Fabrication and Evaluation of Nitinol Thin Film Heart Valves
- 300 Downloads
In recent years new innovations improved the technology of devices for transcatheter heart valve replacement. In order to fabricate an efficient, durable, less thrombogenic and very thin heart valved stent for transcatheter implantation, heart valve leaflets of Nitinol (NiTi) thin film were developed. This work presents a method to fabricate NiTi thin film leaflets for transcatheter aortic valve replacements. NiTi valve leaflets were fabricated in different film thicknesses (10, 15 and 20 µm) by magnetron sputter deposition and a subsequent forming process. Pulsatile testing was performed to investigate the transvalvular pressure differences during systole and diastole as well as the effective orifice area of the valves. Furthermore, the steady backflow leakage rate under hydrostatic pressure was evaluated. Two samples from Edwards Lifescience (SAV 19 mm, and Perimount Magna 25 mm) were taken as a reference. In this experimental study, we present first in vitro test results of the newly developed NiTi leaflets.
KeywordsHeart valves Transcatheter aortic valve implantation Prosthetic valves Shape memory alloy Nitinol
This work was supported by the German Research Foundation (QU146/11-1) (LU 663/11-1).
Conflict of Interest
Authors Klaas Loger, Alexander Engel and Martin Marczynski-Bühlow declare that they have no conflict of interest. Author Rodrigo Lima de Miranda is partner of the Acquandas GmbH (Kiel) and holds several patents on shape memory thin films techniques. Author Eckhard Quandt reports funding of the project by the German Research Foundation (DFG). He is partner of the Acquandas GmbH (Kiel) and holds several patents on shape memory thin films techniques. Author Georg Lutter reports funding of the project by the German Research Foundation (DFG).
- 2.Carabello, B. A., and W. Grossman. Calculation of stenotic valve orifice area. In: Grossman’s Cardiac Catheterization, Angiography and Intervention6th, edited by D. S. Baim, and W. Grossman. Philadelphia: Lippincott Williams & Wilkins, 2000.Google Scholar
- 3.Cribier, A., E. Eltchaninoff, A. Bash, N. Borenstein, C. Tron, F. Bauer, G. Derumeaux, F. Anselme, F. Laborde, and M. B. Leon. Percutaneous transcatheter implantation of an aortic valve prosthesis for calcific aortic stenosis: first human case description. Circulation 106(24):3006–3008, 2002.CrossRefGoogle Scholar
- 5.Lima de Miranda, R., C. Zamponi, and E. Quandt. Micropatterned freestanding superelastic TiNi films. Adv. Eng. Mater. 15(1–2):66–69, 2013.Google Scholar
- 8.Habijan, T., R. Lima de Miranda, C. Zamponi, E. Quandt, C. Greulich, T. A. Schildhauer, and M. Köller. The biocompatibility and mechanical properties of cylindrical NiTi thin films produced by magnetron sputtering. Mater. Sci. Eng. C 32(8):2523–2528, 2012.Google Scholar
- 9.International Standard ISO 5840. Cardiovascular Implants—Cardiac Valve Prostheses. ISO, 2005.Google Scholar
- 12.Marquez, S., R. T. Hon, and A. Yonganathan. Comparative hydrodynamic evaluation of bioprosthetic heart valves. J. Heart Valve Dis. 10(6):802–811, 2001.Google Scholar
- 15.Ryhänen, J. Biocompatibility Evolution of Nickel-Titanium Shape Memory Alloy. Oulu, Finland: Academic Dissertation, Faculty of Medicine, University of Oulu, 1999.Google Scholar
- 16.Shabalovskaya, S. A. Surface, corrosion and biocompatibility aspects of nitinol as an implant material. Bio-Med. Mater. Eng. 12(1):69–109, 2002.Google Scholar