Kalman filter-based EM-optical sensor fusion for needle deflection estimation
- 136 Downloads
In many clinical procedures such as cryoablation that involves needle insertion, accurate placement of the needle’s tip at the desired target is the major issue for optimizing the treatment and minimizing damage to the neighboring anatomy. However, due to the interaction force between the needle and tissue, considerable error in intraoperative tracking of the needle tip can be observed as needle deflects.
In this paper, measurements data from an optical sensor at the needle base and a magnetic resonance (MR) gradient field-driven electromagnetic (EM) sensor placed 10 cm from the needle tip are used within a model-integrated Kalman filter-based sensor fusion scheme. Bending model-based estimations and EM-based direct estimation are used as the measurement vectors in the Kalman filter, thus establishing an online estimation approach.
Static tip bending experiments show that the fusion method can reduce the mean error of the tip position estimation from 29.23 mm of the optical sensor-based approach to 3.15 mm of the fusion-based approach and from 39.96 to 6.90 mm, at the MRI isocenter and the MRI entrance, respectively.
This work established a novel sensor fusion scheme that incorporates model information, which enables real-time tracking of needle deflection with MRI compatibility, in a free-hand operating setup.
KeywordsSensor fusion Needle deflection estimation Kalman filter (KF) Surgical navigation
Compliance with ethical standards
Conflict of interest
The authors declare that they have no conflict of interest.
- 4.Park YL, Elayaperumal S, Daniel B, Ryu SC, Shin M, Savall J, Black RJ, Moslehi B, Cutkosky MR (2010) Real-time estimation of 3-D needle shape and deflection for MRI-guided interventions. IEEE/ASME Trans Mechatron 15(6):906–915Google Scholar
- 8.Tesei M, Saccomandi P, Massaroni C, Quarta R, Carassiti M, Schena E, Setola R (2016) A cost-effective, non-invasive system for pressure monitoring during epidural needle insertion: Design, development and bench tests. In: IEEE 38th annual international conference of the engineering in medicine and biology society (EMBC), IEEE, pp 194–197Google Scholar
- 9.Gao W, Jiang B, Kacher DF, Fetics B, Nevo E, Lee TC, Jayender J (2017) Real-time probe tracking using EM-optical sensor for MRI-guided cryoablation. Int J Med Rob. https://doi.org/10.1002/rcs.1871
- 10.Dorileo E, Zemiti N, Poignet P (2015) Needle deflection prediction using adaptive slope model. In: International conference on advanced robotics (ICAR), IEEE, pp 60–65Google Scholar
- 11.Roesthuis RJ, Van Veen YR, Jahya A, Misra S (2011) Mechanics of needle-tissue interaction. In: IEEE/RSJ international conference on intelligent robots and systems (IROS), IEEE, pp 2557–2563Google Scholar
- 12.Asadian A, Kermani MR, Patel RV (2011) An analytical model for deflection of flexible needles during needle insertion. In: IEEE/RSJ international conference on intelligent robots and systems (IROS), IEEE, pp 2551–2556Google Scholar
- 13.Park W, Reed KB, Okamura AM, Chirikjian GS (2010) Estimation of model parameters for steerable needles. In: IEEE International conference on robotics and automation (ICRA), 2010, IEEE, pp 3703–3708Google Scholar
- 16.Jiang B, Gao W, Kacher DF, Lee TC, Jayender J (2016) Kalman filter based data fusion for needle deflection estimation using optical-em sensor. In: International conference on medical image computing and computer-assisted intervention, Springer, pp 457–464Google Scholar
- 17.Roth A, Nevo E (2015) Method and apparatus to estimate location and orientation of objects during magnetic resonance imaging. US Patent 9,037,213Google Scholar
- 20.Chui CK, Chen G (2008) Kalman filtering: with real-time applications. Springer, BerlinGoogle Scholar
- 22.Yan KG, Podder T, Xiao D, Yu Y, Liu TI, Ling KV, Ng WS (2006) Online parameter estimation for surgical needle steering model. In: Medical image computing and computer-assisted intervention–MICCAI 2006, Springer, pp 321–329Google Scholar
- 23.www.symbowmed.com. Accessed 14 Mar 2016
- 24.Nevo E (2003) Method and apparatus to estimate location and orientation of objects during magnetic resonance imaging. US Patent 6,516,213Google Scholar
- 25.Yaniv Z (2015) Which pivot calibration? In: SPIE medical imaging, international society for optics and photonics, pp 941527–941527Google Scholar
- 26.Crouch JR, Schneider CM, Wainer J, Okamura AM (2005) A velocity-dependent model for needle insertion in soft tissue. In: Medical image computing and computer-assisted intervention–Miccai 2005, Springer, pp 624–632Google Scholar
- 27.Bar-Shalom Y, Li XR, Kirubarajan T (2004) Estimation with applications to tracking and navigation: theory algorithms and software. Wiley, HobokenGoogle Scholar