Non-contact measurement of elastic modulus is necessary for the in-line assessment of material in harsh environments such as high temperature. In this paper, a fully non-contact method to measure elastic modulus is proposed based on the laser ultrasonic technique (LUT) that uses a short-pulsed laser to generate ultrasound and the other laser coupled to an interferometer using a photorefractive crystal to detect the ultrasonic wave displacement. Basically, this method measures velocities of shear wave and longitudinal wave to obtain the elastic modulus. The uniqueness is that the velocity of shear wave is measured in the thermo-elastic regime first and then the velocity of longitudinal wave is measured in the ablation regime. This is because the strong mode of generated ultrasound is the shear wave in the thermo-elastic regime while the longitudinal wave in the ablation regime. Regime change can be achieved simply by switching the laser power, with no change in the measurement setup. In order to demonstrate the usefulness of the proposed method, the elastic modulus of aluminum casting alloy has been measured and the results were compared with a conventional contact method and a destructive tensile test. They showed good agreement with each other, which verified the usefulness of the proposed noncontact elastic modulus measurement method.
This is a preview of subscription content, access via your institution.
Buy single article
Instant access to the full article PDF.
Tax calculation will be finalised during checkout.
Li, P., Hao, J., Zhao, J., and Duan, H., “The influence of Ageing Treatment on the Microstructure and the Elastic Modulus of Ti27Nb8Zr Alloy,” Materials Science and Engineering: A, Vol. 527, No. 29, pp. 7469–7474, 2010.
Olympus, “Ultrasonic Transducers,” http://www.olympus-ims.com/ko/ultrasonic-transducers/ (Accessed 23 MAR 2015)
Scruby, C. B. and Drain, L. E., “Laser Ultrasonics Techniques and Applications,” CRC Press, pp. 223–324, 1990.
Davies, S. J., Edwards, C., Taylor, G., S. and Palmer, S. B., “Laser- Generated Ultrasound: Its Properties, Mechanisms and Multifarious Applications,” Journal of Physics D: Applied Physics, Vol. 26, No. 3, pp. 329–348, 1993.
Lee, H., Chung, C., Kim, C. S., and Jhang, K. Y., “Fully Non- Contact Assessment of Acoustic Nonlinearity according to Plastic Deformation in Al6061 Alloy,” Journal of the Korean Society for Nondestructive Testing, Vol. 32, No. 4, pp. 388–392, 2012.
Aussel, J. D. and Monchalin, J. P., “Precision Laser-Ultrasonic Velocity Measurement and Elastic Constant Determination,” Ultrasonics, Vol. 27, No. 3, pp. 165–177, 1989.
Li, Z. Q., Zhang, X. R., Zhang, S. Y., and Shen, Z. H., “Determination of the Elastic Constants of Metal-Matrix Composites by a Laser Ultrasound Technique,” Composites Science and Technology, Vol. 61, No. 10, pp. 1457–1463, 2001.
Sun, G., Zhou, Z., Chen, X., and Wang, J., “Ultrasonic Characterization of Delamination in Aeronautical Composites using Noncontact Laser Generation and Detection,” Applied Optics, Vol. 52, No. 26, pp. 6481–6486, 2013.
Monchalin, J. P. and Aussel, J. D., “Ultrasonic Velocity and Attenuation Determination by Laser-Ultrasonics,” Journal of Nondestructive Evaluation, Vol. 9, No. 4, pp. 211–221, 1990.
Wild, G. and Hinckley, S., “Acousto-Ultrasonic Optical Fiber Sensors: Overview and State-of-the-Art,” Sensors, Vol. 8, No. 7, pp. 1184–1193, 2008.
Monchalin, J., “Non Contact Generation and Detection of Ultrasound with Lasers,” Proc. of the 16th World Conference on Nondestructive Testing, 2004.
About this article
Cite this article
Kim, J., Jhang, KY. Non-contact measurement of elastic modulus by using laser ultrasound. Int. J. Precis. Eng. Manuf. 16, 905–909 (2015). https://doi.org/10.1007/s12541-015-0118-x
- Elastic modulus
- Laser ultrasound