Utility of second-generation single-energy metal artifact reduction in helical lung computed tomography for patients with pulmonary arteriovenous malformation after coil embolization
- 151 Downloads
The quality of images acquired using single-energy metal artifact reduction (SEMAR) on helical lung computed tomography (CT) in patients with pulmonary arteriovenous malformation (PAVM) after coil embolization was retrospectively evaluated.
Materials and methods
CT images were reconstructed with and without SEMAR. Twenty-seven lesions [20 patients (2 males, 18 females), mean age 61.2 ± 11.0 years; number of embolization coils, 9.8 ± 5.0] on contrast-enhanced CT and 18 lesions of non-enhanced lung CT concurrently performed were evaluated. Regions of interest were positioned around the coils and mean standard deviation value was compared as noise index. Two radiologists visually evaluated metallic coil artifacts using a four-point scale: 4 = minimal; 3 = mild; 2 = strong; 1 = extensive.
Noise index was significantly improved with SEMAR versus without SEMAR (median [interquartile range]; 194.4 [161.6–211.9] Hounsfield units [HU] vs. 243.9 [220.4–286.0] HU; p < 0.001). Visual score was significantly improved with SEMAR versus without SEMAR (Reader 1, 3  vs.1 ; Reader 2, 3  vs.1 ; p < 0.001). Significant differences were similarly demonstrated on lung CT (p < 0.001).
SEMAR provided clear chest CT images in patients who underwent PAVM coil embolization.
KeywordsArtifact reduction Metal implants CT Pulmonary arteriovenous malformation Single-energy metal artifact reduction algorithm
Compliance with ethical standards
Conflict of interest
The authors declare no conflicts of interest or funding sources to disclose.
This study was approved by the ethics committee of the authors’ institution, and requirements for informed consent were waived. Information from this study was presented on the institutional website, and all patients were informed of their right to opt out.
- 9.Yasaka K, Kamiya K, Irie R, Maeda E, Sato J, Ohtomo K. Metal artefact reduction for patients with metallic dental fillings in helical neck computed tomography: comparison of adaptive iterative dose reduction 3D (AIDR 3D), forward-projected model-based iterative reconstruction solution (FIRST) and AIDR 3D with single-energy metal artefact reduction (SEMAR). Dentomaxillofac Radiol. 2016. https://doi.org/10.1259/dmfr.20160114.PubMedPubMedCentralGoogle Scholar
- 10.Sonoda A, Nitta N, Ushio N, Nagatani Y, Okumura N, Otani H, et al. Evaluation of the quality of CT images acquired with the single energy metal artifact reduction (SEMAR) algorithm in patients with hip and dental prostheses and aneurysm embolization coils. Jpn J Radiol. 2015;33:710–6.CrossRefPubMedGoogle Scholar
- 11.Gondim Teixeira PA, Meyer JB, Baumann C, Raymond A, Sirveaux F, Coudane H, et al. Total hip prosthesis CT with single energy projection-based metallic artifact reduction: impact on the visualization of specific periprosthetic soft tissue structures. Skelet Radiol. 2014;43:1237–46.CrossRefGoogle Scholar
- 13.Kidoh M, Utsunomiya D, Oda S, Nakaura T, Funama Y, Yuki H, et al. CT venography after knee replacement surgery: comparison of dual-energy CT-based monochromatic imaging and single-energy metal artifact reduction techniques on a 320-row CT scanner. Acta Radiol Open. 2017;6:1–8.Google Scholar
- 14.Sofue K, Yoshikawa T, Ohno Y, Negi N, Inokawa H, Sugihara N, et al. Improved image quality in abdominal CT in patients who underwent treatment for hepatocellular carcinoma with small metal implants using a raw data-based metal artifact reduction algorithm. Eur Radiol. 2017;27:2978–88.CrossRefPubMedGoogle Scholar
- 24.Makimoto S, Hiraki T, Gobara H, Fujiwara H, Iguchi T, Matsui Y, et al. Association between reperfusion and shrinkage percentage of the aneurysmal sac after embolization of pulmonary arteriovenous malformation: evaluation based on contrast-enhanced thin-section CT images. Jpn J Radiol. 2014;32:266–73.CrossRefPubMedGoogle Scholar