects.
KeywordsTranscatheter tricuspid valve Caval valves Hemodynamics Stent oversizing Finite element analysis Particle image velocimetry
- 6.Debusschere, N., et al. A finite element strategy to investigate the free expansion behaviour of a biodegradable polymeric stent (1873-2380 (Electronic)).Google Scholar
- 8.Ellis, J. T., B. R. Travis, and A. P. Yoganathan. An in vitro study of the hinge and near-field forward flow dynamics of the St. Jude Medical Regent bileaflet mechanical heart valve. Ann. Biomed. Eng. 2000(0090-6964 (Print)).Google Scholar
- 9.Famaey, N., et al. Arterial clamping: finite element simulation and in vivo validation (1878-0180 (Electronic)).Google Scholar
- 11.Hansen, D. C. Metal corrosion in the human body: the ultimate bio-corrosion scenario. Electrochem. Soc. Interface 17:31, 2008.Google Scholar
- 12.Higdon, A. Mechanics of materials (4th ed.). New York: Wiley, 1985; (xvii, 744 p.).Google Scholar
- 14.Hsiao, H. ., et al. Effects of through-hole drug reservoirs on key clinical attributes for drug-eluting depot stent (1873-4030 (Electronic)).Google Scholar
- 15.Ismail, M., et al. In vitro investigation of the hemodynamics of transcatheter heterotopic valves implantation in the cavo-atrial junction. doi:10.1111/aor.12457 (1525-1594 (Electronic)).
- 21.Mortier, P., et al. Finite element analysis of side branch access during bifurcation stenting (1873-4030 (Electronic)).Google Scholar
- 22.Ragkousis, G. E., N. Curzen, and N. W. Bressloff. Simulation of longitudinal stent deformation in a patient-specific coronary artery (1873-4030 (Electronic)).Google Scholar
- 24.Schiavone, A., L. G. Zhao, and A. A. Abdel-Wahab. Effects of material, coating, design and plaque composition on stent deployment inside a stenotic artery—finite element simulation (1873-0191 (Electronic)).Google Scholar
- 26.Sobin, P. B. Mechanical Properties of Human Vein. New York: Springer, 1977.Google Scholar
- 27.Systèmes, D. Best practices for stent analysis with ABAQUS, 2015. Available from: www.3ds.com.
- 28.Vulev, I., et al. Endovascular treatment of internal carotid and vertebral artery aneurysms using a novel pericardium covered stent. Interv. Neuroradiol. 18(2):164–171, 2012.Google Scholar