Ultrasound thermal monitoring with an external ultrasound source for customized bipolar RF ablation shapes
- 183 Downloads
Thermotherapy is a clinical procedure which delivers thermal energy to a target, and it has been applied for various medical treatments. Temperature monitoring during thermotherapy is important to achieve precise and reproducible results. Medical ultrasound can be used for thermal monitoring and is an attractive medical imaging modality due to its advantages including non-ionizing radiation, cost-effectiveness and portability. We propose an ultrasound thermal monitoring method using a speed-of-sound tomographic approach coupled with a biophysical heat diffusion model.
We implement an ultrasound thermometry approach using an external ultrasound source. We reconstruct the speed-of-sound images using time-of-flight information from the external ultrasound source and convert the speed-of-sound information into temperature by using the a priori knowledge brought by a biophysical heat diffusion model.
Customized treatment shapes can be created using switching channels of radio frequency bipolar needle electrodes. Simulations of various ablation lesion shapes in the temperature range of 21–59 \(^\circ \)C are performed to study the feasibility of the proposed method. We also evaluated our method with ex vivo porcine liver experiments, in which we generated temperature images between 22 and 45 \(^\circ \)C.
In this paper, we present a proof of concept showing the feasibility of our ultrasound thermal monitoring method. The proposed method could be applied to various thermotherapy procedures by only adding an ultrasound source.
KeywordsThermal monitoring Speed-of-sound reconstruction Ultrasound RFA modeling Bipolar ablation Hyperthermia Ablation therapy Thermotherapy
The research reported in this paper was supported by the National Institute of Biomedical Imaging and Bioengineering of the National Institutes of Health under Award Number R01EB021396 and National Science Foundation under Proposal Number 1653322.
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
For this type of study formal consent is not required.
This article does not contain patient data.
- 1.Ali G, Krug JW, Friebe M (2017) A four-electrode radiofrequency ablation system designed for more complex and tumor specific ablation patterns. In: Proceedings of BioSpine 2017, 6th international Congress on biotechnologies for spinal surgeryGoogle Scholar
- 6.Duric N, Boyd N, Littrup P, Sak M, Myc L, Li C, West E, Minkin S, Martin L, Yaffe M, Schmidt S, Faiz M, Shen J, Melnichouk O, Li Q, Albrecht T (2013) Breast density measurements with ultrasound tomography: a comparison with film and digital mammography. Med Phys 40(1):013501CrossRefPubMedPubMedCentralGoogle Scholar
- 8.Norton SJ, Testardi LR, Wadley HNG (Oct 1983) Reconstructing internal temperature distributions from ultrasonic time-of-flight tomography and dimensional resonance measurements. In: 1983 ultrasonics symposium, pp 850–855Google Scholar
- 10.Kim Y, Audigier C, Ellens N, Boctor EM (2017) A novel 3D ultrasound thermometry method for HIFU ablation using an ultrasound element. In: 2017 IEEE international ultrasonics symposium (IUS), pp 1–4Google Scholar
- 12.Audigier C, Mansi T, Delingette H, Rapaka S, Passerini T, Mihalef V, Jolly M-P, Pop R, Diana M, Soler L, Kamen A, Comaniciu D, Ayache N (2017) Comprehensive preclinical evaluation of a multi-physics model of liver tumor radiofrequency ablation. Int J Comput Assist Radiol Surg 12(9):1543–1559CrossRefPubMedGoogle Scholar
- 18.Guo X, Kang H-J, Etienne-Cummings R, Boctor EM (2014) Active ultrasound pattern injection system (AUSPIS) for interventional tool guidance. PLoS ONE 9(10):1–13Google Scholar