The goal of this study was to evaluate the safety and the impact on weight loss of a new swallowable gastric balloon. In this prospective pilot study, 17 overweight or obese patients were included. Up to three balloons were ingested under fluoroscopic control. All balloons were removed by upper GI endoscopy, 12 weeks after the ingestion of the first balloon. 43 out of 44 attempts (98 %) to swallow a balloon were successful. Nausea and stomach pain were the most frequent side effects. Endoscopic procedures for balloon removal were uneventful. Weight loss was significant at weeks 4, 8, and 12. This pilot study showed no significant side effects induced by up to three balloons, and a significant weight loss.
Intragastric balloon Human Obesity
This is a preview of subscription content, log in to check access.
The study was sponsored by Obalon Therapeutics Inc. (Carlsbad, CA, USA). FM, TP, JD received honoraria for the study.
Conflict of Interest
The other authors declared no conflict of interest.
Genco A, Bruni T, Doldi SB, et al. BioEnterics intragastric balloon: the Italian experience with 2,515 patients. Obes Surg. 2005;15:1161–4.PubMedCrossRefGoogle Scholar
Roman S, Napoleon B, Mion F, et al. Intragastric balloon for "non-morbid" obesity: a retrospective evaluation of tolerance and efficacy. Obes Surg. 2004;14:539–44.PubMedCrossRefGoogle Scholar
Dumonceau JM. Evidence-based review of the Bioenterics intragastric balloon for weight loss. Obes Surg. 2008;18:1611–7.PubMedCrossRefGoogle Scholar
Genco A, Balducci S, Bacci V, et al. Intragastric balloon or diet alone? A retrospective evaluation. Obes Surg. 2008;18:989–92.PubMedCrossRefGoogle Scholar
Sacks FM, Bray GA, Carey VJ, et al. Comparison of weight-loss diets with different compositions of fat, protein, and carbohydrates. N Engl J Med. 2009;360:859–73.PubMedCrossRefGoogle Scholar
Messina T, Genco A, Favaro R, et al. Intragastric balloon positioning and removal: sedation or general anesthesia? Surg Endosc. 2011;25:3811–4.PubMedCrossRefGoogle Scholar
Machytka E, Klvana P, Kornbluth A, et al. Adjustable intragastric balloons: a 12-month pilot trial in endoscopic weight loss management. Obes Surg. 2011;21:1499–507.PubMedCrossRefGoogle Scholar
Mion F, Gincul R, Roman S, et al. Tolerance and efficacy of an air-filled balloon in non-morbidly obese patients: results of a prospective multicenter study. Obes Surg. 2007;17:764–9.PubMedCrossRefGoogle Scholar
Lecumberri E, Krekshi W, Matia P, et al. Effectiveness and safety of air-filled balloon Heliosphere BAG® in 82 consecutive obese patients. Obes Surg. 2011;21:1508–12.PubMedCrossRefGoogle Scholar
Ofman JJ, Rabeneck L. The effectiveness of endoscopy in the management of dyspepsia: a qualitative systematic review. Am J Med. 1999;106:335–46.PubMedCrossRefGoogle Scholar
Peker Y, Durak E, Ozgurbuz U. Intragastric balloon treatment for obesity: prospective single-center study findings. Obes Facts. 2010;3:105–8.PubMedCrossRefGoogle Scholar
Dastis NS, Francois E, Deviere J, et al. Intragastric balloon for weight loss: results in 100 individuals followed for at least 2.5 years. Endoscopy. 2009;41:575–80.PubMedCrossRefGoogle Scholar
Martin CK, Bellanger DE, Rau KK, Coulon S, Greenway FL. Safety of the Ullorex oral intragastric balloon for the treatment of obesity. J Diabetes Sci Technol. 2007;1:574–81.PubMedGoogle Scholar