Study of complexation of aluminium with cinnamoyl derivative in the presence of fluoride ions using an optical probe: automated determination of fluoride
- 42 Downloads
In this work, the complexation of aluminium (III) with 3-[4-(dimethylamino)cinnamoyl]-4-hydroxy-6-methyl-3,4-2H-pyran-2-one (ligand) in the presence of fluoride using an optical probe for absorbance measurement was studied. It was adjusted that the complex was formed in the molar ratio (Al–Lig)3 to F 1:1. The reaction was applied to determine of fluoride. The optimum conditions were achieved at pH 8 with concentrations of Al3+ and ligand 6.75 and 0.195 mg L−1, respectively. Under optimized conditions, the calibration curve showed good linearity (r = –0.9976) in the range 0.23–1.9 mg L−1 of fluoride. The limit of detection of fluoride was found to be 0.02 mg L−1. The proposed method was applied for determination of fluoride concentration in mineral water samples and fully automated by sequential injection system.
KeywordsFluoride 3-[4-(dimethylamino)cinnamoyl]-4-hydroxy-6-methyl-3,4-2H-pyran-2-one Probes Sequential injection analysis
A.D. would like to say a special thanks to the International Visegrad Fund for providing a 10-month scholarship.
Compliance with ethical standards
Conflict of interest
Authors have declared no conflict of interest.
Human and animal rights statement
This article does not contain any studies with human or animal subjects.
- Bayón MM, Rodrígues Garsia A, García Alonso JI, Sanz-Medel A (1999) Indirect determination of trace amounts of fluoride in natural waters by ion chromatography: a comparison of on-line post-column fluorimetry and ICP-MS detectors. Analyst 124:27–31. https://doi.org/10.1039/A807079B CrossRefGoogle Scholar
- Gleisner H, Welz B, Einax JW (2010) Optimization of fluorine determination via the molecular absorption of gallium mono-fluoride in a graphite furnace using a high-resolution continuum source spectrometer. Spectrochim Acta B 65(9–10):864–869. https://doi.org/10.1016/j.sab.2010.08.003 CrossRefGoogle Scholar
- Miyake Y, Yamashita N, So MK, Rostkowski P, Taniyasu S, Lam PKS, Kannan K (2007) Trace analysis of total fluorine in human blood using combustion ion chromatography for fluorine: a mass balance approach for the determination of known and unknown organofluorine compounds. J Chromatogr A 1154:214–221. https://doi.org/10.1016/j.chroma.2007.03.084 CrossRefGoogle Scholar
- Morés S, Monteiro GC, Santos Fda S, Carasek E, Welz B (2011) Determination of fluorine in tea using high-resolution molecular absorption spectrometry with electrothermal vaporization of the calcium mono-fluoride CaF. Talanta 85:2681–2685. https://doi.org/10.1016/j.talanta.2011.08.044 CrossRefGoogle Scholar
- Ozbek N, Akman S (2012) Method development for the determination of fluorine in toothpaste via molecular absorption of aluminum mono fluoride using a high-resolution continuum source nitrous oxide/acetylene flame atomic absorption spectrophotometer. Talanta 94:246–250. https://doi.org/10.1016/j.talanta.2012.03.034 CrossRefGoogle Scholar
- Pochivalov A, Vakh C, Andruch V, Moskvin L, Bulatov A (2017) Automated alkaline-induced salting-out homogeneous liquid–liquid extraction coupled with in-line organic-phase detection by an optical probe for the determination of diclofenac. Talanta 169:159–162. https://doi.org/10.1016/j.talanta.2017.03.074 CrossRefGoogle Scholar
- Zaruba S, Vishnikin AB, Škrliková J, Diuzheva A, Ozimaničova I, Gavazov K, Andruch V (2017) A two-in-one device for online monitoring of direct immersion single-drop microextraction: an optical probe as both microdrop holder and measuring cell. RSD Adv 7:29421–29427. https://doi.org/10.1039/C7RA02326J CrossRefGoogle Scholar